function [calib]=hydrocalTC4034SN1101051();
disp('Note:  Hydrophone Not Calibrated Between 0-5 KHz')
disp('Note:  Preamp Not Calibrated With Hydrophone and May Suffer From Improper Impedance Match')
disp('Note:  Preamp Phase Not Included In Calibration Data')
calib=zeros(36,2);
calib(1,1)=0; calib(1,2)=-214;
calib(2,1)=5e3; calib(2,2)=-214;
calib(3,1)=6e3; calib(3,2)=-215;
calib(4,1)=7e3; calib(4,2)=-215;
calib(5,1)=8e3; calib(5,2)=-214;
calib(6,1)=9e3; calib(6,2)=-215;
calib(7,1)=1e4; calib(7,2)=-215;
calib(8,1)=2e4; calib(8,2)=-215;
calib(9,1)=3e4; calib(9,2)=-216;
calib(10,1)=4e4; calib(10,2)=-216;
calib(11,1)=5e4; calib(11,2)=-216;
calib(12,1)=6e4; calib(12,2)=-216;
calib(13,1)=7e4; calib(13,2)=-216;
calib(14,1)=8e4; calib(14,2)=-216;
calib(15,1)=9e4; calib(15,2)=-217;
calib(16,1)=1.0e5; calib(16,2)=-217;
calib(17,1)=1.2e5; calib(17,2)=-218;
calib(18,1)=1.4e5; calib(18,2)=-218;
calib(19,1)=1.6e5; calib(19,2)=-218;
calib(20,1)=1.8e5; calib(20,2)=-218;
calib(21,1)=2.0e5; calib(21,2)=-218;
calib(22,1)=2.2e5; calib(22,2)=-218;
calib(23,1)=2.4e5; calib(23,2)=-216;
calib(24,1)=2.6e5; calib(24,2)=-215;
calib(25,1)=2.8e5; calib(25,2)=-214;
calib(26,1)=3.0e5; calib(26,2)=-213;
calib(27,1)=3.2e5; calib(27,2)=-213;
calib(28,1)=3.4e5; calib(28,2)=-214;
calib(29,1)=3.6e5; calib(29,2)=-214;
calib(30,1)=3.8e5; calib(30,2)=-215;
calib(31,1)=4.0e5; calib(31,2)=-218;
calib(32,1)=4.2e5; calib(32,2)=-221;
calib(33,1)=4.4e5; calib(33,2)=-222;
calib(34,1)=4.6e5; calib(34,2)=-222;
calib(35,1)=4.8e5; calib(35,2)=-224;
calib(36,1)=5.0e5; calib(36,2)=-226;

