//
// ctdDriver.cfg
//
// * This config file is normally used for CTD board 1.
// * Used this config file if 1 seabird tasks is used.
//
// Calibration constants TP4122 and CP2514
// For boardset #1 in ALTEX Science Payload
// 
// Updated with new sensor coefficients 24 July 2002 dg
// Swapped O2 sensor 25 Nov 03 dg 
// Updated with new sensor coefficients 10 Aug 2004 drthom
// Updated with new sensor coefficients 3 Jan 2004 dougie
// Updated with new sensor coefficients 3 May 2005 dougie
// Updated with new sensor coefficients 15 Aug 2005 drthom
// Updated with new sensor coefficients 27 AUg 2005 hthomas
// Updated with new sensor coefficients 24 Apr 2006 drthom
// Updated with new sensor coefficients 19 Sep 2006 dougie
// Updated with new sensor coefficients 28 Dec 2006 drthom
// Updated with new sensor coefficients 26 Jun 2007 dougie
// Updated with new sensor coefficients 19 Jun 2008 dougie
// Added CDOM fluorometer on voltage channel 5 15 Nov 2007 dc
// Updated with new sensor coefficients 18 Feb 2009 dougie
// Updated with new sensor coefficients 08 Jan 2010 dougie
// Updated with new sensor coefficients 17 Jun 2010 dougie
// changed CDOM to Rhodamine June 28 2010 hjt
// kept the analog channel set up for rhodamine October 7 2010 hjt
// Updated with new sensor coefficients 09 Feb 2011 dougie

//TempSerialNumber = 4152; 
t_a = 3.68121126e-03;
t_b = 6.04158512e-04;
t_c = 1.63156852e-05;
t_d = 2.04277895e-06;
t_f0 = 3304.280;

//CondSerialNumber = 2478;
c_a = 6.12730389e-04;
c_b = 1.52109695e00;
c_c = -1.04891747e01;  
c_d = -8.42893300e-05;
c_m = 3.3;
eps = -9.57e-08;

//OxygenSerialNumber = 0461;
SOc = 0.5556;
BOc = 0.0000;
Voff = -0.5281;
TCor = -0.0007;
PCor = 1.350e-04;

//Secondar Oxygen Serial Number = 0564
//Cal Date: 26-Feb-11
//SOc = 0.4294;
//BOc = 0.0000;
//Voff = -0.5307
//TCor = 0.0005;
//PCor = 1.350e-4;

//CDOMSerialNumber = FLCDMBA-812;
//ScaleFactor = 50;
//DarkCounts = 0.047;

//RhodaminSerialNumber = FLRHMBA-1376
//ScaleFactor=0.141
//DarkCounts=55

NUMVOLTAGES = 6;
v1name = spare;
v2name = spare;
v3name = humidity; 
v4name = flow1;
v5name = SecondaryDissolvedO2;
v6name = dissolvedO2;

//whether or not the boardset needs the pressure sensor config cmds sent
PressureConfig = N;
