$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	02-Dec-2016	17:09:55.877	Surface Ops	Optode Message	-0.360000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],374.646,AirSaturation[%],127.648,Temperature[Deg.C],18.352,CalPhase[Deg],26.896,TCPhase[Deg],27.717,C1RPh[Deg],35.661,C2RPh[Deg],7.944,C1Amp[mV],547.3,C2Amp[mV],804.1,RawTemp[mV],296.6							
$	02-Dec-2016	17:09:55.877	Surface Ops	Optode Message	-0.360000																											#							
$	02-Dec-2016	17:11:21.433	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,52,700,112,539								
$	02-Dec-2016	17:11:21.537	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,47,700,110,539								
$	02-Dec-2016	17:11:21.546	Surface Ops	FLBB Message	0.130000																																		
$	02-Dec-2016	17:11:21.748	Surface Ops	FLBB Message	0.130000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:11:21.757	Surface Ops	FLBB Message	0.130000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:11:21.861	Surface Ops	FLBB Message	0.130000																										Ave 3								
$	02-Dec-2016	17:11:21.965	Surface Ops	FLBB Message	0.130000																										Pkt 2								
$	02-Dec-2016	17:11:21.853	Surface Ops	Optode Message	0.130000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.716,AirSaturation[%],100.870,Temperature[Deg.C],17.593,CalPhase[Deg],29.540,TCPhase[Deg],30.361,C1RPh[Deg],38.365,C2RPh[Deg],8.004,C1Amp[mV],636.8,C2Amp[mV],813.8,RawTemp[mV],320.8							
$	02-Dec-2016	17:11:21.853	Surface Ops	Optode Message	0.130000																											#							
$	02-Dec-2016	17:11:22.393	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,50,700,112,539								
$	02-Dec-2016	17:11:22.497	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,47,700,112,539								
$	02-Dec-2016	17:11:22.507	Surface Ops	FLBB Message	0.130000																																		
$	02-Dec-2016	17:11:22.610	Surface Ops	FLBB Message	0.130000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:11:22.713	Surface Ops	FLBB Message	0.130000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:11:22.816	Surface Ops	FLBB Message	0.130000																										Ave 3								
$	02-Dec-2016	17:11:23.016	Surface Ops	FLBB Message	0.130000																										Pkt 2								
$	02-Dec-2016	17:14:48.710	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,109,539								
$	02-Dec-2016	17:14:48.340	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.738,AirSaturation[%],100.603,Temperature[Deg.C],17.143,CalPhase[Deg],29.653,TCPhase[Deg],30.474,C1RPh[Deg],38.478,C2RPh[Deg],8.004,C1Amp[mV],641.5,C2Amp[mV],818.6,RawTemp[mV],335.1							
$	02-Dec-2016	17:14:48.343	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	17:14:49.041	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,129,539								
$	02-Dec-2016	17:14:49.050	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	17:14:49.152	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:14:49.353	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:14:49.456	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	17:14:49.559	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	17:14:50.316	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:15:01.906	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:15:49.585	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,120,539								
$	02-Dec-2016	17:15:48.317	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:15:48.426	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],359.841,AirSaturation[%],122.598,Temperature[Deg.C],18.350,CalPhase[Deg],27.310,TCPhase[Deg],28.131,C1RPh[Deg],36.127,C2RPh[Deg],7.997,C1Amp[mV],538.0,C2Amp[mV],789.1,RawTemp[mV],296.6							
$	02-Dec-2016	17:15:48.429	Anchor	Optode Message	9.520000																											#							
$	02-Dec-2016	17:15:50.028	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,52,700,113,539								
$	02-Dec-2016	17:15:50.131	Anchor	FLBB Message	9.520000																																		
$	02-Dec-2016	17:15:50.236	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:15:50.338	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:15:50.538	Anchor	FLBB Message	9.520000																										Ave 3								
$	02-Dec-2016	17:15:50.545	Anchor	FLBB Message	9.520000																										Pkt 2								
$	02-Dec-2016	17:15:59.907	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:16:50.535	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,53,700,115,539								
$	02-Dec-2016	17:16:48.339	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.935,AirSaturation[%],124.013,Temperature[Deg.C],18.357,CalPhase[Deg],27.190,TCPhase[Deg],28.011,C1RPh[Deg],36.008,C2RPh[Deg],7.997,C1Amp[mV],534.3,C2Amp[mV],788.3,RawTemp[mV],296.4							
$	02-Dec-2016	17:16:48.342	Anchor	Optode Message	9.520000																											#							
$	02-Dec-2016	17:16:50.769	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,112,539								
$	02-Dec-2016	17:16:50.873	Anchor	FLBB Message	9.520000																																		
$	02-Dec-2016	17:16:51.071	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:16:51.079	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:16:51.184	Anchor	FLBB Message	9.520000																										Ave 3								
$	02-Dec-2016	17:16:51.286	Anchor	FLBB Message	9.520000																										Pkt 2								
$	02-Dec-2016	17:16:49.316	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:17:00.906	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:17:48.614	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,54,700,112,538								
$	02-Dec-2016	17:17:48.407	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.172,AirSaturation[%],125.139,Temperature[Deg.C],18.367,CalPhase[Deg],27.096,TCPhase[Deg],27.917,C1RPh[Deg],35.917,C2RPh[Deg],8.000,C1Amp[mV],531.6,C2Amp[mV],787.9,RawTemp[mV],296.1							
$	02-Dec-2016	17:17:48.409	Anchor	Optode Message	9.530000																											#							
$	02-Dec-2016	17:17:48.948	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,52,700,111,539								
$	02-Dec-2016	17:17:48.956	Anchor	FLBB Message	9.530000																																		
$	02-Dec-2016	17:17:49.059	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:17:49.258	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:17:49.266	Anchor	FLBB Message	9.530000																										Ave 3								
$	02-Dec-2016	17:17:49.371	Anchor	FLBB Message	9.530000																										Pkt 2								
$	02-Dec-2016	17:17:50.316	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:18:01.905	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:18:48.317	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:18:49.705	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,112,539								
$	02-Dec-2016	17:18:48.434	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],368.585,AirSaturation[%],125.617,Temperature[Deg.C],18.365,CalPhase[Deg],27.057,TCPhase[Deg],27.878,C1RPh[Deg],35.878,C2RPh[Deg],8.000,C1Amp[mV],530.4,C2Amp[mV],787.7,RawTemp[mV],296.1							
$	02-Dec-2016	17:18:48.437	Anchor	Optode Message	9.520000																											#							
$	02-Dec-2016	17:18:50.036	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,112,539								
$	02-Dec-2016	17:18:50.044	Anchor	FLBB Message	9.520000																																		
$	02-Dec-2016	17:18:50.147	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:18:50.249	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:18:50.450	Anchor	FLBB Message	9.520000																										Ave 3								
$	02-Dec-2016	17:18:50.458	Anchor	FLBB Message	9.520000																										Pkt 2								
$	02-Dec-2016	17:18:59.905	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:19:50.449	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,114,539								
$	02-Dec-2016	17:19:48.343	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],369.735,AirSaturation[%],126.018,Temperature[Deg.C],18.369,CalPhase[Deg],27.024,TCPhase[Deg],27.845,C1RPh[Deg],35.844,C2RPh[Deg],7.999,C1Amp[mV],529.4,C2Amp[mV],787.5,RawTemp[mV],296.0							
$	02-Dec-2016	17:19:48.346	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	17:19:50.683	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,115,539								
$	02-Dec-2016	17:19:50.786	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	17:19:50.987	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:19:50.995	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:19:51.193	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	17:19:51.296	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	17:19:49.316	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	17:19:51.718	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,117,539								
$	02-Dec-2016	17:19:49.482	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.589,AirSaturation[%],126.314,Temperature[Deg.C],18.371,CalPhase[Deg],27.000,TCPhase[Deg],27.821,C1RPh[Deg],35.821,C2RPh[Deg],8.000,C1Amp[mV],528.8,C2Amp[mV],787.6,RawTemp[mV],296.0							
$	02-Dec-2016	17:19:49.485	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	17:19:52.143	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,116,539								
$	02-Dec-2016	17:19:52.151	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	17:19:52.254	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:19:52.359	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:19:52.558	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	17:19:52.565	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	17:20:00.905	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:20:05.543	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:20:11.632	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	02-Dec-2016	17:20:11.633	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	02-Dec-2016	17:20:20.601	DeAnchor	Unparsed SBE41 Message						S>    9.43, 18.3405, 35.2110  																													
$	02-Dec-2016	17:22:39.531	Ascend	FLBB Message	5.240000																										99/99/99,99:99:99,695,50,700,112,539								
$	02-Dec-2016	17:22:39.639	Ascend	FLBB Message	5.240000																										99/99/99,99:99:99,695,50,700,106,539								
$	02-Dec-2016	17:22:39.647	Ascend	FLBB Message	5.240000																																		
$	02-Dec-2016	17:22:39.750	Ascend	FLBB Message	5.240000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:22:39.853	Ascend	FLBB Message	5.240000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:22:39.957	Ascend	FLBB Message	5.240000																										Ave 3								
$	02-Dec-2016	17:22:40.156	Ascend	FLBB Message	5.240000																										Pkt 2								
$	02-Dec-2016	17:22:39.220	Ascend	Optode Message	5.240000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.877,AirSaturation[%],126.413,Temperature[Deg.C],18.371,CalPhase[Deg],26.992,TCPhase[Deg],27.813,C1RPh[Deg],35.746,C2RPh[Deg],7.933,C1Amp[mV],528.8,C2Amp[mV],784.2,RawTemp[mV],296.0							
$	02-Dec-2016	17:22:39.220	Ascend	Optode Message	5.240000																											#							
$	02-Dec-2016	17:22:52.153	Ascend	FLBB Message	3.150000																										99/99/99,99:99:99,695,51,700,116,539								
$	02-Dec-2016	17:22:51.834	Ascend	Optode Message	3.150000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.386,AirSaturation[%],127.228,Temperature[Deg.C],18.355,CalPhase[Deg],26.929,TCPhase[Deg],27.750,C1RPh[Deg],35.688,C2RPh[Deg],7.938,C1Amp[mV],534.7,C2Amp[mV],794.2,RawTemp[mV],296.5							
$	02-Dec-2016	17:22:51.837	Ascend	Optode Message	3.150000																											#							
$	02-Dec-2016	17:22:52.579	Ascend	FLBB Message	3.150000																										99/99/99,99:99:99,695,49,700,117,539								
$	02-Dec-2016	17:22:52.589	Ascend	FLBB Message	3.150000																																		
$	02-Dec-2016	17:22:52.692	Ascend	FLBB Message	3.150000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:22:52.794	Ascend	FLBB Message	3.150000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:22:52.897	Ascend	FLBB Message	3.150000																										Ave 3								
$	02-Dec-2016	17:22:53.097	Ascend	FLBB Message	3.150000																										Pkt 2								
$	02-Dec-2016	17:23:07.945	Ascend	FLBB Message	1.320000																										99/99/99,99:99:99,695,50,700,116,539								
$	02-Dec-2016	17:23:08.144	Ascend	FLBB Message	1.320000																										99/99/99,99:99:99,695,50,700,111,539								
$	02-Dec-2016	17:23:08.152	Ascend	FLBB Message	1.320000																																		
$	02-Dec-2016	17:23:08.254	Ascend	FLBB Message	1.320000																										Ser FLBB2K-3310								
$	02-Dec-2016	17:23:08.455	Ascend	FLBB Message	1.320000																										Ver FLNTU 4.08								
$	02-Dec-2016	17:23:08.463	Ascend	FLBB Message	1.320000																										Ave 3								
$	02-Dec-2016	17:23:08.662	Ascend	FLBB Message	1.320000																										Pkt 2								
$	02-Dec-2016	17:23:07.931	Ascend	Optode Message	1.320000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.939,AirSaturation[%],127.417,Temperature[Deg.C],18.356,CalPhase[Deg],26.914,TCPhase[Deg],27.735,C1RPh[Deg],35.737,C2RPh[Deg],8.002,C1Amp[mV],540.7,C2Amp[mV],798.9,RawTemp[mV],296.4							
$	02-Dec-2016	17:23:07.931	Ascend	Optode Message	1.320000																											#							
$	02-Dec-2016	17:23:09.988	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	02-Dec-2016	17:23:12.940	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	17:23:14.034	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	02-Dec-2016	17:23:14.035	Init Profile	Unparsed SBE41 Message						powering down  																													
$	02-Dec-2016	17:23:38.951	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	02-Dec-2016	17:24:40.948	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	17:24:47.050	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	02-Dec-2016	17:24:47.220	Dump CP Data	Unparsed SBE41 Message						samples = 174, maxPress = 9.60  																													
$	02-Dec-2016	17:24:55.170	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	02-Dec-2016	17:24:55.851	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	02-Dec-2016	17:24:56.153	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	02-Dec-2016	17:24:56.236	Dump CP Data	Unparsed SBE41 Message						  																													
$	02-Dec-2016	17:24:56.254	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	02-Dec-2016	17:25:02.233	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	02-Dec-2016	17:25:02.234	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.334	Dump CP Data	Unparsed SBE41 Message						    1.95,  18.3397,  35.2111, 18  																													
$	02-Dec-2016	17:25:02.334	Dump CP Data	Unparsed SBE41 Message						    3.99,  18.3389,  35.2111, 14  																													
$	02-Dec-2016	17:25:02.378	Dump CP Data	Unparsed SBE41 Message						    6.06,  18.3424,  35.2108, 12  																													
$	02-Dec-2016	17:25:02.427	Dump CP Data	Unparsed SBE41 Message						    8.14,  18.3420,  35.2112, 19  																													
$	02-Dec-2016	17:25:02.487	Dump CP Data	Unparsed SBE41 Message						    9.48,  18.3399,  35.2113, 111  																													
$	02-Dec-2016	17:25:02.532	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.641	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.642	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.745	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.850	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	17:25:02.851	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	17:25:08.343	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	02-Dec-2016	17:25:08.343	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.443	Dump CP Data	Unparsed SBE41 Message						001347A4898B12  																													
$	02-Dec-2016	17:25:08.443	Dump CP Data	Unparsed SBE41 Message						002847A3898B0E  																													
$	02-Dec-2016	17:25:08.443	Dump CP Data	Unparsed SBE41 Message						003D47A6898B0C  																													
$	02-Dec-2016	17:25:08.444	Dump CP Data	Unparsed SBE41 Message						005147A6898B13  																													
$	02-Dec-2016	17:25:08.459	Dump CP Data	Unparsed SBE41 Message						005F47A4898B6F  																													
$	02-Dec-2016	17:25:08.485	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.510	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.536	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.561	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.595	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	17:25:08.615	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	17:25:14.261	Dump CP Data	Unparsed SBE41 Message						S>  																													
