$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	04-Dec-2016	13:50:39.591	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:50:43.878	Surface Ops	Optode Message	1.040000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],371.872,AirSaturation[%],126.282,Temperature[Deg.C],18.190,CalPhase[Deg],27.032,TCPhase[Deg],27.853,C1RPh[Deg],35.828,C2RPh[Deg],7.974,C1Amp[mV],543.3,C2Amp[mV],800.8,RawTemp[mV],301.8							
$	04-Dec-2016	13:50:43.879	Surface Ops	Optode Message	1.040000																											#							
$	04-Dec-2016	13:50:51.158	Surface Ops	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	13:52:15.034	Surface Ops	FLBB Message	0.110000																										99/99/99,99:99:99,695,49,700,114,539								
$	04-Dec-2016	13:52:15.238	Surface Ops	FLBB Message	0.110000																										99/99/99,99:99:99,695,49,700,112,539								
$	04-Dec-2016	13:52:15.343	Surface Ops	FLBB Message	0.110000																																		
$	04-Dec-2016	13:52:15.446	Surface Ops	FLBB Message	0.110000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:52:15.549	Surface Ops	FLBB Message	0.110000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:52:15.652	Surface Ops	FLBB Message	0.110000																										Ave 3								
$	04-Dec-2016	13:52:15.761	Surface Ops	FLBB Message	0.110000																										Pkt 2								
$	04-Dec-2016	13:52:15.864	AB Retract Fast	Optode Message	0.110000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],304.767,AirSaturation[%],101.397,Temperature[Deg.C],17.200,CalPhase[Deg],29.554,TCPhase[Deg],30.375,C1RPh[Deg],38.350,C2RPh[Deg],7.974,C1Amp[mV],625.2,C2Amp[mV],808.8,RawTemp[mV],333.3							
$	04-Dec-2016	13:52:15.864	AB Retract Fast	Optode Message	0.110000																											#							
$	04-Dec-2016	13:52:19.254	AB Retract Fast	FLBB Message	0.110000																										99/99/99,99:99:99,695,47,700,116,539								
$	04-Dec-2016	13:52:19.357	AB Retract Fast	FLBB Message	0.110000																										99/99/99,99:99:99,695,51,700,115,539								
$	04-Dec-2016	13:52:19.556	AB Retract Fast	FLBB Message	0.110000																																		
$	04-Dec-2016	13:52:19.663	AB Retract Fast	FLBB Message	0.110000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:52:19.769	AB Retract Fast	FLBB Message	0.110000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:52:19.871	AB Retract Fast	FLBB Message	0.110000																										Ave 3								
$	04-Dec-2016	13:52:19.975	AB Retract Fast	FLBB Message	0.110000																										Pkt 2								
$	04-Dec-2016	13:55:42.605	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,53,700,110,539								
$	04-Dec-2016	13:55:42.534	Anchor	Optode Message	9.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],303.829,AirSaturation[%],100.123,Temperature[Deg.C],16.743,CalPhase[Deg],29.781,TCPhase[Deg],30.602,C1RPh[Deg],38.575,C2RPh[Deg],7.973,C1Amp[mV],633.6,C2Amp[mV],814.9,RawTemp[mV],347.8							
$	04-Dec-2016	13:55:42.537	Anchor	Optode Message	9.490000																											#							
$	04-Dec-2016	13:55:43.032	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,50,700,122,539								
$	04-Dec-2016	13:55:43.136	Anchor	FLBB Message	9.490000																																		
$	04-Dec-2016	13:55:43.143	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:55:43.247	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:55:43.358	Anchor	FLBB Message	9.490000																										Ave 3								
$	04-Dec-2016	13:55:43.460	Anchor	FLBB Message	9.490000																										Pkt 2								
$	04-Dec-2016	13:55:43.664	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:55:55.236	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	13:56:43.423	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,114,539								
$	04-Dec-2016	13:56:42.646	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],357.732,AirSaturation[%],121.469,Temperature[Deg.C],18.185,CalPhase[Deg],27.433,TCPhase[Deg],28.254,C1RPh[Deg],36.222,C2RPh[Deg],7.968,C1Amp[mV],536.0,C2Amp[mV],787.5,RawTemp[mV],301.9							
$	04-Dec-2016	13:56:42.649	Anchor	Optode Message	9.510000																											#							
$	04-Dec-2016	13:56:43.753	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,110,539								
$	04-Dec-2016	13:56:43.856	Anchor	FLBB Message	9.510000																																		
$	04-Dec-2016	13:56:43.959	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:56:44.161	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:56:44.169	Anchor	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	13:56:44.272	Anchor	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	13:56:44.567	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:56:56.158	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	13:57:44.420	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,49,700,111,539								
$	04-Dec-2016	13:57:42.534	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.275,AirSaturation[%],123.370,Temperature[Deg.C],18.192,CalPhase[Deg],27.271,TCPhase[Deg],28.092,C1RPh[Deg],36.062,C2RPh[Deg],7.970,C1Amp[mV],531.1,C2Amp[mV],786.6,RawTemp[mV],301.7							
$	04-Dec-2016	13:57:42.538	Anchor	Optode Message	9.510000																											#							
$	04-Dec-2016	13:57:44.750	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,114,539								
$	04-Dec-2016	13:57:44.759	Anchor	FLBB Message	9.510000																																		
$	04-Dec-2016	13:57:44.861	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:57:44.964	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:57:45.165	Anchor	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	13:57:42.568	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:57:45.381	Anchor	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	13:57:54.158	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	13:58:42.605	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,47,700,113,539								
$	04-Dec-2016	13:58:42.535	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.743,AirSaturation[%],124.224,Temperature[Deg.C],18.199,CalPhase[Deg],27.199,TCPhase[Deg],28.020,C1RPh[Deg],35.992,C2RPh[Deg],7.971,C1Amp[mV],529.0,C2Amp[mV],786.2,RawTemp[mV],301.5							
$	04-Dec-2016	13:58:42.538	Anchor	Optode Message	9.500000																											#							
$	04-Dec-2016	13:58:42.937	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,52,700,118,539								
$	04-Dec-2016	13:58:43.171	Anchor	FLBB Message	9.500000																																		
$	04-Dec-2016	13:58:43.179	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:58:43.377	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:58:43.480	Anchor	FLBB Message	9.500000																										Ave 3								
$	04-Dec-2016	13:58:43.488	Anchor	FLBB Message	9.500000																										Pkt 2								
$	04-Dec-2016	13:58:43.695	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:58:55.159	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	13:59:43.307	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,112,539								
$	04-Dec-2016	13:59:42.535	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.759,AirSaturation[%],124.589,Temperature[Deg.C],18.206,CalPhase[Deg],27.168,TCPhase[Deg],27.989,C1RPh[Deg],35.960,C2RPh[Deg],7.971,C1Amp[mV],528.0,C2Amp[mV],785.9,RawTemp[mV],301.2							
$	04-Dec-2016	13:59:42.538	Anchor	Optode Message	9.510000																											#							
$	04-Dec-2016	13:59:43.637	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,114,539								
$	04-Dec-2016	13:59:43.742	Anchor	FLBB Message	9.510000																																		
$	04-Dec-2016	13:59:43.845	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	13:59:44.044	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	13:59:44.051	Anchor	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	13:59:44.250	Anchor	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	13:59:44.663	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	13:59:56.158	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	14:00:44.551	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,116,539								
$	04-Dec-2016	14:00:44.654	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,51,700,122,539								
$	04-Dec-2016	14:00:44.662	Start CP Mode	FLBB Message	9.500000																																		
$	04-Dec-2016	14:00:44.767	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	04-Dec-2016	14:00:44.966	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	04-Dec-2016	14:00:44.974	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	04-Dec-2016	14:00:45.076	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	04-Dec-2016	14:00:42.627	Start CP Mode	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.343,AirSaturation[%],124.794,Temperature[Deg.C],18.209,CalPhase[Deg],27.151,TCPhase[Deg],27.972,C1RPh[Deg],35.944,C2RPh[Deg],7.973,C1Amp[mV],527.5,C2Amp[mV],785.9,RawTemp[mV],301.1							
$	04-Dec-2016	14:00:42.627	Start CP Mode	Optode Message	9.500000																											#							
$	04-Dec-2016	14:00:42.640	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	14:00:45.913	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,46,700,120,539								
$	04-Dec-2016	14:00:43.588	Start CP Mode	Optode Message	9.500000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.630,AirSaturation[%],124.890,Temperature[Deg.C],18.208,CalPhase[Deg],27.143,TCPhase[Deg],27.964,C1RPh[Deg],35.877,C2RPh[Deg],7.913,C1Amp[mV],527.2,C2Amp[mV],785.8,RawTemp[mV],301.2							
$	04-Dec-2016	14:00:43.592	Start CP Mode	Optode Message	9.500000																											#							
$	04-Dec-2016	14:00:46.242	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,51,700,118,539								
$	04-Dec-2016	14:00:46.348	Start CP Mode	FLBB Message	9.500000																																		
$	04-Dec-2016	14:00:46.355	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	04-Dec-2016	14:00:46.460	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	04-Dec-2016	14:00:46.563	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	04-Dec-2016	14:00:46.666	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	04-Dec-2016	14:00:54.159	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	14:00:59.669	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	14:01:05.762	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	04-Dec-2016	14:01:05.763	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	04-Dec-2016	14:01:13.975	DeAnchor	Unparsed SBE41 Message						S>    9.49, 18.1739, 35.2035  																													
$	04-Dec-2016	14:03:21.377	Ascend	FLBB Message	5.230000																										99/99/99,99:99:99,695,50,700,118,539								
$	04-Dec-2016	14:03:21.363	Ascend	Optode Message	5.230000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.973,AirSaturation[%],125.008,Temperature[Deg.C],18.209,CalPhase[Deg],27.133,TCPhase[Deg],27.954,C1RPh[Deg],35.859,C2RPh[Deg],7.905,C1Amp[mV],527.3,C2Amp[mV],782.3,RawTemp[mV],301.1							
$	04-Dec-2016	14:03:21.363	Ascend	Optode Message	5.230000																											#							
$	04-Dec-2016	14:03:21.710	Ascend	FLBB Message	5.230000																										99/99/99,99:99:99,695,53,700,117,539								
$	04-Dec-2016	14:03:21.813	Ascend	FLBB Message	5.230000																																		
$	04-Dec-2016	14:03:21.917	Ascend	FLBB Message	5.230000																										Ser FLBB2K-3310								
$	04-Dec-2016	14:03:22.019	Ascend	FLBB Message	5.230000																										Ver FLNTU 4.08								
$	04-Dec-2016	14:03:22.224	Ascend	FLBB Message	5.230000																										Ave 3								
$	04-Dec-2016	14:03:22.231	Ascend	FLBB Message	5.230000																										Pkt 2								
$	04-Dec-2016	14:03:37.618	Ascend	FLBB Message	3.030000																										99/99/99,99:99:99,695,50,700,116,539								
$	04-Dec-2016	14:03:37.066	Ascend	Optode Message	3.030000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.516,AirSaturation[%],125.819,Temperature[Deg.C],18.188,CalPhase[Deg],27.070,TCPhase[Deg],27.891,C1RPh[Deg],35.861,C2RPh[Deg],7.970,C1Amp[mV],531.7,C2Amp[mV],791.7,RawTemp[mV],301.8							
$	04-Dec-2016	14:03:37.066	Ascend	Optode Message	3.030000																											#							
$	04-Dec-2016	14:03:37.953	Ascend	FLBB Message	3.030000																										99/99/99,99:99:99,695,50,700,113,539								
$	04-Dec-2016	14:03:37.961	Ascend	FLBB Message	3.030000																																		
$	04-Dec-2016	14:03:38.160	Ascend	FLBB Message	3.030000																										Ser FLBB2K-3310								
$	04-Dec-2016	14:03:38.263	Ascend	FLBB Message	3.030000																										Ver FLNTU 4.08								
$	04-Dec-2016	14:03:38.273	Ascend	FLBB Message	3.030000																										Ave 3								
$	04-Dec-2016	14:03:38.376	Ascend	FLBB Message	3.030000																										Pkt 2								
$	04-Dec-2016	14:03:54.288	Ascend	FLBB Message	1.190000																										99/99/99,99:99:99,695,48,700,114,539								
$	04-Dec-2016	14:03:53.171	Ascend	Optode Message	1.190000																											MEASUREMENT,4330,2592,O2Concentration[uM],371.133,AirSaturation[%],126.032,Temperature[Deg.C],18.190,CalPhase[Deg],27.053,TCPhase[Deg],27.874,C1RPh[Deg],35.845,C2RPh[Deg],7.972,C1Amp[mV],537.2,C2Amp[mV],796.1,RawTemp[mV],301.7							
$	04-Dec-2016	14:03:53.172	Ascend	Optode Message	1.190000																											#							
$	04-Dec-2016	14:03:54.726	Ascend	FLBB Message	1.190000																										99/99/99,99:99:99,695,49,700,117,539								
$	04-Dec-2016	14:03:54.831	Ascend	FLBB Message	1.190000																																		
$	04-Dec-2016	14:03:54.840	Ascend	FLBB Message	1.190000																										Ser FLBB2K-3310								
$	04-Dec-2016	14:03:54.943	Ascend	FLBB Message	1.190000																										Ver FLNTU 4.08								
$	04-Dec-2016	14:03:55.046	Ascend	FLBB Message	1.190000																										Ave 3								
$	04-Dec-2016	14:03:55.148	Ascend	FLBB Message	1.190000																										Pkt 2								
$	04-Dec-2016	14:03:54.519	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	04-Dec-2016	14:03:56.350	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	04-Dec-2016	14:03:57.707	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	04-Dec-2016	14:03:58.802	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	04-Dec-2016	14:03:58.802	Init Profile	Unparsed SBE41 Message						powering down  																													
$	04-Dec-2016	14:04:23.730	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	04-Dec-2016	14:05:25.714	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	14:05:31.792	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	04-Dec-2016	14:05:31.963	Dump CP Data	Unparsed SBE41 Message						samples = 165, maxPress = 9.57  																													
$	04-Dec-2016	14:05:39.913	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	04-Dec-2016	14:05:40.598	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	04-Dec-2016	14:05:40.800	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	04-Dec-2016	14:05:41.002	Dump CP Data	Unparsed SBE41 Message						  																													
$	04-Dec-2016	14:05:41.002	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	04-Dec-2016	14:05:46.974	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	04-Dec-2016	14:05:46.974	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.076	Dump CP Data	Unparsed SBE41 Message						    1.88,  18.1739,  35.2032, 18  																													
$	04-Dec-2016	14:05:47.077	Dump CP Data	Unparsed SBE41 Message						    3.98,  18.1742,  35.2033, 16  																													
$	04-Dec-2016	14:05:47.119	Dump CP Data	Unparsed SBE41 Message						    6.04,  18.1728,  35.2034, 12  																													
$	04-Dec-2016	14:05:47.168	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.1731,  35.2034, 21  																													
$	04-Dec-2016	14:05:47.226	Dump CP Data	Unparsed SBE41 Message						    9.47,  18.1734,  35.2037, 98  																													
$	04-Dec-2016	14:05:47.270	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.382	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.383	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.487	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.590	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	14:05:47.590	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	04-Dec-2016	14:05:53.083	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	04-Dec-2016	14:05:53.084	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.183	Dump CP Data	Unparsed SBE41 Message						001346FE898312  																													
$	04-Dec-2016	14:05:53.184	Dump CP Data	Unparsed SBE41 Message						002846FE898310  																													
$	04-Dec-2016	14:05:53.184	Dump CP Data	Unparsed SBE41 Message						003C46FD89830C  																													
$	04-Dec-2016	14:05:53.185	Dump CP Data	Unparsed SBE41 Message						005146FD898315  																													
$	04-Dec-2016	14:05:53.199	Dump CP Data	Unparsed SBE41 Message						005F46FD898462  																													
$	04-Dec-2016	14:05:53.225	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.250	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.276	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.301	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.336	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	14:05:53.356	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	04-Dec-2016	14:05:58.927	Dump CP Data	Unparsed SBE41 Message						S>  																													
