$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	02-Dec-2016	14:51:25.745	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],375.288,AirSaturation[%],127.899,Temperature[Deg.C],18.364,CalPhase[Deg],26.874,TCPhase[Deg],27.695,C1RPh[Deg],35.701,C2RPh[Deg],8.005,C1Amp[mV],546.4,C2Amp[mV],803.7,RawTemp[mV],296.2							
$	02-Dec-2016	14:51:25.745	Surface Ops	Optode Message	0.150000																											#							
$	02-Dec-2016	14:52:50.494	Surface Ops	FLBB Message	0.110000																										99/99/99,99:99:99,695,56,700,146,539								
$	02-Dec-2016	14:52:50.601	Surface Ops	FLBB Message	0.110000																										99/99/99,99:99:99,695,50,700,156,539								
$	02-Dec-2016	14:52:50.610	Surface Ops	FLBB Message	0.110000																																		
$	02-Dec-2016	14:52:50.809	Surface Ops	FLBB Message	0.110000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:52:50.818	Surface Ops	FLBB Message	0.110000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:52:50.921	Surface Ops	FLBB Message	0.110000																										Ave 3								
$	02-Dec-2016	14:52:51.024	Surface Ops	FLBB Message	0.110000																										Pkt 2								
$	02-Dec-2016	14:52:51.339	Surface Ops	Optode Message	0.380000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.885,AirSaturation[%],102.172,Temperature[Deg.C],17.866,CalPhase[Deg],29.347,TCPhase[Deg],30.168,C1RPh[Deg],38.175,C2RPh[Deg],8.006,C1Amp[mV],630.6,C2Amp[mV],813.7,RawTemp[mV],312.1							
$	02-Dec-2016	14:52:51.340	Surface Ops	Optode Message	0.380000																											#							
$	02-Dec-2016	14:52:51.676	Surface Ops	FLBB Message	0.380000																										99/99/99,99:99:99,695,55,700,186,539								
$	02-Dec-2016	14:52:51.878	Surface Ops	FLBB Message	0.380000																										99/99/99,99:99:99,695,56,700,194,539								
$	02-Dec-2016	14:52:51.886	Surface Ops	FLBB Message	0.380000																																		
$	02-Dec-2016	14:52:51.989	Surface Ops	FLBB Message	0.380000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:52:52.092	Surface Ops	FLBB Message	0.380000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:52:52.195	Surface Ops	FLBB Message	0.380000																										Ave 3								
$	02-Dec-2016	14:52:52.396	Surface Ops	FLBB Message	0.380000																										Pkt 2								
$	02-Dec-2016	14:56:14.659	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,48,700,120,539								
$	02-Dec-2016	14:56:14.609	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.650,AirSaturation[%],100.919,Temperature[Deg.C],17.308,CalPhase[Deg],29.587,TCPhase[Deg],30.408,C1RPh[Deg],38.416,C2RPh[Deg],8.008,C1Amp[mV],639.4,C2Amp[mV],818.2,RawTemp[mV],329.9							
$	02-Dec-2016	14:56:14.613	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	14:56:15.008	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,54,700,117,539								
$	02-Dec-2016	14:56:15.112	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	14:56:15.217	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:56:15.416	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:56:15.423	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	14:56:15.528	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	14:56:17.143	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:56:28.733	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:57:15.628	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,51,700,118,539								
$	02-Dec-2016	14:57:14.766	Anchor	Optode Message	9.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.079,AirSaturation[%],123.388,Temperature[Deg.C],18.360,CalPhase[Deg],27.242,TCPhase[Deg],28.063,C1RPh[Deg],36.004,C2RPh[Deg],7.941,C1Amp[mV],536.0,C2Amp[mV],788.7,RawTemp[mV],296.3							
$	02-Dec-2016	14:57:14.770	Anchor	Optode Message	9.490000																											#							
$	02-Dec-2016	14:57:15.958	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,51,700,118,539								
$	02-Dec-2016	14:57:15.968	Anchor	FLBB Message	9.490000																																		
$	02-Dec-2016	14:57:16.071	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:57:16.174	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:57:16.277	Anchor	FLBB Message	9.490000																										Ave 3								
$	02-Dec-2016	14:57:16.476	Anchor	FLBB Message	9.490000																										Pkt 2								
$	02-Dec-2016	14:57:15.734	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:57:27.233	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:58:16.728	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,118,539								
$	02-Dec-2016	14:58:14.611	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.104,AirSaturation[%],125.115,Temperature[Deg.C],18.366,CalPhase[Deg],27.098,TCPhase[Deg],27.919,C1RPh[Deg],35.920,C2RPh[Deg],8.001,C1Amp[mV],531.6,C2Amp[mV],788.0,RawTemp[mV],296.1							
$	02-Dec-2016	14:58:14.614	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	14:58:17.153	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,48,700,114,539								
$	02-Dec-2016	14:58:17.163	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	14:58:17.362	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:58:17.370	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:58:17.473	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	14:58:17.575	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	14:58:16.642	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:58:28.233	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:59:14.976	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,122,539								
$	02-Dec-2016	14:59:14.643	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:59:14.771	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],369.084,AirSaturation[%],125.800,Temperature[Deg.C],18.370,CalPhase[Deg],27.041,TCPhase[Deg],27.862,C1RPh[Deg],35.865,C2RPh[Deg],8.003,C1Amp[mV],529.9,C2Amp[mV],787.6,RawTemp[mV],296.0							
$	02-Dec-2016	14:59:14.771	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	14:59:15.421	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,52,700,124,539								
$	02-Dec-2016	14:59:15.527	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	14:59:15.629	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:59:15.828	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:59:15.836	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	14:59:15.939	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	14:59:26.234	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	15:00:15.488	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,115,539								
$	02-Dec-2016	15:00:14.607	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],369.764,AirSaturation[%],126.067,Temperature[Deg.C],18.384,CalPhase[Deg],27.018,TCPhase[Deg],27.839,C1RPh[Deg],35.842,C2RPh[Deg],8.003,C1Amp[mV],529.3,C2Amp[mV],787.5,RawTemp[mV],295.6							
$	02-Dec-2016	15:00:14.610	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	15:00:15.927	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,117,539								
$	02-Dec-2016	15:00:15.936	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	15:00:16.038	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:00:16.143	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:00:16.246	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	15:00:16.445	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	15:00:15.816	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	15:00:27.233	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	15:01:16.634	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,117,539								
$	02-Dec-2016	15:01:14.617	Start CP Mode	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.107,AirSaturation[%],126.199,Temperature[Deg.C],18.390,CalPhase[Deg],27.006,TCPhase[Deg],27.827,C1RPh[Deg],35.771,C2RPh[Deg],7.944,C1Amp[mV],528.8,C2Amp[mV],787.4,RawTemp[mV],295.4							
$	02-Dec-2016	15:01:16.954	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,121,539								
$	02-Dec-2016	15:01:14.621	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	15:01:17.073	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	15:01:17.272	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:01:17.281	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:01:17.479	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	15:01:17.582	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	15:01:17.909	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,121,539								
$	02-Dec-2016	15:01:15.662	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.386,AirSaturation[%],126.269,Temperature[Deg.C],18.380,CalPhase[Deg],27.002,TCPhase[Deg],27.823,C1RPh[Deg],35.825,C2RPh[Deg],8.003,C1Amp[mV],528.7,C2Amp[mV],787.3,RawTemp[mV],295.7							
$	02-Dec-2016	15:01:15.665	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	15:01:18.239	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,117,539								
$	02-Dec-2016	15:01:18.247	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	15:01:18.352	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:01:18.455	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:01:18.654	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	15:01:18.661	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	15:01:16.838	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	15:02:08.144	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	15:02:14.258	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	02-Dec-2016	15:02:14.259	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	02-Dec-2016	15:02:22.987	DeAnchor	Unparsed SBE41 Message						S>    9.56, 18.3491, 35.2108  																													
$	02-Dec-2016	15:04:29.048	Ascend	FLBB Message	5.420000																										99/99/99,99:99:99,695,50,700,113,539								
$	02-Dec-2016	15:04:29.033	Ascend	Optode Message	5.420000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.875,AirSaturation[%],126.436,Temperature[Deg.C],18.380,CalPhase[Deg],26.988,TCPhase[Deg],27.809,C1RPh[Deg],35.799,C2RPh[Deg],7.990,C1Amp[mV],528.8,C2Amp[mV],783.6,RawTemp[mV],295.7							
$	02-Dec-2016	15:04:29.033	Ascend	Optode Message	5.420000																											#							
$	02-Dec-2016	15:04:29.715	Ascend	FLBB Message	5.130000																										99/99/99,99:99:99,695,53,700,119,539								
$	02-Dec-2016	15:04:29.725	Ascend	FLBB Message	5.130000																																		
$	02-Dec-2016	15:04:29.924	Ascend	FLBB Message	5.130000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:04:29.932	Ascend	FLBB Message	5.130000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:04:30.131	Ascend	FLBB Message	5.130000																										Ave 3								
$	02-Dec-2016	15:04:30.139	Ascend	FLBB Message	5.130000																										Pkt 2								
$	02-Dec-2016	15:04:42.763	Ascend	FLBB Message	3.240000																										99/99/99,99:99:99,695,50,700,116,539								
$	02-Dec-2016	15:04:42.868	Ascend	FLBB Message	3.240000																										99/99/99,99:99:99,695,51,700,114,539								
$	02-Dec-2016	15:04:42.879	Ascend	FLBB Message	3.240000																																		
$	02-Dec-2016	15:04:42.981	Ascend	FLBB Message	3.240000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:04:43.084	Ascend	FLBB Message	3.240000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:04:43.187	Ascend	FLBB Message	3.240000																										Ave 3								
$	02-Dec-2016	15:04:43.387	Ascend	FLBB Message	3.240000																										Pkt 2								
$	02-Dec-2016	15:04:42.453	Ascend	Optode Message	3.240000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.273,AirSaturation[%],127.208,Temperature[Deg.C],18.363,CalPhase[Deg],26.929,TCPhase[Deg],27.750,C1RPh[Deg],35.690,C2RPh[Deg],7.940,C1Amp[mV],534.2,C2Amp[mV],793.6,RawTemp[mV],296.2							
$	02-Dec-2016	15:04:42.454	Ascend	Optode Message	3.240000																											#							
$	02-Dec-2016	15:04:59.522	Ascend	FLBB Message	1.640000																										99/99/99,99:99:99,695,53,700,117,539								
$	02-Dec-2016	15:04:58.310	Ascend	Optode Message	1.640000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.979,AirSaturation[%],127.449,Temperature[Deg.C],18.363,CalPhase[Deg],26.910,TCPhase[Deg],27.731,C1RPh[Deg],35.736,C2RPh[Deg],8.005,C1Amp[mV],540.1,C2Amp[mV],798.3,RawTemp[mV],296.2							
$	02-Dec-2016	15:04:58.310	Ascend	Optode Message	1.640000																											#							
$	02-Dec-2016	15:04:59.854	Ascend	FLBB Message	1.640000																										99/99/99,99:99:99,695,56,700,114,539								
$	02-Dec-2016	15:04:59.864	Ascend	FLBB Message	1.640000																																		
$	02-Dec-2016	15:04:59.967	Ascend	FLBB Message	1.640000																										Ser FLBB2K-3310								
$	02-Dec-2016	15:05:00.069	Ascend	FLBB Message	1.640000																										Ver FLNTU 4.08								
$	02-Dec-2016	15:05:00.268	Ascend	FLBB Message	1.640000																										Ave 3								
$	02-Dec-2016	15:05:00.372	Ascend	FLBB Message	1.640000																										Pkt 2								
$	02-Dec-2016	15:05:03.524	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	02-Dec-2016	15:05:05.883	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	15:05:06.979	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	02-Dec-2016	15:05:06.979	Init Profile	Unparsed SBE41 Message						powering down  																													
$	02-Dec-2016	15:05:32.323	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	02-Dec-2016	15:06:31.996	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	15:06:38.076	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	02-Dec-2016	15:06:38.247	Dump CP Data	Unparsed SBE41 Message						samples = 165, maxPress = 9.59  																													
$	02-Dec-2016	15:06:46.197	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	02-Dec-2016	15:06:46.882	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	02-Dec-2016	15:06:47.073	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	02-Dec-2016	15:06:47.210	Dump CP Data	Unparsed SBE41 Message						  																													
$	02-Dec-2016	15:06:47.228	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	02-Dec-2016	15:06:53.252	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	02-Dec-2016	15:06:53.253	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.352	Dump CP Data	Unparsed SBE41 Message						    1.95,  18.3484,  35.2106, 20  																													
$	02-Dec-2016	15:06:53.353	Dump CP Data	Unparsed SBE41 Message						    3.93,  18.3480,  35.2106, 14  																													
$	02-Dec-2016	15:06:53.440	Dump CP Data	Unparsed SBE41 Message						    5.98,  18.3483,  35.2106, 13  																													
$	02-Dec-2016	15:06:53.454	Dump CP Data	Unparsed SBE41 Message						    8.08,  18.3487,  35.2106, 18  																													
$	02-Dec-2016	15:06:53.551	Dump CP Data	Unparsed SBE41 Message						    9.47,  18.3488,  35.2107, 100  																													
$	02-Dec-2016	15:06:53.552	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.596	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.641	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.685	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.739	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	15:06:53.856	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	15:06:59.344	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	02-Dec-2016	15:06:59.345	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.444	Dump CP Data	Unparsed SBE41 Message						001447AC898B14  																													
$	02-Dec-2016	15:06:59.445	Dump CP Data	Unparsed SBE41 Message						002747AC898B0E  																													
$	02-Dec-2016	15:06:59.445	Dump CP Data	Unparsed SBE41 Message						003C47AC898B0D  																													
$	02-Dec-2016	15:06:59.446	Dump CP Data	Unparsed SBE41 Message						005147AD898B12  																													
$	02-Dec-2016	15:06:59.461	Dump CP Data	Unparsed SBE41 Message						005F47AD898B64  																													
$	02-Dec-2016	15:06:59.486	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.512	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.537	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.563	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.598	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	15:06:59.618	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	15:07:05.257	Dump CP Data	Unparsed SBE41 Message						S>  																													
