$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	02-Dec-2016	08:15:27.820	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:15:31.824	Surface Ops	Optode Message	1.130000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],374.601,AirSaturation[%],127.646,Temperature[Deg.C],18.357,CalPhase[Deg],26.895,TCPhase[Deg],27.716,C1RPh[Deg],35.730,C2RPh[Deg],8.013,C1Amp[mV],546.9,C2Amp[mV],802.1,RawTemp[mV],296.4							
$	02-Dec-2016	08:15:31.825	Surface Ops	Optode Message	1.130000																											#							
$	02-Dec-2016	08:15:39.389	Surface Ops	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:16:56.840	Surface Ops	FLBB Message	-0.850000																										99/99/99,99:99:99,695,54,700,279,539								
$	02-Dec-2016	08:16:57.043	Surface Ops	FLBB Message	-0.850000																										99/99/99,99:99:99,695,53,700,258,539								
$	02-Dec-2016	08:16:57.146	Surface Ops	FLBB Message	-0.850000																																		
$	02-Dec-2016	08:16:57.154	Surface Ops	FLBB Message	-0.850000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:16:57.355	Surface Ops	FLBB Message	-0.850000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:16:57.461	Surface Ops	FLBB Message	-0.850000																										Ave 3								
$	02-Dec-2016	08:16:57.469	Surface Ops	FLBB Message	-0.850000																										Pkt 2								
$	02-Dec-2016	08:16:57.564	Surface Ops	Optode Message	-0.850000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.538,AirSaturation[%],102.241,Temperature[Deg.C],17.954,CalPhase[Deg],29.324,TCPhase[Deg],30.145,C1RPh[Deg],38.160,C2RPh[Deg],8.015,C1Amp[mV],628.9,C2Amp[mV],809.9,RawTemp[mV],309.3							
$	02-Dec-2016	08:16:57.565	Surface Ops	Optode Message	-0.850000																											#							
$	02-Dec-2016	08:16:57.800	Surface Ops	FLBB Message	-0.850000																										99/99/99,99:99:99,695,54,700,439,539								
$	02-Dec-2016	08:16:57.906	Surface Ops	FLBB Message	-0.850000																										99/99/99,99:99:99,695,50,700,380,539								
$	02-Dec-2016	08:16:58.011	Surface Ops	FLBB Message	-0.850000																																		
$	02-Dec-2016	08:16:58.114	Surface Ops	FLBB Message	-0.850000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:16:58.313	Surface Ops	FLBB Message	-0.850000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:16:58.321	Surface Ops	FLBB Message	-0.850000																										Ave 3								
$	02-Dec-2016	08:16:58.427	Surface Ops	FLBB Message	-0.850000																										Pkt 2								
$	02-Dec-2016	08:20:24.566	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,145,539								
$	02-Dec-2016	08:20:24.234	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.324,AirSaturation[%],100.863,Temperature[Deg.C],17.652,CalPhase[Deg],29.530,TCPhase[Deg],30.351,C1RPh[Deg],38.308,C2RPh[Deg],7.957,C1Amp[mV],636.5,C2Amp[mV],813.4,RawTemp[mV],318.9							
$	02-Dec-2016	08:20:24.234	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	08:20:24.993	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,144,539								
$	02-Dec-2016	08:20:25.097	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	08:20:25.104	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:20:25.207	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:20:25.312	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	08:20:25.415	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	08:20:25.799	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:20:37.389	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:21:25.152	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,139,539								
$	02-Dec-2016	08:21:24.195	Anchor	Optode Message	9.500000																											MEASUREMENT,4330,2592,O2Concentration[uM],358.728,AirSaturation[%],122.233,Temperature[Deg.C],18.355,CalPhase[Deg],27.340,TCPhase[Deg],28.161,C1RPh[Deg],36.170,C2RPh[Deg],8.010,C1Amp[mV],538.1,C2Amp[mV],787.0,RawTemp[mV],296.5							
$	02-Dec-2016	08:21:24.198	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	08:21:25.501	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,51,700,139,539								
$	02-Dec-2016	08:21:25.604	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	08:21:25.707	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:21:25.906	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:21:25.916	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	08:21:26.018	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	08:21:26.798	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:21:38.388	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:22:26.281	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,49,700,146,539								
$	02-Dec-2016	08:22:24.195	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.918,AirSaturation[%],124.353,Temperature[Deg.C],18.360,CalPhase[Deg],27.162,TCPhase[Deg],27.983,C1RPh[Deg],35.994,C2RPh[Deg],8.011,C1Amp[mV],532.9,C2Amp[mV],786.4,RawTemp[mV],296.3							
$	02-Dec-2016	08:22:24.198	Anchor	Optode Message	9.500000																											#							
$	02-Dec-2016	08:22:26.611	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,53,700,142,539								
$	02-Dec-2016	08:22:26.619	Anchor	FLBB Message	9.500000																																		
$	02-Dec-2016	08:22:26.722	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:22:26.824	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:22:26.927	Anchor	FLBB Message	9.500000																										Ave 3								
$	02-Dec-2016	08:22:27.128	Anchor	FLBB Message	9.500000																										Pkt 2								
$	02-Dec-2016	08:22:24.804	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:22:36.389	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:23:24.755	Anchor	FLBB Message	9.480000																										99/99/99,99:99:99,695,50,700,137,539								
$	02-Dec-2016	08:23:24.330	Anchor	Optode Message	9.480000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.402,AirSaturation[%],125.209,Temperature[Deg.C],18.363,CalPhase[Deg],27.091,TCPhase[Deg],27.912,C1RPh[Deg],35.923,C2RPh[Deg],8.012,C1Amp[mV],530.6,C2Amp[mV],786.2,RawTemp[mV],296.2							
$	02-Dec-2016	08:23:24.330	Anchor	Optode Message	9.480000																											#							
$	02-Dec-2016	08:23:25.085	Anchor	FLBB Message	9.480000																										99/99/99,99:99:99,695,51,700,136,539								
$	02-Dec-2016	08:23:25.195	Anchor	FLBB Message	9.480000																																		
$	02-Dec-2016	08:23:25.202	Anchor	FLBB Message	9.480000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:23:25.305	Anchor	FLBB Message	9.480000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:23:25.408	Anchor	FLBB Message	9.480000																										Ave 3								
$	02-Dec-2016	08:23:25.511	Anchor	FLBB Message	9.480000																										Pkt 2								
$	02-Dec-2016	08:23:25.797	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:23:37.388	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:24:25.233	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,144,539								
$	02-Dec-2016	08:24:24.276	Anchor	Optode Message	9.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],369.332,AirSaturation[%],125.872,Temperature[Deg.C],18.365,CalPhase[Deg],27.036,TCPhase[Deg],27.857,C1RPh[Deg],35.869,C2RPh[Deg],8.012,C1Amp[mV],529.1,C2Amp[mV],786.0,RawTemp[mV],296.1							
$	02-Dec-2016	08:24:24.279	Anchor	Optode Message	9.490000																											#							
$	02-Dec-2016	08:24:25.579	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,55,700,152,539								
$	02-Dec-2016	08:24:25.684	Anchor	FLBB Message	9.490000																																		
$	02-Dec-2016	08:24:25.787	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:24:25.986	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:24:25.994	Anchor	FLBB Message	9.490000																										Ave 3								
$	02-Dec-2016	08:24:26.097	Anchor	FLBB Message	9.490000																										Pkt 2								
$	02-Dec-2016	08:24:26.799	Anchor	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:24:38.389	Anchor	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:25:26.505	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,142,539								
$	02-Dec-2016	08:25:26.514	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,57,700,144,539								
$	02-Dec-2016	08:25:26.617	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	08:25:26.722	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:25:26.825	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:25:27.024	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	08:25:27.031	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	08:25:24.273	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.163,AirSaturation[%],126.161,Temperature[Deg.C],18.368,CalPhase[Deg],27.013,TCPhase[Deg],27.834,C1RPh[Deg],35.844,C2RPh[Deg],8.011,C1Amp[mV],528.2,C2Amp[mV],786.0,RawTemp[mV],296.1							
$	02-Dec-2016	08:25:24.273	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	08:25:24.798	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	02-Dec-2016	08:25:27.872	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,150,539								
$	02-Dec-2016	08:25:25.637	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],371.205,AirSaturation[%],126.515,Temperature[Deg.C],18.367,CalPhase[Deg],26.984,TCPhase[Deg],27.805,C1RPh[Deg],35.757,C2RPh[Deg],7.951,C1Amp[mV],527.5,C2Amp[mV],785.9,RawTemp[mV],296.1							
$	02-Dec-2016	08:25:25.640	Start CP Mode	Optode Message	9.510000																											#							
$	02-Dec-2016	08:25:28.203	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,48,700,137,539								
$	02-Dec-2016	08:25:28.307	Start CP Mode	FLBB Message	9.510000																																		
$	02-Dec-2016	08:25:28.314	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:25:28.417	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:25:28.523	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	02-Dec-2016	08:25:28.626	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	02-Dec-2016	08:25:36.388	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:25:41.632	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:25:47.720	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	02-Dec-2016	08:25:47.721	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	02-Dec-2016	08:25:56.154	DeAnchor	Unparsed SBE41 Message						S>    9.43, 18.3431, 35.2096  																													
$	02-Dec-2016	08:28:12.495	Ascend	FLBB Message	5.160000																										99/99/99,99:99:99,695,53,700,111,539								
$	02-Dec-2016	08:28:12.694	Ascend	FLBB Message	5.160000																										99/99/99,99:99:99,695,50,700,116,539								
$	02-Dec-2016	08:28:12.703	Ascend	FLBB Message	5.160000																																		
$	02-Dec-2016	08:28:12.805	Ascend	FLBB Message	5.160000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:28:13.006	Ascend	FLBB Message	5.160000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:28:13.015	Ascend	FLBB Message	5.160000																										Ave 3								
$	02-Dec-2016	08:28:13.118	Ascend	FLBB Message	5.160000																										Pkt 2								
$	02-Dec-2016	08:28:12.481	Ascend	Optode Message	5.160000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],371.225,AirSaturation[%],126.521,Temperature[Deg.C],18.367,CalPhase[Deg],26.984,TCPhase[Deg],27.805,C1RPh[Deg],35.810,C2RPh[Deg],8.005,C1Amp[mV],527.5,C2Amp[mV],783.2,RawTemp[mV],296.1							
$	02-Dec-2016	08:28:12.481	Ascend	Optode Message	5.160000																											#							
$	02-Dec-2016	08:28:25.777	Ascend	FLBB Message	3.270000																										99/99/99,99:99:99,695,50,700,117,539								
$	02-Dec-2016	08:28:25.223	Ascend	Optode Message	3.270000																											MEASUREMENT,4330,2592,O2Concentration[uM],373.517,AirSaturation[%],127.276,Temperature[Deg.C],18.357,CalPhase[Deg],26.925,TCPhase[Deg],27.746,C1RPh[Deg],35.755,C2RPh[Deg],8.009,C1Amp[mV],533.5,C2Amp[mV],792.2,RawTemp[mV],296.4							
$	02-Dec-2016	08:28:25.223	Ascend	Optode Message	3.270000																											#							
$	02-Dec-2016	08:28:26.107	Ascend	FLBB Message	3.270000																										99/99/99,99:99:99,695,53,700,119,539								
$	02-Dec-2016	08:28:26.212	Ascend	FLBB Message	3.270000																																		
$	02-Dec-2016	08:28:26.315	Ascend	FLBB Message	3.270000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:28:26.419	Ascend	FLBB Message	3.270000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:28:26.521	Ascend	FLBB Message	3.270000																										Ave 3								
$	02-Dec-2016	08:28:26.529	Ascend	FLBB Message	3.270000																										Pkt 2								
$	02-Dec-2016	08:28:41.329	Ascend	FLBB Message	1.710000																										99/99/99,99:99:99,695,52,700,115,539								
$	02-Dec-2016	08:28:41.672	Ascend	FLBB Message	1.540000																										99/99/99,99:99:99,695,52,700,119,539								
$	02-Dec-2016	08:28:41.871	Ascend	FLBB Message	1.540000																																		
$	02-Dec-2016	08:28:41.880	Ascend	FLBB Message	1.540000																										Ser FLBB2K-3310								
$	02-Dec-2016	08:28:41.983	Ascend	FLBB Message	1.540000																										Ver FLNTU 4.08								
$	02-Dec-2016	08:28:42.086	Ascend	FLBB Message	1.540000																										Ave 3								
$	02-Dec-2016	08:28:42.286	Ascend	FLBB Message	1.540000																										Pkt 2								
$	02-Dec-2016	08:28:41.317	Ascend	Optode Message	1.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],374.043,AirSaturation[%],127.452,Temperature[Deg.C],18.356,CalPhase[Deg],26.911,TCPhase[Deg],27.732,C1RPh[Deg],35.743,C2RPh[Deg],8.011,C1Amp[mV],539.5,C2Amp[mV],796.6,RawTemp[mV],296.5							
$	02-Dec-2016	08:28:41.317	Ascend	Optode Message	1.540000																											#							
$	02-Dec-2016	08:28:43.625	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	02-Dec-2016	08:28:43.625	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	08:28:46.193	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	02-Dec-2016	08:28:47.287	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	02-Dec-2016	08:28:47.287	Init Profile	Unparsed SBE41 Message						powering down  																													
$	02-Dec-2016	08:29:12.109	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	02-Dec-2016	08:30:12.097	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:30:18.174	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	02-Dec-2016	08:30:18.344	Dump CP Data	Unparsed SBE41 Message						samples = 172, maxPress = 9.56  																													
$	02-Dec-2016	08:30:26.294	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	02-Dec-2016	08:30:26.975	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	02-Dec-2016	08:30:27.276	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	02-Dec-2016	08:30:27.344	Dump CP Data	Unparsed SBE41 Message						  																													
$	02-Dec-2016	08:30:27.363	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	02-Dec-2016	08:30:33.356	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	02-Dec-2016	08:30:33.356	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.456	Dump CP Data	Unparsed SBE41 Message						    2.13,  18.3425,  35.2092, 17  																													
$	02-Dec-2016	08:30:33.457	Dump CP Data	Unparsed SBE41 Message						    4.00,  18.3417,  35.2094, 15  																													
$	02-Dec-2016	08:30:33.501	Dump CP Data	Unparsed SBE41 Message						    6.02,  18.3420,  35.2093, 12  																													
$	02-Dec-2016	08:30:33.550	Dump CP Data	Unparsed SBE41 Message						    8.04,  18.3424,  35.2093, 19  																													
$	02-Dec-2016	08:30:33.609	Dump CP Data	Unparsed SBE41 Message						    9.45,  18.3424,  35.2094, 109  																													
$	02-Dec-2016	08:30:33.654	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.763	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.764	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.868	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.869	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	02-Dec-2016	08:30:33.869	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	08:30:39.463	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	02-Dec-2016	08:30:39.464	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.563	Dump CP Data	Unparsed SBE41 Message						001547A6898911  																													
$	02-Dec-2016	08:30:39.564	Dump CP Data	Unparsed SBE41 Message						002847A689890F  																													
$	02-Dec-2016	08:30:39.564	Dump CP Data	Unparsed SBE41 Message						003C47A689890C  																													
$	02-Dec-2016	08:30:39.565	Dump CP Data	Unparsed SBE41 Message						005047A6898913  																													
$	02-Dec-2016	08:30:39.579	Dump CP Data	Unparsed SBE41 Message						005F47A689896D  																													
$	02-Dec-2016	08:30:39.605	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.630	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.655	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.681	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.716	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	02-Dec-2016	08:30:39.736	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	02-Dec-2016	08:30:45.376	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	02-Dec-2016	08:30:46.983	Dump CP Data	Unparsed SBE41 Message						S>pts  																													
