$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	22-Nov-2016	19:36:27.943	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:36:31.823	Surface Ops	Optode Message	1.110000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],372.401,AirSaturation[%],127.701,Temperature[Deg.C],18.667,CalPhase[Deg],26.840,TCPhase[Deg],27.661,C1RPh[Deg],35.637,C2RPh[Deg],7.976,C1Amp[mV],526.8,C2Amp[mV],799.5,RawTemp[mV],286.5							
$	22-Nov-2016	19:36:31.823	Surface Ops	Optode Message	1.110000																											#							
$	22-Nov-2016	19:36:39.507	Surface Ops	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:40:51.617	Surface Ops	FLBB Message	0.830000																										99/99/99,99:99:99,695,50,700,109,538								
$	22-Nov-2016	19:40:51.728	Surface Ops	FLBB Message	0.830000																										99/99/99,99:99:99,695,50,700,111,539								
$	22-Nov-2016	19:40:51.835	Surface Ops	FLBB Message	0.830000																																		
$	22-Nov-2016	19:40:51.940	Surface Ops	FLBB Message	0.830000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:40:52.044	Surface Ops	FLBB Message	0.830000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:40:52.150	Surface Ops	FLBB Message	0.830000																										Ave 3								
$	22-Nov-2016	19:40:52.257	Surface Ops	FLBB Message	0.830000																										Pkt 2								
$	22-Nov-2016	19:40:51.939	Surface Ops	Optode Message	0.830000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.873,AirSaturation[%],100.686,Temperature[Deg.C],18.292,CalPhase[Deg],29.432,TCPhase[Deg],30.253,C1RPh[Deg],38.296,C2RPh[Deg],8.042,C1Amp[mV],608.2,C2Amp[mV],803.9,RawTemp[mV],298.5							
$	22-Nov-2016	19:40:51.939	Surface Ops	Optode Message	0.830000																											#							
$	22-Nov-2016	19:40:52.598	Surface Ops	FLBB Message	0.830000																										99/99/99,99:99:99,695,51,700,113,539								
$	22-Nov-2016	19:40:52.703	Surface Ops	FLBB Message	0.830000																										99/99/99,99:99:99,695,47,700,113,539								
$	22-Nov-2016	19:40:52.808	Surface Ops	FLBB Message	0.830000																																		
$	22-Nov-2016	19:40:52.912	Surface Ops	FLBB Message	0.830000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:40:53.025	Surface Ops	FLBB Message	0.830000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:40:53.129	Surface Ops	FLBB Message	0.830000																										Ave 3								
$	22-Nov-2016	19:40:53.233	Surface Ops	FLBB Message	0.830000																										Pkt 2								
$	22-Nov-2016	19:43:51.605	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,47,700,118,539								
$	22-Nov-2016	19:43:51.439	Anchor	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],297.920,AirSaturation[%],100.502,Temperature[Deg.C],17.868,CalPhase[Deg],29.530,TCPhase[Deg],30.351,C1RPh[Deg],38.405,C2RPh[Deg],8.054,C1Amp[mV],611.0,C2Amp[mV],807.8,RawTemp[mV],312.0							
$	22-Nov-2016	19:43:51.442	Anchor	Optode Message	9.570000																											#							
$	22-Nov-2016	19:43:51.944	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,49,700,118,539								
$	22-Nov-2016	19:43:52.051	Anchor	FLBB Message	9.570000																																		
$	22-Nov-2016	19:43:52.155	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:43:52.267	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:43:52.371	Anchor	FLBB Message	9.570000																										Ave 3								
$	22-Nov-2016	19:43:52.475	Anchor	FLBB Message	9.570000																										Pkt 2								
$	22-Nov-2016	19:44:52.405	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,47,700,115,539								
$	22-Nov-2016	19:44:51.435	Anchor	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],353.256,AirSaturation[%],121.133,Temperature[Deg.C],18.666,CalPhase[Deg],27.381,TCPhase[Deg],28.202,C1RPh[Deg],36.249,C2RPh[Deg],8.047,C1Amp[mV],518.9,C2Amp[mV],771.9,RawTemp[mV],286.6							
$	22-Nov-2016	19:44:51.438	Anchor	Optode Message	9.570000																											#							
$	22-Nov-2016	19:44:52.743	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,116,538								
$	22-Nov-2016	19:44:52.848	Anchor	FLBB Message	9.570000																																		
$	22-Nov-2016	19:44:52.954	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:44:53.058	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:44:53.162	Anchor	FLBB Message	9.570000																										Ave 3								
$	22-Nov-2016	19:44:53.268	Anchor	FLBB Message	9.570000																										Pkt 2								
$	22-Nov-2016	19:44:53.918	Anchor	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:45:05.508	Anchor	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:45:53.094	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,112,538								
$	22-Nov-2016	19:45:51.438	Anchor	Optode Message	9.570000																											MEASUREMENT,4330,2592,O2Concentration[uM],359.144,AirSaturation[%],123.168,Temperature[Deg.C],18.672,CalPhase[Deg],27.208,TCPhase[Deg],28.029,C1RPh[Deg],36.074,C2RPh[Deg],8.045,C1Amp[mV],513.3,C2Amp[mV],770.6,RawTemp[mV],286.4							
$	22-Nov-2016	19:45:51.441	Anchor	Optode Message	9.570000																											#							
$	22-Nov-2016	19:45:53.432	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,46,700,110,538								
$	22-Nov-2016	19:45:53.537	Anchor	FLBB Message	9.570000																																		
$	22-Nov-2016	19:45:53.643	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:45:53.749	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:45:53.853	Anchor	FLBB Message	9.570000																										Ave 3								
$	22-Nov-2016	19:45:53.957	Anchor	FLBB Message	9.570000																										Pkt 2								
$	22-Nov-2016	19:45:52.417	Anchor	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:46:04.008	Anchor	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:46:53.720	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,51,700,116,538								
$	22-Nov-2016	19:46:51.437	Anchor	Optode Message	9.590000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.654,AirSaturation[%],124.381,Temperature[Deg.C],18.676,CalPhase[Deg],27.107,TCPhase[Deg],27.928,C1RPh[Deg],35.972,C2RPh[Deg],8.044,C1Amp[mV],509.8,C2Amp[mV],769.7,RawTemp[mV],286.2							
$	22-Nov-2016	19:46:51.440	Anchor	Optode Message	9.590000																											#							
$	22-Nov-2016	19:46:54.058	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,45,700,115,538								
$	22-Nov-2016	19:46:54.163	Anchor	FLBB Message	9.590000																																		
$	22-Nov-2016	19:46:54.267	Anchor	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:46:54.374	Anchor	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:46:54.479	Anchor	FLBB Message	9.590000																										Ave 3								
$	22-Nov-2016	19:46:54.583	Anchor	FLBB Message	9.590000																										Pkt 2								
$	22-Nov-2016	19:46:53.417	Anchor	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:47:05.008	Anchor	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:47:51.901	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,49,700,113,538								
$	22-Nov-2016	19:47:51.417	Anchor	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:47:51.438	Anchor	Optode Message	9.590000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.332,AirSaturation[%],124.975,Temperature[Deg.C],18.683,CalPhase[Deg],27.057,TCPhase[Deg],27.878,C1RPh[Deg],35.921,C2RPh[Deg],8.043,C1Amp[mV],507.8,C2Amp[mV],769.0,RawTemp[mV],286.0							
$	22-Nov-2016	19:47:51.441	Anchor	Optode Message	9.590000																											#							
$	22-Nov-2016	19:47:52.454	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,47,700,110,538								
$	22-Nov-2016	19:47:52.559	Anchor	FLBB Message	9.590000																																		
$	22-Nov-2016	19:47:52.662	Anchor	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:47:52.768	Anchor	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:47:52.875	Anchor	FLBB Message	9.590000																										Ave 3								
$	22-Nov-2016	19:47:52.979	Anchor	FLBB Message	9.590000																										Pkt 2								
$	22-Nov-2016	19:48:03.008	Anchor	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:48:52.830	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,112,538								
$	22-Nov-2016	19:48:51.447	Start CP Mode	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.305,AirSaturation[%],125.331,Temperature[Deg.C],18.692,CalPhase[Deg],27.026,TCPhase[Deg],27.847,C1RPh[Deg],35.891,C2RPh[Deg],8.043,C1Amp[mV],506.5,C2Amp[mV],768.6,RawTemp[mV],285.7							
$	22-Nov-2016	19:48:51.449	Start CP Mode	Optode Message	9.570000																											#							
$	22-Nov-2016	19:48:53.168	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,52,700,116,538								
$	22-Nov-2016	19:48:53.277	Start CP Mode	FLBB Message	9.570000																																		
$	22-Nov-2016	19:48:53.381	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:48:53.485	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:48:53.590	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	22-Nov-2016	19:48:53.694	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	22-Nov-2016	19:48:52.524	Start CP Mode	Optode Message	9.570000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.102,AirSaturation[%],125.624,Temperature[Deg.C],18.699,CalPhase[Deg],27.001,TCPhase[Deg],27.822,C1RPh[Deg],35.866,C2RPh[Deg],8.044,C1Amp[mV],505.4,C2Amp[mV],768.1,RawTemp[mV],285.5							
$	22-Nov-2016	19:48:52.524	Start CP Mode	Optode Message	9.570000																											#							
$	22-Nov-2016	19:48:52.525	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	22-Nov-2016	19:48:54.246	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,51,700,119,538								
$	22-Nov-2016	19:48:54.351	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,120,538								
$	22-Nov-2016	19:48:54.456	Start CP Mode	FLBB Message	9.570000																																		
$	22-Nov-2016	19:48:54.562	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:48:54.668	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:48:54.772	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	22-Nov-2016	19:48:54.876	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	22-Nov-2016	19:49:04.008	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:49:07.739	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:49:13.826	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	22-Nov-2016	19:49:13.827	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	22-Nov-2016	19:49:22.835	DeAnchor	Unparsed SBE41 Message						S>    9.60, 18.6546, 35.1807  																													
$	22-Nov-2016	19:51:38.697	Ascend	FLBB Message	5.150000																										99/99/99,99:99:99,695,49,700,110,538								
$	22-Nov-2016	19:51:38.270	Ascend	Optode Message	5.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.330,AirSaturation[%],125.704,Temperature[Deg.C],18.700,CalPhase[Deg],26.995,TCPhase[Deg],27.816,C1RPh[Deg],35.844,C2RPh[Deg],8.028,C1Amp[mV],505.5,C2Amp[mV],764.4,RawTemp[mV],285.5							
$	22-Nov-2016	19:51:38.270	Ascend	Optode Message	5.150000																											#							
$	22-Nov-2016	19:51:39.037	Ascend	FLBB Message	5.150000																										99/99/99,99:99:99,695,51,700,112,538								
$	22-Nov-2016	19:51:39.144	Ascend	FLBB Message	5.150000																																		
$	22-Nov-2016	19:51:39.248	Ascend	FLBB Message	5.150000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:51:39.352	Ascend	FLBB Message	5.150000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:51:39.456	Ascend	FLBB Message	5.150000																										Ave 3								
$	22-Nov-2016	19:51:39.563	Ascend	FLBB Message	5.150000																										Pkt 2								
$	22-Nov-2016	19:51:51.643	Ascend	FLBB Message	3.230000																										99/99/99,99:99:99,695,48,700,111,539								
$	22-Nov-2016	19:51:51.218	Ascend	Optode Message	3.230000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.943,AirSaturation[%],126.178,Temperature[Deg.C],18.669,CalPhase[Deg],26.962,TCPhase[Deg],27.783,C1RPh[Deg],35.827,C2RPh[Deg],8.045,C1Amp[mV],507.7,C2Amp[mV],770.6,RawTemp[mV],286.5							
$	22-Nov-2016	19:51:51.221	Ascend	Optode Message	3.230000																											#							
$	22-Nov-2016	19:51:52.225	Ascend	FLBB Message	3.100000																										99/99/99,99:99:99,695,49,700,114,538								
$	22-Nov-2016	19:51:52.331	Ascend	FLBB Message	3.100000																																		
$	22-Nov-2016	19:51:52.435	Ascend	FLBB Message	3.100000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:51:52.539	Ascend	FLBB Message	3.100000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:51:52.645	Ascend	FLBB Message	3.100000																										Ave 3								
$	22-Nov-2016	19:51:52.751	Ascend	FLBB Message	3.100000																										Pkt 2								
$	22-Nov-2016	19:52:07.576	Ascend	FLBB Message	1.360000																										99/99/99,99:99:99,695,48,700,424,538								
$	22-Nov-2016	19:52:07.351	Ascend	Optode Message	1.360000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.934,AirSaturation[%],126.172,Temperature[Deg.C],18.668,CalPhase[Deg],26.962,TCPhase[Deg],27.783,C1RPh[Deg],35.824,C2RPh[Deg],8.040,C1Amp[mV],512.7,C2Amp[mV],774.1,RawTemp[mV],286.5							
$	22-Nov-2016	19:52:07.351	Ascend	Optode Message	1.360000																											#							
$	22-Nov-2016	19:52:07.915	Ascend	FLBB Message	1.360000																										99/99/99,99:99:99,695,50,700,125,538								
$	22-Nov-2016	19:52:08.020	Ascend	FLBB Message	1.360000																																		
$	22-Nov-2016	19:52:08.123	Ascend	FLBB Message	1.360000																										Ser FLBB2K-3310								
$	22-Nov-2016	19:52:08.230	Ascend	FLBB Message	1.360000																										Ver FLNTU 4.08								
$	22-Nov-2016	19:52:08.334	Ascend	FLBB Message	1.360000																										Ave 3								
$	22-Nov-2016	19:52:08.445	Ascend	FLBB Message	1.360000																										Pkt 2								
$	22-Nov-2016	19:52:10.210	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	22-Nov-2016	19:52:12.766	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	22-Nov-2016	19:52:13.946	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	22-Nov-2016	19:52:13.969	Init Profile	Unparsed SBE41 Message						powering down  																													
$	22-Nov-2016	19:52:39.750	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	22-Nov-2016	19:53:39.309	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	22-Nov-2016	19:53:45.408	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	22-Nov-2016	19:53:45.563	Dump CP Data	Unparsed SBE41 Message						samples = 172, maxPress = 9.63  																													
$	22-Nov-2016	19:53:53.513	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	22-Nov-2016	19:53:54.198	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	22-Nov-2016	19:53:54.506	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	22-Nov-2016	19:53:54.609	Dump CP Data	Unparsed SBE41 Message						  																													
$	22-Nov-2016	19:53:54.610	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	22-Nov-2016	19:54:00.480	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	22-Nov-2016	19:54:00.481	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.582	Dump CP Data	Unparsed SBE41 Message						    1.94,  18.6534,  35.1802, 18  																													
$	22-Nov-2016	19:54:00.582	Dump CP Data	Unparsed SBE41 Message						    3.99,  18.6541,  35.1801, 15  																													
$	22-Nov-2016	19:54:00.625	Dump CP Data	Unparsed SBE41 Message						    6.06,  18.6541,  35.1803, 12  																													
$	22-Nov-2016	19:54:00.674	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.6547,  35.1801, 19  																													
$	22-Nov-2016	19:54:00.734	Dump CP Data	Unparsed SBE41 Message						    9.53,  18.6565,  35.1793, 108  																													
$	22-Nov-2016	19:54:00.779	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.889	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.890	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.998	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.998	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	22-Nov-2016	19:54:00.999	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	22-Nov-2016	19:54:06.705	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	22-Nov-2016	19:54:06.706	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:06.706	Dump CP Data	Unparsed SBE41 Message						001348DD896C12  																													
$	22-Nov-2016	19:54:06.728	Dump CP Data	Unparsed SBE41 Message						002848DE896C0F  																													
$	22-Nov-2016	19:54:06.754	Dump CP Data	Unparsed SBE41 Message						003D48DE896C0C  																													
$	22-Nov-2016	19:54:06.779	Dump CP Data	Unparsed SBE41 Message						005148DF896C13  																													
$	22-Nov-2016	19:54:06.814	Dump CP Data	Unparsed SBE41 Message						005F48E0896B6C  																													
$	22-Nov-2016	19:54:06.839	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:06.865	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:06.890	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:07.013	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:07.013	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	22-Nov-2016	19:54:07.014	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	22-Nov-2016	19:54:12.531	Dump CP Data	Unparsed SBE41 Message						S>  																													
