$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	02-Dec-2016	20:35:39.953	Surface Ops	Optode Message	0.130000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],374.281,AirSaturation[%],127.506,Temperature[Deg.C],18.345,CalPhase[Deg],26.908,TCPhase[Deg],27.729,C1RPh[Deg],35.730,C2RPh[Deg],8.001,C1Amp[mV],547.0,C2Amp[mV],803.9,RawTemp[mV],296.8							
$	02-Dec-2016	20:35:39.953	Surface Ops	Optode Message	0.130000																											#							
$	02-Dec-2016	20:37:04.974	Surface Ops	FLBB Message	0.160000																										99/99/99,99:99:99,695,48,700,111,539								
$	02-Dec-2016	20:37:05.175	Surface Ops	FLBB Message	0.160000																										99/99/99,99:99:99,695,52,700,112,539								
$	02-Dec-2016	20:37:05.183	Surface Ops	FLBB Message	0.160000																																		
$	02-Dec-2016	20:37:05.384	Surface Ops	FLBB Message	0.160000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:37:05.490	Surface Ops	FLBB Message	0.160000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:37:05.595	Surface Ops	FLBB Message	0.160000																										Ave 3								
$	02-Dec-2016	20:37:05.698	Surface Ops	FLBB Message	0.160000																										Pkt 2								
$	02-Dec-2016	20:37:05.594	Surface Ops	Optode Message	0.160000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.403,AirSaturation[%],101.474,Temperature[Deg.C],17.611,CalPhase[Deg],29.470,TCPhase[Deg],30.291,C1RPh[Deg],38.292,C2RPh[Deg],8.001,C1Amp[mV],634.2,C2Amp[mV],812.8,RawTemp[mV],320.2							
$	02-Dec-2016	20:37:05.698	Surface Ops	Optode Message	0.160000																											#							
$	02-Dec-2016	20:37:05.934	Surface Ops	FLBB Message	0.160000																										99/99/99,99:99:99,695,47,700,110,539								
$	02-Dec-2016	20:37:06.339	Surface Ops	FLBB Message	0.160000																										99/99/99,99:99:99,695,54,700,111,539								
$	02-Dec-2016	20:37:06.444	Surface Ops	FLBB Message	0.160000																																		
$	02-Dec-2016	20:37:06.547	Surface Ops	FLBB Message	0.160000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:37:06.650	Surface Ops	FLBB Message	0.160000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:37:06.753	Surface Ops	FLBB Message	0.160000																										Ave 3								
$	02-Dec-2016	20:37:06.857	Surface Ops	FLBB Message	0.160000																										Pkt 2								
$	02-Dec-2016	20:40:26.309	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,50,700,113,539								
$	02-Dec-2016	20:40:26.037	Anchor	Optode Message	9.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.199,AirSaturation[%],100.374,Temperature[Deg.C],17.278,CalPhase[Deg],29.653,TCPhase[Deg],30.474,C1RPh[Deg],38.476,C2RPh[Deg],8.002,C1Amp[mV],641.2,C2Amp[mV],817.4,RawTemp[mV],330.8							
$	02-Dec-2016	20:40:26.040	Anchor	Optode Message	9.490000																											#							
$	02-Dec-2016	20:40:26.643	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,51,700,116,539								
$	02-Dec-2016	20:40:26.747	Anchor	FLBB Message	9.490000																																		
$	02-Dec-2016	20:40:26.850	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:40:27.049	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:40:27.058	Anchor	FLBB Message	9.490000																										Ave 3								
$	02-Dec-2016	20:40:27.161	Anchor	FLBB Message	9.490000																										Pkt 2								
$	02-Dec-2016	20:40:28.315	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:40:39.906	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:41:27.096	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,113,539								
$	02-Dec-2016	20:41:26.139	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],358.736,AirSaturation[%],122.195,Temperature[Deg.C],18.339,CalPhase[Deg],27.345,TCPhase[Deg],28.166,C1RPh[Deg],36.160,C2RPh[Deg],7.994,C1Amp[mV],539.1,C2Amp[mV],789.3,RawTemp[mV],297.0							
$	02-Dec-2016	20:41:26.142	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	20:41:27.428	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,112,539								
$	02-Dec-2016	20:41:27.436	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	20:41:27.538	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:41:27.641	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:41:27.744	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	20:41:27.945	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	20:41:26.815	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:41:38.406	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:42:28.133	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,116,539								
$	02-Dec-2016	20:42:26.136	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.736,AirSaturation[%],124.597,Temperature[Deg.C],18.346,CalPhase[Deg],27.144,TCPhase[Deg],27.965,C1RPh[Deg],35.900,C2RPh[Deg],7.935,C1Amp[mV],533.0,C2Amp[mV],788.5,RawTemp[mV],296.8							
$	02-Dec-2016	20:42:26.139	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	20:42:28.463	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,123,539								
$	02-Dec-2016	20:42:28.566	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	20:42:28.765	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:42:28.773	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:42:28.876	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	20:42:28.980	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	20:42:27.815	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:42:39.406	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:43:25.816	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:43:26.405	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,50,700,113,539								
$	02-Dec-2016	20:43:26.036	Anchor	Optode Message	9.490000																											MEASUREMENT,4330,2592,O2Concentration[uM],368.195,AirSaturation[%],125.449,Temperature[Deg.C],18.351,CalPhase[Deg],27.073,TCPhase[Deg],27.894,C1RPh[Deg],35.891,C2RPh[Deg],7.997,C1Amp[mV],530.9,C2Amp[mV],788.2,RawTemp[mV],296.6							
$	02-Dec-2016	20:43:26.039	Anchor	Optode Message	9.490000																											#							
$	02-Dec-2016	20:43:26.734	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,53,700,116,539								
$	02-Dec-2016	20:43:26.935	Anchor	FLBB Message	9.490000																																		
$	02-Dec-2016	20:43:26.943	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:43:27.142	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:43:27.149	Anchor	FLBB Message	9.490000																										Ave 3								
$	02-Dec-2016	20:43:27.252	Anchor	FLBB Message	9.490000																										Pkt 2								
$	02-Dec-2016	20:43:37.406	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:44:26.995	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,49,700,117,539								
$	02-Dec-2016	20:44:26.038	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],369.506,AirSaturation[%],125.901,Temperature[Deg.C],18.354,CalPhase[Deg],27.036,TCPhase[Deg],27.857,C1RPh[Deg],35.854,C2RPh[Deg],7.997,C1Amp[mV],529.8,C2Amp[mV],788.0,RawTemp[mV],296.5							
$	02-Dec-2016	20:44:26.040	Anchor	Optode Message	9.510000																											#							
$	02-Dec-2016	20:44:27.327	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,117,539								
$	02-Dec-2016	20:44:27.335	Anchor	FLBB Message	9.510000																																		
$	02-Dec-2016	20:44:27.440	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:44:27.543	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:44:27.645	Anchor	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	20:44:27.844	Anchor	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	20:44:26.816	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:44:38.406	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:45:28.460	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,109,539								
$	02-Dec-2016	20:45:26.039	Start CP Mode	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.642,AirSaturation[%],126.288,Temperature[Deg.C],18.354,CalPhase[Deg],27.005,TCPhase[Deg],27.826,C1RPh[Deg],35.823,C2RPh[Deg],7.998,C1Amp[mV],529.0,C2Amp[mV],787.9,RawTemp[mV],296.5							
$	02-Dec-2016	20:45:26.042	Start CP Mode	Optode Message	9.500000																											#							
$	02-Dec-2016	20:45:28.695	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,113,539								
$	02-Dec-2016	20:45:28.798	Start CP Mode	FLBB Message	9.500000																																		
$	02-Dec-2016	20:45:28.997	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:45:29.007	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:45:29.205	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	20:45:29.307	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	20:45:29.809	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,113,539								
$	02-Dec-2016	20:45:27.378	Start CP Mode	Optode Message	9.500000																											MEASUREMENT,4330,2592,O2Concentration[uM],371.311,AirSaturation[%],126.527,Temperature[Deg.C],18.358,CalPhase[Deg],26.985,TCPhase[Deg],27.806,C1RPh[Deg],35.802,C2RPh[Deg],7.996,C1Amp[mV],528.3,C2Amp[mV],787.8,RawTemp[mV],296.4							
$	02-Dec-2016	20:45:27.381	Start CP Mode	Optode Message	9.500000																											#							
$	02-Dec-2016	20:45:30.142	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,51,700,115,539								
$	02-Dec-2016	20:45:30.150	Start CP Mode	FLBB Message	9.500000																																		
$	02-Dec-2016	20:45:30.253	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:45:30.356	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:45:30.557	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	20:45:30.564	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	20:45:28.460	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	20:46:19.841	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:46:25.932	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	02-Dec-2016	20:46:25.932	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	02-Dec-2016	20:46:34.813	DeAnchor	Unparsed SBE41 Message						S>    9.51, 18.3303, 35.2117  																													
$	02-Dec-2016	20:48:42.947	Ascend	FLBB Message	5.190000																										99/99/99,99:99:99,695,51,700,114,539								
$	02-Dec-2016	20:48:41.734	Ascend	Optode Message	5.190000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],371.344,AirSaturation[%],126.539,Temperature[Deg.C],18.358,CalPhase[Deg],26.984,TCPhase[Deg],27.805,C1RPh[Deg],35.796,C2RPh[Deg],7.991,C1Amp[mV],528.3,C2Amp[mV],784.8,RawTemp[mV],296.4							
$	02-Dec-2016	20:48:41.734	Ascend	Optode Message	5.190000																											#							
$	02-Dec-2016	20:48:43.182	Ascend	FLBB Message	5.190000																										99/99/99,99:99:99,695,55,700,119,539								
$	02-Dec-2016	20:48:43.285	Ascend	FLBB Message	5.190000																																		
$	02-Dec-2016	20:48:43.485	Ascend	FLBB Message	5.190000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:48:43.492	Ascend	FLBB Message	5.190000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:48:43.595	Ascend	FLBB Message	5.190000																										Ave 3								
$	02-Dec-2016	20:48:43.706	Ascend	FLBB Message	5.190000																										Pkt 2								
$	02-Dec-2016	20:48:54.370	Ascend	FLBB Message	3.380000																										99/99/99,99:99:99,695,48,700,113,539								
$	02-Dec-2016	20:48:54.477	Ascend	FLBB Message	3.380000																										99/99/99,99:99:99,695,51,700,112,539								
$	02-Dec-2016	20:48:54.486	Ascend	FLBB Message	3.380000																																		
$	02-Dec-2016	20:48:54.588	Ascend	FLBB Message	3.380000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:48:54.693	Ascend	FLBB Message	3.380000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:48:54.796	Ascend	FLBB Message	3.380000																										Ave 3								
$	02-Dec-2016	20:48:54.995	Ascend	FLBB Message	3.380000																										Pkt 2								
$	02-Dec-2016	20:48:54.059	Ascend	Optode Message	3.380000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.803,AirSaturation[%],127.335,Temperature[Deg.C],18.342,CalPhase[Deg],26.923,TCPhase[Deg],27.744,C1RPh[Deg],35.739,C2RPh[Deg],7.996,C1Amp[mV],534.5,C2Amp[mV],794.5,RawTemp[mV],296.9							
$	02-Dec-2016	20:48:54.059	Ascend	Optode Message	3.380000																											#							
$	02-Dec-2016	20:49:13.157	Ascend	FLBB Message	1.520000																										99/99/99,99:99:99,695,54,700,112,539								
$	02-Dec-2016	20:49:12.946	Ascend	Optode Message	1.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.877,AirSaturation[%],127.361,Temperature[Deg.C],18.342,CalPhase[Deg],26.920,TCPhase[Deg],27.741,C1RPh[Deg],35.740,C2RPh[Deg],7.999,C1Amp[mV],539.6,C2Amp[mV],798.3,RawTemp[mV],296.9							
$	02-Dec-2016	20:49:12.949	Ascend	Optode Message	1.520000																											#							
$	02-Dec-2016	20:49:13.584	Ascend	FLBB Message	1.520000																										99/99/99,99:99:99,695,56,700,112,539								
$	02-Dec-2016	20:49:13.592	Ascend	FLBB Message	1.520000																																		
$	02-Dec-2016	20:49:13.695	Ascend	FLBB Message	1.520000																										Ser FLBB2K-3310								
$	02-Dec-2016	20:49:13.800	Ascend	FLBB Message	1.520000																										Ver FLNTU 4.08								
$	02-Dec-2016	20:49:14.000	Ascend	FLBB Message	1.520000																										Ave 3								
$	02-Dec-2016	20:49:14.007	Ascend	FLBB Message	1.520000																										Pkt 2								
$	02-Dec-2016	20:49:15.348	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	02-Dec-2016	20:49:18.087	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	20:49:19.181	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	02-Dec-2016	20:49:19.181	Init Profile	Unparsed SBE41 Message						powering down  																													
$	02-Dec-2016	20:49:44.533	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	02-Dec-2016	20:50:44.218	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	20:50:50.295	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	02-Dec-2016	20:50:50.481	Dump CP Data	Unparsed SBE41 Message						samples = 165, maxPress = 9.57  																													
$	02-Dec-2016	20:50:58.416	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	02-Dec-2016	20:50:59.176	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	02-Dec-2016	20:50:59.390	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	02-Dec-2016	20:50:59.599	Dump CP Data	Unparsed SBE41 Message						  																													
$	02-Dec-2016	20:50:59.599	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	02-Dec-2016	20:51:05.602	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	02-Dec-2016	20:51:05.602	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:05.637	Dump CP Data	Unparsed SBE41 Message						    2.04,  18.3265,  35.2118, 19  																													
$	02-Dec-2016	20:51:05.687	Dump CP Data	Unparsed SBE41 Message						    3.91,  18.3280,  35.2116, 14  																													
$	02-Dec-2016	20:51:05.736	Dump CP Data	Unparsed SBE41 Message						    5.93,  18.3299,  35.2115, 14  																													
$	02-Dec-2016	20:51:05.786	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.3312,  35.2116, 19  																													
$	02-Dec-2016	20:51:05.908	Dump CP Data	Unparsed SBE41 Message						    9.46,  18.3300,  35.2117, 99  																													
$	02-Dec-2016	20:51:06.012	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:06.115	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:06.116	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:06.117	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:06.117	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	20:51:06.118	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	20:51:11.703	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	02-Dec-2016	20:51:11.704	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:11.704	Dump CP Data	Unparsed SBE41 Message						00144796898C13  																													
$	02-Dec-2016	20:51:11.726	Dump CP Data	Unparsed SBE41 Message						00274798898C0E  																													
$	02-Dec-2016	20:51:11.752	Dump CP Data	Unparsed SBE41 Message						003B479A898C0E  																													
$	02-Dec-2016	20:51:11.777	Dump CP Data	Unparsed SBE41 Message						0051479B898C13  																													
$	02-Dec-2016	20:51:11.912	Dump CP Data	Unparsed SBE41 Message						005F479A898C63  																													
$	02-Dec-2016	20:51:11.912	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:11.913	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:11.913	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:11.914	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:12.020	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	20:51:12.020	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	20:51:17.518	Dump CP Data	Unparsed SBE41 Message						S>  																													
