$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	02-Dec-2016	14:06:45.838	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:06:49.849	Surface Ops	Optode Message	1.330000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],375.360,AirSaturation[%],127.919,Temperature[Deg.C],18.363,CalPhase[Deg],26.873,TCPhase[Deg],27.694,C1RPh[Deg],35.700,C2RPh[Deg],8.006,C1Amp[mV],546.4,C2Amp[mV],803.8,RawTemp[mV],296.2							
$	02-Dec-2016	14:06:49.849	Surface Ops	Optode Message	1.330000																											#							
$	02-Dec-2016	14:06:57.404	Surface Ops	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:08:14.916	Surface Ops	FLBB Message	0.260000																										99/99/99,99:99:99,695,49,700,201,539								
$	02-Dec-2016	14:08:15.121	Surface Ops	FLBB Message	0.260000																										99/99/99,99:99:99,695,52,700,204,539								
$	02-Dec-2016	14:08:15.129	Surface Ops	FLBB Message	0.260000																																		
$	02-Dec-2016	14:08:15.234	Surface Ops	FLBB Message	0.260000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:08:15.337	Surface Ops	FLBB Message	0.260000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:08:15.538	Surface Ops	FLBB Message	0.260000																										Ave 3								
$	02-Dec-2016	14:08:15.642	Surface Ops	FLBB Message	0.260000																										Pkt 2								
$	02-Dec-2016	14:08:15.537	Surface Ops	Optode Message	0.260000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.718,AirSaturation[%],102.373,Temperature[Deg.C],17.988,CalPhase[Deg],29.303,TCPhase[Deg],30.124,C1RPh[Deg],38.073,C2RPh[Deg],7.949,C1Amp[mV],629.2,C2Amp[mV],812.2,RawTemp[mV],308.2							
$	02-Dec-2016	14:08:15.641	Surface Ops	Optode Message	0.260000																											#							
$	02-Dec-2016	14:08:18.630	AB Retract Fast	FLBB Message	0.260000																										99/99/99,99:99:99,695,53,700,193,539								
$	02-Dec-2016	14:08:18.644	AB Retract Fast	FLBB Message	0.260000																										99/99/99,99:99:99,695,50,700,177,539								
$	02-Dec-2016	14:08:18.845	AB Retract Fast	FLBB Message	0.260000																																		
$	02-Dec-2016	14:08:18.947	AB Retract Fast	FLBB Message	0.260000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:08:19.052	AB Retract Fast	FLBB Message	0.260000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:08:19.059	AB Retract Fast	FLBB Message	0.260000																										Ave 3								
$	02-Dec-2016	14:08:19.164	AB Retract Fast	FLBB Message	0.260000																										Pkt 2								
$	02-Dec-2016	14:11:39.136	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,47,700,802,539								
$	02-Dec-2016	14:11:38.964	Anchor	Optode Message	9.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.509,AirSaturation[%],100.831,Temperature[Deg.C],17.607,CalPhase[Deg],29.542,TCPhase[Deg],30.363,C1RPh[Deg],38.371,C2RPh[Deg],8.008,C1Amp[mV],637.7,C2Amp[mV],816.2,RawTemp[mV],320.3							
$	02-Dec-2016	14:11:38.967	Anchor	Optode Message	9.490000																											#							
$	02-Dec-2016	14:11:39.469	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,49,700,153,539								
$	02-Dec-2016	14:11:39.669	Anchor	FLBB Message	9.490000																																		
$	02-Dec-2016	14:11:39.676	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:11:39.781	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:11:39.884	Anchor	FLBB Message	9.490000																										Ave 3								
$	02-Dec-2016	14:11:39.986	Anchor	FLBB Message	9.490000																										Pkt 2								
$	02-Dec-2016	14:11:40.813	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:11:52.404	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:12:38.813	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:12:40.233	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,128,539								
$	02-Dec-2016	14:12:39.064	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],361.290,AirSaturation[%],123.119,Temperature[Deg.C],18.361,CalPhase[Deg],27.264,TCPhase[Deg],28.085,C1RPh[Deg],36.087,C2RPh[Deg],8.002,C1Amp[mV],536.7,C2Amp[mV],788.5,RawTemp[mV],296.3							
$	02-Dec-2016	14:12:39.068	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	14:12:40.563	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,124,539								
$	02-Dec-2016	14:12:40.571	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	14:12:40.772	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:12:40.780	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:12:40.882	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	14:12:40.985	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	14:12:50.404	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:13:41.057	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,126,539								
$	02-Dec-2016	14:13:38.978	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.240,AirSaturation[%],125.158,Temperature[Deg.C],18.365,CalPhase[Deg],27.095,TCPhase[Deg],27.916,C1RPh[Deg],35.918,C2RPh[Deg],8.003,C1Amp[mV],531.6,C2Amp[mV],787.9,RawTemp[mV],296.2							
$	02-Dec-2016	14:13:38.980	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	14:13:41.386	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,135,539								
$	02-Dec-2016	14:13:41.395	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	14:13:41.497	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:13:41.600	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:13:41.801	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	14:13:41.809	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	14:13:39.814	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:13:51.404	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:14:39.036	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,124,539								
$	02-Dec-2016	14:14:38.961	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],369.485,AirSaturation[%],125.934,Temperature[Deg.C],18.369,CalPhase[Deg],27.031,TCPhase[Deg],27.852,C1RPh[Deg],35.855,C2RPh[Deg],8.004,C1Amp[mV],529.7,C2Amp[mV],787.6,RawTemp[mV],296.0							
$	02-Dec-2016	14:14:38.964	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	14:14:39.368	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,49,700,128,539								
$	02-Dec-2016	14:14:39.567	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	14:14:39.575	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:14:39.678	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:14:39.781	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	14:14:39.886	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	14:14:40.814	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:14:52.404	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:15:38.814	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:15:40.129	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,58,700,132,539								
$	02-Dec-2016	14:15:39.055	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.643,AirSaturation[%],126.336,Temperature[Deg.C],18.372,CalPhase[Deg],26.998,TCPhase[Deg],27.819,C1RPh[Deg],35.822,C2RPh[Deg],8.003,C1Amp[mV],528.8,C2Amp[mV],787.4,RawTemp[mV],295.9							
$	02-Dec-2016	14:15:39.058	Anchor	Optode Message	9.520000																											#							
$	02-Dec-2016	14:15:40.554	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,128,539								
$	02-Dec-2016	14:15:40.562	Anchor	FLBB Message	9.520000																																		
$	02-Dec-2016	14:15:40.761	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:15:40.769	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:15:40.873	Anchor	FLBB Message	9.520000																										Ave 3								
$	02-Dec-2016	14:15:40.976	Anchor	FLBB Message	9.520000																										Pkt 2								
$	02-Dec-2016	14:15:50.404	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:16:41.272	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,129,539								
$	02-Dec-2016	14:16:39.048	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],371.425,AirSaturation[%],126.610,Temperature[Deg.C],18.375,CalPhase[Deg],26.975,TCPhase[Deg],27.796,C1RPh[Deg],35.800,C2RPh[Deg],8.004,C1Amp[mV],528.1,C2Amp[mV],787.4,RawTemp[mV],295.8							
$	02-Dec-2016	14:16:39.048	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	14:16:41.603	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,124,539								
$	02-Dec-2016	14:16:41.611	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	14:16:41.716	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:16:41.819	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:16:42.017	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	14:16:42.025	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	14:16:39.814	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	14:16:42.338	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,126,539								
$	02-Dec-2016	14:16:40.034	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],372.171,AirSaturation[%],126.854,Temperature[Deg.C],18.371,CalPhase[Deg],26.956,TCPhase[Deg],27.777,C1RPh[Deg],35.781,C2RPh[Deg],8.003,C1Amp[mV],527.5,C2Amp[mV],787.3,RawTemp[mV],296.0							
$	02-Dec-2016	14:16:40.036	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	14:16:42.763	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,49,700,125,539								
$	02-Dec-2016	14:16:42.867	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	14:16:42.877	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:16:42.980	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:16:43.083	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	14:16:43.185	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	14:16:51.404	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:16:56.179	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:17:02.272	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	02-Dec-2016	14:17:02.273	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	02-Dec-2016	14:17:10.957	DeAnchor	Unparsed SBE41 Message						S>    9.50, 18.3483, 35.2108  																													
$	02-Dec-2016	14:19:20.282	Ascend	FLBB Message	5.430000																										99/99/99,99:99:99,695,50,700,118,539								
$	02-Dec-2016	14:19:19.969	Ascend	Optode Message	5.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],372.364,AirSaturation[%],126.921,Temperature[Deg.C],18.371,CalPhase[Deg],26.951,TCPhase[Deg],27.772,C1RPh[Deg],35.766,C2RPh[Deg],7.994,C1Amp[mV],527.6,C2Amp[mV],783.6,RawTemp[mV],295.9							
$	02-Dec-2016	14:19:19.969	Ascend	Optode Message	5.150000																											#							
$	02-Dec-2016	14:19:20.759	Ascend	FLBB Message	5.150000																										99/99/99,99:99:99,695,52,700,121,539								
$	02-Dec-2016	14:19:20.863	Ascend	FLBB Message	5.150000																																		
$	02-Dec-2016	14:19:20.966	Ascend	FLBB Message	5.150000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:19:21.069	Ascend	FLBB Message	5.150000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:19:21.268	Ascend	FLBB Message	5.150000																										Ave 3								
$	02-Dec-2016	14:19:21.277	Ascend	FLBB Message	5.150000																										Pkt 2								
$	02-Dec-2016	14:19:35.775	Ascend	FLBB Message	2.880000																										99/99/99,99:99:99,695,52,700,120,539								
$	02-Dec-2016	14:19:35.761	Ascend	Optode Message	2.880000																											MEASUREMENT,4330,2592,O2Concentration[uM],374.190,AirSaturation[%],127.521,Temperature[Deg.C],18.363,CalPhase[Deg],26.904,TCPhase[Deg],27.725,C1RPh[Deg],35.668,C2RPh[Deg],7.942,C1Amp[mV],533.5,C2Amp[mV],793.5,RawTemp[mV],296.2							
$	02-Dec-2016	14:19:35.761	Ascend	Optode Message	2.880000																											#							
$	02-Dec-2016	14:19:36.129	Ascend	FLBB Message	2.880000																										99/99/99,99:99:99,695,50,700,122,539								
$	02-Dec-2016	14:19:36.327	Ascend	FLBB Message	2.880000																																		
$	02-Dec-2016	14:19:36.430	Ascend	FLBB Message	2.880000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:19:36.534	Ascend	FLBB Message	2.880000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:19:36.542	Ascend	FLBB Message	2.880000																										Ave 3								
$	02-Dec-2016	14:19:36.742	Ascend	FLBB Message	2.880000																										Pkt 2								
$	02-Dec-2016	14:19:52.476	Ascend	FLBB Message	1.220000																										99/99/99,99:99:99,695,51,700,122,539								
$	02-Dec-2016	14:19:51.163	Ascend	Optode Message	1.220000																											MEASUREMENT,4330,2592,O2Concentration[uM],374.726,AirSaturation[%],127.703,Temperature[Deg.C],18.362,CalPhase[Deg],26.890,TCPhase[Deg],27.711,C1RPh[Deg],35.717,C2RPh[Deg],8.006,C1Amp[mV],540.8,C2Amp[mV],799.2,RawTemp[mV],296.2							
$	02-Dec-2016	14:19:51.163	Ascend	Optode Message	1.220000																											#							
$	02-Dec-2016	14:19:52.818	Ascend	FLBB Message	1.220000																										99/99/99,99:99:99,695,48,700,115,539								
$	02-Dec-2016	14:19:52.924	Ascend	FLBB Message	1.220000																																		
$	02-Dec-2016	14:19:52.932	Ascend	FLBB Message	1.220000																										Ser FLBB2K-3310								
$	02-Dec-2016	14:19:53.035	Ascend	FLBB Message	1.220000																										Ver FLNTU 4.08								
$	02-Dec-2016	14:19:53.138	Ascend	FLBB Message	1.220000																										Ave 3								
$	02-Dec-2016	14:19:53.241	Ascend	FLBB Message	1.220000																										Pkt 2								
$	02-Dec-2016	14:19:53.465	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	02-Dec-2016	14:19:54.683	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	14:19:55.740	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	14:19:56.835	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	02-Dec-2016	14:19:56.835	Init Profile	Unparsed SBE41 Message						powering down  																													
$	02-Dec-2016	14:20:21.262	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	02-Dec-2016	14:21:20.867	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	14:21:26.942	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	02-Dec-2016	14:21:27.113	Dump CP Data	Unparsed SBE41 Message						samples = 167, maxPress = 9.58  																													
$	02-Dec-2016	14:21:35.063	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	02-Dec-2016	14:21:35.748	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	02-Dec-2016	14:21:35.943	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	02-Dec-2016	14:21:36.090	Dump CP Data	Unparsed SBE41 Message						  																													
$	02-Dec-2016	14:21:36.108	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	02-Dec-2016	14:21:42.119	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	02-Dec-2016	14:21:42.119	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.219	Dump CP Data	Unparsed SBE41 Message						    1.92,  18.3473,  35.2104, 19  																													
$	02-Dec-2016	14:21:42.220	Dump CP Data	Unparsed SBE41 Message						    3.98,  18.3479,  35.2104, 14  																													
$	02-Dec-2016	14:21:42.264	Dump CP Data	Unparsed SBE41 Message						    5.95,  18.3482,  35.2105, 13  																													
$	02-Dec-2016	14:21:42.313	Dump CP Data	Unparsed SBE41 Message						    8.05,  18.3482,  35.2105, 19  																													
$	02-Dec-2016	14:21:42.372	Dump CP Data	Unparsed SBE41 Message						    9.46,  18.3482,  35.2106, 102  																													
$	02-Dec-2016	14:21:42.417	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.526	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.526	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.629	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.630	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	14:21:42.630	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	14:21:48.221	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	02-Dec-2016	14:21:48.221	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.321	Dump CP Data	Unparsed SBE41 Message						001347AB898A13  																													
$	02-Dec-2016	14:21:48.322	Dump CP Data	Unparsed SBE41 Message						002847AC898A0E  																													
$	02-Dec-2016	14:21:48.322	Dump CP Data	Unparsed SBE41 Message						003C47AC898A0D  																													
$	02-Dec-2016	14:21:48.323	Dump CP Data	Unparsed SBE41 Message						005047AC898A13  																													
$	02-Dec-2016	14:21:48.339	Dump CP Data	Unparsed SBE41 Message						005F47AC898B66  																													
$	02-Dec-2016	14:21:48.364	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.389	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.414	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.440	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.474	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	14:21:48.494	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	14:21:54.127	Dump CP Data	Unparsed SBE41 Message						S>  																													
