$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	04-Dec-2016	05:12:40.080	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.822,AirSaturation[%],125.903,Temperature[Deg.C],18.181,CalPhase[Deg],27.065,TCPhase[Deg],27.886,C1RPh[Deg],35.864,C2RPh[Deg],7.979,C1Amp[mV],544.7,C2Amp[mV],801.2,RawTemp[mV],302.0							
$	04-Dec-2016	05:12:40.080	Surface Ops	Optode Message	0.150000																											#							
$	04-Dec-2016	05:14:05.441	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,53,700,288,539								
$	04-Dec-2016	05:14:05.641	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,52,700,279,539								
$	04-Dec-2016	05:14:05.649	Surface Ops	FLBB Message	0.120000																																		
$	04-Dec-2016	05:14:05.852	Surface Ops	FLBB Message	0.120000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:14:05.860	Surface Ops	FLBB Message	0.120000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:14:06.061	Surface Ops	FLBB Message	0.120000																										Ave 3								
$	04-Dec-2016	05:14:06.068	Surface Ops	FLBB Message	0.120000																										Pkt 2								
$	04-Dec-2016	05:14:06.164	Surface Ops	Optode Message	0.120000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],305.317,AirSaturation[%],102.553,Temperature[Deg.C],17.659,CalPhase[Deg],29.344,TCPhase[Deg],30.165,C1RPh[Deg],38.144,C2RPh[Deg],7.979,C1Amp[mV],621.5,C2Amp[mV],809.4,RawTemp[mV],318.7							
$	04-Dec-2016	05:14:06.164	Surface Ops	Optode Message	0.120000																											#							
$	04-Dec-2016	05:14:06.402	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,54,700,156,539								
$	04-Dec-2016	05:14:06.604	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,50,700,168,539								
$	04-Dec-2016	05:14:06.612	Surface Ops	FLBB Message	0.120000																																		
$	04-Dec-2016	05:14:06.715	Surface Ops	FLBB Message	0.120000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:14:06.818	Surface Ops	FLBB Message	0.120000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:14:06.920	Surface Ops	FLBB Message	0.120000																										Ave 3								
$	04-Dec-2016	05:14:07.121	Surface Ops	FLBB Message	0.120000																										Pkt 2								
$	04-Dec-2016	05:17:26.886	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,54,700,112,539								
$	04-Dec-2016	05:17:26.614	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.013,AirSaturation[%],100.943,Temperature[Deg.C],17.420,CalPhase[Deg],29.564,TCPhase[Deg],30.385,C1RPh[Deg],38.362,C2RPh[Deg],7.977,C1Amp[mV],629.1,C2Amp[mV],813.3,RawTemp[mV],326.3							
$	04-Dec-2016	05:17:26.616	Anchor	Optode Message	9.520000																											#							
$	04-Dec-2016	05:17:27.217	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,54,700,113,539								
$	04-Dec-2016	05:17:27.322	Anchor	FLBB Message	9.520000																																		
$	04-Dec-2016	05:17:27.424	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:17:27.624	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:17:27.727	Anchor	FLBB Message	9.520000																										Ave 3								
$	04-Dec-2016	05:17:27.829	Anchor	FLBB Message	9.520000																										Pkt 2								
$	04-Dec-2016	05:17:28.737	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:17:40.328	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:18:27.669	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,113,539								
$	04-Dec-2016	05:18:26.713	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],359.097,AirSaturation[%],121.916,Temperature[Deg.C],18.179,CalPhase[Deg],27.396,TCPhase[Deg],28.217,C1RPh[Deg],36.131,C2RPh[Deg],7.914,C1Amp[mV],536.5,C2Amp[mV],788.3,RawTemp[mV],302.1							
$	04-Dec-2016	05:18:26.716	Anchor	Optode Message	9.510000																											#							
$	04-Dec-2016	05:18:27.998	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,112,539								
$	04-Dec-2016	05:18:28.007	Anchor	FLBB Message	9.510000																																		
$	04-Dec-2016	05:18:28.109	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:18:26.738	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:18:28.422	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:18:28.621	Anchor	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	05:18:28.630	Anchor	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	05:18:38.328	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:19:28.604	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,57,700,119,539								
$	04-Dec-2016	05:19:26.614	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.242,AirSaturation[%],123.335,Temperature[Deg.C],18.183,CalPhase[Deg],27.276,TCPhase[Deg],28.097,C1RPh[Deg],36.071,C2RPh[Deg],7.975,C1Amp[mV],532.8,C2Amp[mV],787.7,RawTemp[mV],302.0							
$	04-Dec-2016	05:19:26.617	Anchor	Optode Message	9.520000																											#							
$	04-Dec-2016	05:19:28.838	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,53,700,123,539								
$	04-Dec-2016	05:19:28.942	Anchor	FLBB Message	9.520000																																		
$	04-Dec-2016	05:19:29.143	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:19:29.246	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:19:29.349	Anchor	FLBB Message	9.520000																										Ave 3								
$	04-Dec-2016	05:19:29.451	Anchor	FLBB Message	9.520000																										Pkt 2								
$	04-Dec-2016	05:19:27.738	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:19:39.328	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:20:26.883	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,111,539								
$	04-Dec-2016	05:20:26.613	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.862,AirSaturation[%],124.236,Temperature[Deg.C],18.188,CalPhase[Deg],27.200,TCPhase[Deg],28.021,C1RPh[Deg],35.997,C2RPh[Deg],7.976,C1Amp[mV],530.6,C2Amp[mV],787.3,RawTemp[mV],301.8							
$	04-Dec-2016	05:20:26.616	Anchor	Optode Message	9.520000																											#							
$	04-Dec-2016	05:20:27.119	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,112,539								
$	04-Dec-2016	05:20:27.222	Anchor	FLBB Message	9.520000																																		
$	04-Dec-2016	05:20:27.325	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:20:27.524	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:20:27.533	Anchor	FLBB Message	9.520000																										Ave 3								
$	04-Dec-2016	05:20:27.636	Anchor	FLBB Message	9.520000																										Pkt 2								
$	04-Dec-2016	05:20:28.737	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:20:40.328	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:21:27.574	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,115,539								
$	04-Dec-2016	05:21:26.615	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.817,AirSaturation[%],124.583,Temperature[Deg.C],18.196,CalPhase[Deg],27.170,TCPhase[Deg],27.991,C1RPh[Deg],35.966,C2RPh[Deg],7.975,C1Amp[mV],529.6,C2Amp[mV],787.2,RawTemp[mV],301.5							
$	04-Dec-2016	05:21:26.618	Anchor	Optode Message	9.510000																											#							
$	04-Dec-2016	05:21:27.905	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,120,539								
$	04-Dec-2016	05:21:27.913	Anchor	FLBB Message	9.510000																																		
$	04-Dec-2016	05:21:28.016	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:21:28.119	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:21:28.221	Anchor	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	05:21:28.422	Anchor	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	05:21:26.737	Anchor	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:21:38.327	Anchor	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:22:28.636	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,59,700,115,539								
$	04-Dec-2016	05:22:26.615	Start CP Mode	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.407,AirSaturation[%],124.766,Temperature[Deg.C],18.190,CalPhase[Deg],27.156,TCPhase[Deg],27.977,C1RPh[Deg],35.952,C2RPh[Deg],7.975,C1Amp[mV],529.3,C2Amp[mV],787.1,RawTemp[mV],301.8							
$	04-Dec-2016	05:22:26.618	Start CP Mode	Optode Message	9.510000																											#							
$	04-Dec-2016	05:22:28.968	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,118,539								
$	04-Dec-2016	05:22:29.071	Start CP Mode	FLBB Message	9.510000																																		
$	04-Dec-2016	05:22:29.270	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:22:29.278	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:22:29.476	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	05:22:29.485	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	05:22:29.887	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,117,539								
$	04-Dec-2016	05:22:27.737	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	04-Dec-2016	05:22:27.767	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.763,AirSaturation[%],124.876,Temperature[Deg.C],18.185,CalPhase[Deg],27.148,TCPhase[Deg],27.969,C1RPh[Deg],35.944,C2RPh[Deg],7.976,C1Amp[mV],529.0,C2Amp[mV],787.0,RawTemp[mV],301.9							
$	04-Dec-2016	05:22:27.770	Start CP Mode	Optode Message	9.510000																											#							
$	04-Dec-2016	05:22:30.329	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,48,700,118,539								
$	04-Dec-2016	05:22:30.435	Start CP Mode	FLBB Message	9.510000																																		
$	04-Dec-2016	05:22:30.537	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:22:30.640	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:22:30.839	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	04-Dec-2016	05:22:30.847	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	04-Dec-2016	05:22:39.346	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:22:43.836	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:22:49.929	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	04-Dec-2016	05:22:49.930	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	04-Dec-2016	05:22:58.842	DeAnchor	Unparsed SBE41 Message						S>    9.55, 18.1579, 35.2024  																													
$	04-Dec-2016	05:25:06.055	Ascend	FLBB Message	5.340000																										99/99/99,99:99:99,695,52,700,118,539								
$	04-Dec-2016	05:25:04.841	Ascend	Optode Message	5.340000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.766,AirSaturation[%],124.877,Temperature[Deg.C],18.185,CalPhase[Deg],27.148,TCPhase[Deg],27.969,C1RPh[Deg],35.937,C2RPh[Deg],7.968,C1Amp[mV],529.0,C2Amp[mV],783.7,RawTemp[mV],301.9							
$	04-Dec-2016	05:25:04.841	Ascend	Optode Message	5.340000																											#							
$	04-Dec-2016	05:25:06.387	Ascend	FLBB Message	5.340000																										99/99/99,99:99:99,695,50,700,115,539								
$	04-Dec-2016	05:25:06.491	Ascend	FLBB Message	5.340000																																		
$	04-Dec-2016	05:25:06.499	Ascend	FLBB Message	5.340000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:25:06.602	Ascend	FLBB Message	5.340000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:25:06.802	Ascend	FLBB Message	5.340000																										Ave 3								
$	04-Dec-2016	05:25:06.810	Ascend	FLBB Message	5.340000																										Pkt 2								
$	04-Dec-2016	05:25:20.292	Ascend	FLBB Message	3.080000																										99/99/99,99:99:99,695,48,700,112,539								
$	04-Dec-2016	05:25:20.396	Ascend	FLBB Message	3.080000																										99/99/99,99:99:99,695,49,700,110,539								
$	04-Dec-2016	05:25:20.407	Ascend	FLBB Message	3.080000																																		
$	04-Dec-2016	05:25:20.605	Ascend	FLBB Message	3.080000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:25:20.708	Ascend	FLBB Message	3.080000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:25:20.813	Ascend	FLBB Message	3.080000																										Ave 3								
$	04-Dec-2016	05:25:20.915	Ascend	FLBB Message	3.080000																										Pkt 2								
$	04-Dec-2016	05:25:20.184	Ascend	Optode Message	3.080000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.015,AirSaturation[%],125.629,Temperature[Deg.C],18.181,CalPhase[Deg],27.087,TCPhase[Deg],27.908,C1RPh[Deg],35.881,C2RPh[Deg],7.974,C1Amp[mV],533.9,C2Amp[mV],792.8,RawTemp[mV],302.0							
$	04-Dec-2016	05:25:20.185	Ascend	Optode Message	3.080000																											#							
$	04-Dec-2016	05:25:39.122	Ascend	FLBB Message	1.330000																										99/99/99,99:99:99,695,50,700,129,539								
$	04-Dec-2016	05:25:38.772	Ascend	Optode Message	1.330000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.503,AirSaturation[%],125.798,Temperature[Deg.C],18.182,CalPhase[Deg],27.073,TCPhase[Deg],27.894,C1RPh[Deg],35.869,C2RPh[Deg],7.975,C1Amp[mV],539.4,C2Amp[mV],797.3,RawTemp[mV],302.0							
$	04-Dec-2016	05:25:38.773	Ascend	Optode Message	1.330000																											#							
$	04-Dec-2016	05:25:39.357	Ascend	FLBB Message	1.330000																										99/99/99,99:99:99,695,55,700,121,539								
$	04-Dec-2016	05:25:39.461	Ascend	FLBB Message	1.330000																																		
$	04-Dec-2016	05:25:39.659	Ascend	FLBB Message	1.330000																										Ser FLBB2K-3310								
$	04-Dec-2016	05:25:39.666	Ascend	FLBB Message	1.330000																										Ver FLNTU 4.08								
$	04-Dec-2016	05:25:39.866	Ascend	FLBB Message	1.330000																										Ave 3								
$	04-Dec-2016	05:25:39.875	Ascend	FLBB Message	1.330000																										Pkt 2								
$	04-Dec-2016	05:25:41.333	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	04-Dec-2016	05:25:44.151	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	04-Dec-2016	05:25:45.245	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	04-Dec-2016	05:25:45.245	Init Profile	Unparsed SBE41 Message						powering down  																													
$	04-Dec-2016	05:26:09.742	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	04-Dec-2016	05:27:09.012	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	04-Dec-2016	05:27:15.086	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	04-Dec-2016	05:27:15.261	Dump CP Data	Unparsed SBE41 Message						samples = 167, maxPress = 9.59  																													
$	04-Dec-2016	05:27:23.238	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	04-Dec-2016	05:27:23.924	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	04-Dec-2016	05:27:24.213	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	04-Dec-2016	05:27:24.266	Dump CP Data	Unparsed SBE41 Message						  																													
$	04-Dec-2016	05:27:24.284	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	04-Dec-2016	05:27:30.385	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	04-Dec-2016	05:27:30.386	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.422	Dump CP Data	Unparsed SBE41 Message						    1.98,  18.1677,  35.2019, 21  																													
$	04-Dec-2016	05:27:30.472	Dump CP Data	Unparsed SBE41 Message						    3.97,  18.1666,  35.2019, 15  																													
$	04-Dec-2016	05:27:30.594	Dump CP Data	Unparsed SBE41 Message						    6.06,  18.1664,  35.2020, 14  																													
$	04-Dec-2016	05:27:30.595	Dump CP Data	Unparsed SBE41 Message						    8.22,  18.1644,  35.2020, 19  																													
$	04-Dec-2016	05:27:30.700	Dump CP Data	Unparsed SBE41 Message						    9.48,  18.1589,  35.2019, 98  																													
$	04-Dec-2016	05:27:30.700	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.807	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.807	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.912	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.912	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	04-Dec-2016	05:27:30.913	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	04-Dec-2016	05:27:36.401	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	04-Dec-2016	05:27:36.401	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.500	Dump CP Data	Unparsed SBE41 Message						001446F8898215  																													
$	04-Dec-2016	05:27:36.501	Dump CP Data	Unparsed SBE41 Message						002846F789820F  																													
$	04-Dec-2016	05:27:36.501	Dump CP Data	Unparsed SBE41 Message						003D46F689820E  																													
$	04-Dec-2016	05:27:36.502	Dump CP Data	Unparsed SBE41 Message						005246F4898213  																													
$	04-Dec-2016	05:27:36.518	Dump CP Data	Unparsed SBE41 Message						005F46EF898262  																													
$	04-Dec-2016	05:27:36.543	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.569	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.593	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.619	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.654	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	04-Dec-2016	05:27:36.674	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	04-Dec-2016	05:27:42.313	Dump CP Data	Unparsed SBE41 Message						S>  																													
