$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	05-Dec-2016	05:08:18.144	Surface Ops	Optode Message	0.110000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],370.469,AirSaturation[%],125.695,Temperature[Deg.C],18.146,CalPhase[Deg],27.087,TCPhase[Deg],27.908,C1RPh[Deg],35.874,C2RPh[Deg],7.966,C1Amp[mV],538.2,C2Amp[mV],796.8,RawTemp[mV],303.1							
$	05-Dec-2016	05:08:18.144	Surface Ops	Optode Message	0.110000																											#							
$	05-Dec-2016	05:09:43.297	Surface Ops	FLBB Message	0.110000																										99/99/99,99:99:99,695,48,700,165,539								
$	05-Dec-2016	05:09:43.425	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,50,700,187,539								
$	05-Dec-2016	05:09:43.624	Surface Ops	FLBB Message	-0.080000																																		
$	05-Dec-2016	05:09:43.727	Surface Ops	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:09:43.737	Surface Ops	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:09:43.939	Surface Ops	FLBB Message	-0.080000																										Ave 3								
$	05-Dec-2016	05:09:43.947	Surface Ops	FLBB Message	-0.080000																										Pkt 2								
$	05-Dec-2016	05:09:44.043	Surface Ops	Optode Message	-0.080000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],305.922,AirSaturation[%],102.505,Temperature[Deg.C],17.541,CalPhase[Deg],29.371,TCPhase[Deg],30.192,C1RPh[Deg],38.159,C2RPh[Deg],7.967,C1Amp[mV],613.1,C2Amp[mV],804.0,RawTemp[mV],322.4							
$	05-Dec-2016	05:09:44.044	Surface Ops	Optode Message	-0.080000																											#							
$	05-Dec-2016	05:09:47.142	AB Retract Fast	FLBB Message	-0.080000																										99/99/99,99:99:99,695,51,700,170,539								
$	05-Dec-2016	05:09:47.353	AB Retract Fast	FLBB Message	-0.080000																										99/99/99,99:99:99,695,50,700,187,540								
$	05-Dec-2016	05:09:47.456	AB Retract Fast	FLBB Message	-0.080000																																		
$	05-Dec-2016	05:09:47.605	AB Retract Fast	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:09:47.662	AB Retract Fast	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:09:47.863	AB Retract Fast	FLBB Message	-0.080000																										Ave 3								
$	05-Dec-2016	05:09:47.871	AB Retract Fast	FLBB Message	-0.080000																										Pkt 2								
$	05-Dec-2016	05:13:04.805	Anchor	FLBB Message	9.470000																										99/99/99,99:99:99,695,49,700,117,539								
$	05-Dec-2016	05:13:04.633	Anchor	Optode Message	9.470000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.878,AirSaturation[%],100.874,Temperature[Deg.C],17.250,CalPhase[Deg],29.603,TCPhase[Deg],30.424,C1RPh[Deg],38.330,C2RPh[Deg],7.906,C1Amp[mV],621.2,C2Amp[mV],807.8,RawTemp[mV],331.7							
$	05-Dec-2016	05:13:04.637	Anchor	Optode Message	9.470000																											#							
$	05-Dec-2016	05:13:05.236	Anchor	FLBB Message	9.470000																										99/99/99,99:99:99,695,51,700,118,539								
$	05-Dec-2016	05:13:05.246	Anchor	FLBB Message	9.470000																																		
$	05-Dec-2016	05:13:05.349	Anchor	FLBB Message	9.470000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:13:05.548	Anchor	FLBB Message	9.470000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:13:05.556	Anchor	FLBB Message	9.470000																										Ave 3								
$	05-Dec-2016	05:13:05.658	Anchor	FLBB Message	9.470000																										Pkt 2								
$	05-Dec-2016	05:13:06.836	Anchor	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:13:18.426	Anchor	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:14:05.497	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,49,700,112,539								
$	05-Dec-2016	05:14:04.638	Anchor	Optode Message	9.490000																											MEASUREMENT,4330,2592,O2Concentration[uM],360.351,AirSaturation[%],122.260,Temperature[Deg.C],18.146,CalPhase[Deg],27.372,TCPhase[Deg],28.193,C1RPh[Deg],36.157,C2RPh[Deg],7.963,C1Amp[mV],531.1,C2Amp[mV],785.2,RawTemp[mV],303.1							
$	05-Dec-2016	05:14:04.641	Anchor	Optode Message	9.490000																											#							
$	05-Dec-2016	05:14:05.826	Anchor	FLBB Message	9.490000																										99/99/99,99:99:99,695,53,700,113,539								
$	05-Dec-2016	05:14:05.834	Anchor	FLBB Message	9.490000																																		
$	05-Dec-2016	05:14:06.034	Anchor	FLBB Message	9.490000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:14:06.137	Anchor	FLBB Message	9.490000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:14:06.239	Anchor	FLBB Message	9.490000																										Ave 3								
$	05-Dec-2016	05:14:06.343	Anchor	FLBB Message	9.490000																										Pkt 2								
$	05-Dec-2016	05:14:05.336	Anchor	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:14:16.926	Anchor	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:15:06.540	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,113,539								
$	05-Dec-2016	05:15:04.541	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.081,AirSaturation[%],123.877,Temperature[Deg.C],18.150,CalPhase[Deg],27.236,TCPhase[Deg],28.057,C1RPh[Deg],35.964,C2RPh[Deg],7.907,C1Amp[mV],527.1,C2Amp[mV],784.6,RawTemp[mV],303.0							
$	05-Dec-2016	05:15:04.544	Anchor	Optode Message	9.510000																											#							
$	05-Dec-2016	05:15:06.870	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,47,700,115,539								
$	05-Dec-2016	05:15:07.071	Anchor	FLBB Message	9.510000																																		
$	05-Dec-2016	05:15:07.079	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:15:07.182	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:15:07.285	Anchor	FLBB Message	9.510000																										Ave 3								
$	05-Dec-2016	05:15:07.387	Anchor	FLBB Message	9.510000																										Pkt 2								
$	05-Dec-2016	05:15:06.335	Anchor	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:15:17.926	Anchor	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:16:04.335	Anchor	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:16:05.022	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,54,700,119,539								
$	05-Dec-2016	05:16:04.607	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.093,AirSaturation[%],124.240,Temperature[Deg.C],18.158,CalPhase[Deg],27.205,TCPhase[Deg],28.026,C1RPh[Deg],35.932,C2RPh[Deg],7.906,C1Amp[mV],525.9,C2Amp[mV],784.3,RawTemp[mV],302.7							
$	05-Dec-2016	05:16:04.608	Anchor	Optode Message	9.500000																											#							
$	05-Dec-2016	05:16:05.256	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,48,700,117,539								
$	05-Dec-2016	05:16:05.360	Anchor	FLBB Message	9.500000																																		
$	05-Dec-2016	05:16:05.561	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:16:05.664	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:16:05.766	Anchor	FLBB Message	9.500000																										Ave 3								
$	05-Dec-2016	05:16:05.869	Anchor	FLBB Message	9.500000																										Pkt 2								
$	05-Dec-2016	05:16:15.927	Anchor	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:17:05.620	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,49,700,114,539								
$	05-Dec-2016	05:17:04.541	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.307,AirSaturation[%],124.647,Temperature[Deg.C],18.156,CalPhase[Deg],27.171,TCPhase[Deg],27.992,C1RPh[Deg],35.957,C2RPh[Deg],7.965,C1Amp[mV],525.0,C2Amp[mV],784.1,RawTemp[mV],302.8							
$	05-Dec-2016	05:17:04.544	Anchor	Optode Message	9.510000																											#							
$	05-Dec-2016	05:17:06.044	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,52,700,119,539								
$	05-Dec-2016	05:17:06.053	Anchor	FLBB Message	9.510000																																		
$	05-Dec-2016	05:17:06.155	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:17:06.258	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:17:06.458	Anchor	FLBB Message	9.510000																										Ave 3								
$	05-Dec-2016	05:17:06.562	Anchor	FLBB Message	9.510000																										Pkt 2								
$	05-Dec-2016	05:17:05.336	Anchor	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:17:16.926	Anchor	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:18:06.684	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,54,700,116,539								
$	05-Dec-2016	05:18:04.638	Start CP Mode	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],368.033,AirSaturation[%],124.888,Temperature[Deg.C],18.154,CalPhase[Deg],27.152,TCPhase[Deg],27.973,C1RPh[Deg],35.939,C2RPh[Deg],7.966,C1Amp[mV],524.5,C2Amp[mV],784.1,RawTemp[mV],302.9							
$	05-Dec-2016	05:18:04.638	Start CP Mode	Optode Message	9.500000																											#							
$	05-Dec-2016	05:18:07.013	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,118,539								
$	05-Dec-2016	05:18:07.213	Start CP Mode	FLBB Message	9.500000																																		
$	05-Dec-2016	05:18:07.220	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:18:07.323	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:18:07.428	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	05-Dec-2016	05:18:07.531	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	05-Dec-2016	05:18:07.932	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,50,700,117,539								
$	05-Dec-2016	05:18:05.614	Start CP Mode	Optode Message	9.500000																											MEASUREMENT,4330,2592,O2Concentration[uM],368.583,AirSaturation[%],125.076,Temperature[Deg.C],18.155,CalPhase[Deg],27.136,TCPhase[Deg],27.957,C1RPh[Deg],35.923,C2RPh[Deg],7.966,C1Amp[mV],523.9,C2Amp[mV],784.1,RawTemp[mV],302.9							
$	05-Dec-2016	05:18:05.618	Start CP Mode	Optode Message	9.500000																											#							
$	05-Dec-2016	05:18:08.262	Start CP Mode	FLBB Message	9.500000																										99/99/99,99:99:99,695,54,700,114,539								
$	05-Dec-2016	05:18:08.270	Start CP Mode	FLBB Message	9.500000																																		
$	05-Dec-2016	05:18:08.470	Start CP Mode	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:18:08.477	Start CP Mode	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:18:08.676	Start CP Mode	FLBB Message	9.500000																										Ave 3								
$	05-Dec-2016	05:18:08.684	Start CP Mode	FLBB Message	9.500000																										Pkt 2								
$	05-Dec-2016	05:18:06.335	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	05-Dec-2016	05:18:17.927	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:18:21.774	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:18:27.867	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	05-Dec-2016	05:18:27.868	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	05-Dec-2016	05:18:36.457	DeAnchor	Unparsed SBE41 Message						S>    9.50, 18.1301, 35.2055  																													
$	05-Dec-2016	05:20:50.321	Ascend	FLBB Message	5.070000																										99/99/99,99:99:99,695,50,700,118,539								
$	05-Dec-2016	05:20:49.108	Ascend	Optode Message	5.070000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],368.833,AirSaturation[%],125.162,Temperature[Deg.C],18.155,CalPhase[Deg],27.129,TCPhase[Deg],27.950,C1RPh[Deg],35.907,C2RPh[Deg],7.956,C1Amp[mV],524.0,C2Amp[mV],780.4,RawTemp[mV],302.8							
$	05-Dec-2016	05:20:49.108	Ascend	Optode Message	5.070000																											#							
$	05-Dec-2016	05:20:50.656	Ascend	FLBB Message	5.070000																										99/99/99,99:99:99,695,53,700,117,539								
$	05-Dec-2016	05:20:50.760	Ascend	FLBB Message	5.070000																																		
$	05-Dec-2016	05:20:50.767	Ascend	FLBB Message	5.070000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:20:50.872	Ascend	FLBB Message	5.070000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:20:50.975	Ascend	FLBB Message	5.070000																										Ave 3								
$	05-Dec-2016	05:20:51.077	Ascend	FLBB Message	5.070000																										Pkt 2								
$	05-Dec-2016	05:21:01.511	Ascend	FLBB Message	3.380000																										99/99/99,99:99:99,695,50,700,115,539								
$	05-Dec-2016	05:21:01.363	Ascend	Optode Message	3.380000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.295,AirSaturation[%],125.638,Temperature[Deg.C],18.148,CalPhase[Deg],27.092,TCPhase[Deg],27.913,C1RPh[Deg],35.816,C2RPh[Deg],7.904,C1Amp[mV],528.8,C2Amp[mV],789.7,RawTemp[mV],303.1							
$	05-Dec-2016	05:21:01.363	Ascend	Optode Message	3.380000																											#							
$	05-Dec-2016	05:21:01.936	Ascend	FLBB Message	3.380000																										99/99/99,99:99:99,695,52,700,116,539								
$	05-Dec-2016	05:21:01.944	Ascend	FLBB Message	3.380000																																		
$	05-Dec-2016	05:21:02.145	Ascend	FLBB Message	3.380000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:21:02.247	Ascend	FLBB Message	3.380000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:21:02.351	Ascend	FLBB Message	3.380000																										Ave 3								
$	05-Dec-2016	05:21:02.359	Ascend	FLBB Message	3.380000																										Pkt 2								
$	05-Dec-2016	05:21:20.078	Ascend	Optode Message	1.580000																											MEASUREMENT,4330,2592,O2Concentration[uM],370.578,AirSaturation[%],125.736,Temperature[Deg.C],18.148,CalPhase[Deg],27.084,TCPhase[Deg],27.905,C1RPh[Deg],35.866,C2RPh[Deg],7.961,C1Amp[mV],532.4,C2Amp[mV],792.6,RawTemp[mV],303.1							
$	05-Dec-2016	05:21:20.443	Ascend	FLBB Message	1.580000																										99/99/99,99:99:99,695,49,700,156,539								
$	05-Dec-2016	05:21:20.080	Ascend	Optode Message	1.580000																											#							
$	05-Dec-2016	05:21:20.563	Ascend	FLBB Message	1.580000																										99/99/99,99:99:99,695,49,700,129,539								
$	05-Dec-2016	05:21:20.668	Ascend	FLBB Message	1.580000																																		
$	05-Dec-2016	05:21:20.771	Ascend	FLBB Message	1.580000																										Ser FLBB2K-3310								
$	05-Dec-2016	05:21:20.970	Ascend	FLBB Message	1.580000																										Ver FLNTU 4.08								
$	05-Dec-2016	05:21:20.978	Ascend	FLBB Message	1.580000																										Ave 3								
$	05-Dec-2016	05:21:21.080	Ascend	FLBB Message	1.580000																										Pkt 2								
$	05-Dec-2016	05:21:22.858	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	05-Dec-2016	05:21:25.354	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	05-Dec-2016	05:21:26.450	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	05-Dec-2016	05:21:26.450	Init Profile	Unparsed SBE41 Message						powering down  																													
$	05-Dec-2016	05:21:51.120	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	05-Dec-2016	05:22:54.132	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	05-Dec-2016	05:23:00.228	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	05-Dec-2016	05:23:00.398	Dump CP Data	Unparsed SBE41 Message						samples = 171, maxPress = 9.56  																													
$	05-Dec-2016	05:23:08.348	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	05-Dec-2016	05:23:09.033	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	05-Dec-2016	05:23:09.322	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	05-Dec-2016	05:23:09.423	Dump CP Data	Unparsed SBE41 Message						  																													
$	05-Dec-2016	05:23:09.424	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	05-Dec-2016	05:23:15.401	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	05-Dec-2016	05:23:15.402	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.501	Dump CP Data	Unparsed SBE41 Message						    2.14,  18.1345,  35.2051, 20  																													
$	05-Dec-2016	05:23:15.502	Dump CP Data	Unparsed SBE41 Message						    3.96,  18.1343,  35.2052, 14  																													
$	05-Dec-2016	05:23:15.546	Dump CP Data	Unparsed SBE41 Message						    5.97,  18.1329,  35.2054, 14  																													
$	05-Dec-2016	05:23:15.595	Dump CP Data	Unparsed SBE41 Message						    8.15,  18.1291,  35.2053, 19  																													
$	05-Dec-2016	05:23:15.655	Dump CP Data	Unparsed SBE41 Message						    9.46,  18.1298,  35.2054, 104  																													
$	05-Dec-2016	05:23:15.699	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.809	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.810	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.913	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.914	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	05-Dec-2016	05:23:15.915	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	05-Dec-2016	05:23:21.509	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	05-Dec-2016	05:23:21.509	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.609	Dump CP Data	Unparsed SBE41 Message						001546D7898514  																													
$	05-Dec-2016	05:23:21.609	Dump CP Data	Unparsed SBE41 Message						002846D689850E  																													
$	05-Dec-2016	05:23:21.610	Dump CP Data	Unparsed SBE41 Message						003C46D589850E  																													
$	05-Dec-2016	05:23:21.610	Dump CP Data	Unparsed SBE41 Message						005246D1898513  																													
$	05-Dec-2016	05:23:21.627	Dump CP Data	Unparsed SBE41 Message						005F46D2898568  																													
$	05-Dec-2016	05:23:21.652	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.678	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.704	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.729	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.763	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	05-Dec-2016	05:23:21.783	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	05-Dec-2016	05:23:27.422	Dump CP Data	Unparsed SBE41 Message						S>  																													
