$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	23-Nov-2016	13:07:57.431	Surface Ops	Optode Message	0.090000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.021,AirSaturation[%],125.834,Temperature[Deg.C],18.658,CalPhase[Deg],26.991,TCPhase[Deg],27.812,C1RPh[Deg],35.789,C2RPh[Deg],7.977,C1Amp[mV],507.5,C2Amp[mV],767.6,RawTemp[mV],286.8							
$	23-Nov-2016	13:07:57.431	Surface Ops	Optode Message	0.090000																											#							
$	23-Nov-2016	13:09:21.487	Surface Ops	FLBB Message	0.180000																										99/99/99,99:99:99,695,50,700,114,539								
$	23-Nov-2016	13:09:21.708	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,50,700,113,539								
$	23-Nov-2016	13:09:21.813	Surface Ops	FLBB Message	-0.080000																																		
$	23-Nov-2016	13:09:21.920	Surface Ops	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:09:22.028	Surface Ops	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:09:22.134	Surface Ops	FLBB Message	-0.080000																										Ave 3								
$	23-Nov-2016	13:09:22.238	Surface Ops	FLBB Message	-0.080000																										Pkt 2								
$	23-Nov-2016	13:09:22.133	Surface Ops	Optode Message	-0.080000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.322,AirSaturation[%],101.872,Temperature[Deg.C],17.974,CalPhase[Deg],29.360,TCPhase[Deg],30.181,C1RPh[Deg],38.218,C2RPh[Deg],8.036,C1Amp[mV],577.5,C2Amp[mV],771.4,RawTemp[mV],308.6							
$	23-Nov-2016	13:09:22.133	Surface Ops	Optode Message	-0.080000																											#							
$	23-Nov-2016	13:09:22.580	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,50,700,113,539								
$	23-Nov-2016	13:09:22.689	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,48,700,112,539								
$	23-Nov-2016	13:09:22.794	Surface Ops	FLBB Message	-0.080000																																		
$	23-Nov-2016	13:09:22.898	Surface Ops	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:09:23.002	Surface Ops	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:09:23.107	Surface Ops	FLBB Message	-0.080000																										Ave 3								
$	23-Nov-2016	13:09:23.214	Surface Ops	FLBB Message	-0.080000																										Pkt 2								
$	23-Nov-2016	13:12:36.867	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,49,700,114,538								
$	23-Nov-2016	13:12:36.698	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],298.998,AirSaturation[%],100.879,Temperature[Deg.C],17.874,CalPhase[Deg],29.487,TCPhase[Deg],30.308,C1RPh[Deg],38.340,C2RPh[Deg],8.032,C1Amp[mV],581.9,C2Amp[mV],774.8,RawTemp[mV],311.8							
$	23-Nov-2016	13:12:36.701	Anchor	Optode Message	9.540000																											#							
$	23-Nov-2016	13:12:37.208	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,56,700,115,539								
$	23-Nov-2016	13:12:37.314	Anchor	FLBB Message	9.540000																																		
$	23-Nov-2016	13:12:37.417	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:12:37.523	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:12:37.627	Anchor	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	13:12:37.733	Anchor	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	13:12:39.003	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:12:50.602	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:13:37.662	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,50,700,118,538								
$	23-Nov-2016	13:13:36.600	Anchor	Optode Message	9.560000																											MEASUREMENT,4330,2592,O2Concentration[uM],354.318,AirSaturation[%],121.471,Temperature[Deg.C],18.655,CalPhase[Deg],27.354,TCPhase[Deg],28.175,C1RPh[Deg],36.212,C2RPh[Deg],8.037,C1Amp[mV],509.4,C2Amp[mV],762.9,RawTemp[mV],286.9							
$	23-Nov-2016	13:13:36.603	Anchor	Optode Message	9.560000																											#							
$	23-Nov-2016	13:13:37.999	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,49,700,114,538								
$	23-Nov-2016	13:13:38.106	Anchor	FLBB Message	9.560000																																		
$	23-Nov-2016	13:13:38.210	Anchor	FLBB Message	9.560000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:13:38.314	Anchor	FLBB Message	9.560000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:13:38.418	Anchor	FLBB Message	9.560000																										Ave 3								
$	23-Nov-2016	13:13:38.524	Anchor	FLBB Message	9.560000																										Pkt 2								
$	23-Nov-2016	13:13:37.503	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:13:49.093	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:14:38.670	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,52,700,113,539								
$	23-Nov-2016	13:14:36.693	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],358.979,AirSaturation[%],123.089,Temperature[Deg.C],18.663,CalPhase[Deg],27.216,TCPhase[Deg],28.037,C1RPh[Deg],36.073,C2RPh[Deg],8.036,C1Amp[mV],505.3,C2Amp[mV],762.3,RawTemp[mV],286.6							
$	23-Nov-2016	13:14:36.696	Anchor	Optode Message	9.550000																											#							
$	23-Nov-2016	13:14:39.008	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,53,700,118,539								
$	23-Nov-2016	13:14:39.115	Anchor	FLBB Message	9.550000																																		
$	23-Nov-2016	13:14:39.220	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:14:39.325	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:14:39.429	Anchor	FLBB Message	9.550000																										Ave 3								
$	23-Nov-2016	13:14:39.533	Anchor	FLBB Message	9.550000																										Pkt 2								
$	23-Nov-2016	13:14:38.502	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:14:50.093	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:15:36.584	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:15:39.403	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,49,700,112,538								
$	23-Nov-2016	13:15:36.697	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.603,AirSaturation[%],123.657,Temperature[Deg.C],18.668,CalPhase[Deg],27.168,TCPhase[Deg],27.989,C1RPh[Deg],36.025,C2RPh[Deg],8.036,C1Amp[mV],503.9,C2Amp[mV],761.9,RawTemp[mV],286.5							
$	23-Nov-2016	13:15:36.700	Anchor	Optode Message	9.550000																											#							
$	23-Nov-2016	13:15:39.741	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,52,700,118,538								
$	23-Nov-2016	13:15:39.846	Anchor	FLBB Message	9.550000																																		
$	23-Nov-2016	13:15:39.951	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:15:40.055	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:15:40.165	Anchor	FLBB Message	9.550000																										Ave 3								
$	23-Nov-2016	13:15:40.269	Anchor	FLBB Message	9.550000																										Pkt 2								
$	23-Nov-2016	13:15:48.093	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:16:37.268	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,47,700,117,538								
$	23-Nov-2016	13:16:36.601	Anchor	Optode Message	9.570000																											MEASUREMENT,4330,2592,O2Concentration[uM],362.139,AirSaturation[%],124.195,Temperature[Deg.C],18.672,CalPhase[Deg],27.123,TCPhase[Deg],27.944,C1RPh[Deg],35.981,C2RPh[Deg],8.037,C1Amp[mV],502.7,C2Amp[mV],761.8,RawTemp[mV],286.4							
$	23-Nov-2016	13:16:36.605	Anchor	Optode Message	9.570000																											#							
$	23-Nov-2016	13:16:37.613	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,49,700,115,538								
$	23-Nov-2016	13:16:37.718	Anchor	FLBB Message	9.570000																																		
$	23-Nov-2016	13:16:37.822	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:16:37.928	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:16:38.032	Anchor	FLBB Message	9.570000																										Ave 3								
$	23-Nov-2016	13:16:38.138	Anchor	FLBB Message	9.570000																										Pkt 2								
$	23-Nov-2016	13:16:37.612	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:16:49.094	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:17:38.304	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,47,700,115,538								
$	23-Nov-2016	13:17:36.704	Start CP Mode	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.765,AirSaturation[%],124.426,Temperature[Deg.C],18.678,CalPhase[Deg],27.103,TCPhase[Deg],27.924,C1RPh[Deg],35.961,C2RPh[Deg],8.037,C1Amp[mV],501.9,C2Amp[mV],761.7,RawTemp[mV],286.1							
$	23-Nov-2016	13:17:36.707	Start CP Mode	Optode Message	9.570000																											#							
$	23-Nov-2016	13:17:38.647	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,49,700,115,538								
$	23-Nov-2016	13:17:38.754	Start CP Mode	FLBB Message	9.570000																																		
$	23-Nov-2016	13:17:38.858	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:17:38.964	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:17:39.068	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	23-Nov-2016	13:17:39.172	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	23-Nov-2016	13:17:37.750	Start CP Mode	Optode Message	9.570000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.331,AirSaturation[%],124.643,Temperature[Deg.C],18.687,CalPhase[Deg],27.083,TCPhase[Deg],27.904,C1RPh[Deg],35.941,C2RPh[Deg],8.037,C1Amp[mV],501.5,C2Amp[mV],761.6,RawTemp[mV],285.9							
$	23-Nov-2016	13:17:37.753	Start CP Mode	Optode Message	9.570000																											#							
$	23-Nov-2016	13:17:39.509	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,53,700,115,538								
$	23-Nov-2016	13:17:39.620	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,49,700,112,538								
$	23-Nov-2016	13:17:39.727	Start CP Mode	FLBB Message	9.570000																																		
$	23-Nov-2016	13:17:39.830	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:17:39.935	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:17:40.041	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	23-Nov-2016	13:17:40.145	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	23-Nov-2016	13:17:38.646	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	13:18:29.408	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:18:35.496	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	23-Nov-2016	13:18:35.497	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	23-Nov-2016	13:18:44.143	DeAnchor	Unparsed SBE41 Message						S>    9.54, 18.6444, 35.1823  																													
$	23-Nov-2016	13:20:56.917	Ascend	FLBB Message	5.100000																										99/99/99,99:99:99,695,50,700,115,538								
$	23-Nov-2016	13:20:57.023	Ascend	Optode Message	5.100000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.569,AirSaturation[%],124.726,Temperature[Deg.C],18.688,CalPhase[Deg],27.076,TCPhase[Deg],27.897,C1RPh[Deg],35.921,C2RPh[Deg],8.024,C1Amp[mV],501.6,C2Amp[mV],758.0,RawTemp[mV],285.8							
$	23-Nov-2016	13:20:57.023	Ascend	Optode Message	5.100000																											#							
$	23-Nov-2016	13:20:57.511	Ascend	FLBB Message	4.880000																										99/99/99,99:99:99,695,49,700,113,539								
$	23-Nov-2016	13:20:57.616	Ascend	FLBB Message	4.880000																																		
$	23-Nov-2016	13:20:57.722	Ascend	FLBB Message	4.880000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:20:57.828	Ascend	FLBB Message	4.880000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:20:57.933	Ascend	FLBB Message	4.880000																										Ave 3								
$	23-Nov-2016	13:20:58.037	Ascend	FLBB Message	4.880000																										Pkt 2								
$	23-Nov-2016	13:21:06.324	Ascend	FLBB Message	3.500000																										99/99/99,99:99:99,695,46,700,112,539								
$	23-Nov-2016	13:21:06.433	Ascend	Optode Message	3.500000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.820,AirSaturation[%],125.423,Temperature[Deg.C],18.658,CalPhase[Deg],27.024,TCPhase[Deg],27.845,C1RPh[Deg],35.883,C2RPh[Deg],8.038,C1Amp[mV],503.5,C2Amp[mV],764.4,RawTemp[mV],286.8							
$	23-Nov-2016	13:21:06.433	Ascend	Optode Message	3.500000																											#							
$	23-Nov-2016	13:21:06.667	Ascend	FLBB Message	3.500000																										99/99/99,99:99:99,695,47,700,109,539								
$	23-Nov-2016	13:21:06.777	Ascend	FLBB Message	3.500000																																		
$	23-Nov-2016	13:21:06.881	Ascend	FLBB Message	3.500000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:21:06.985	Ascend	FLBB Message	3.500000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:21:07.089	Ascend	FLBB Message	3.500000																										Ave 3								
$	23-Nov-2016	13:21:07.195	Ascend	FLBB Message	3.500000																										Pkt 2								
$	23-Nov-2016	13:21:21.920	Ascend	FLBB Message	1.610000																										99/99/99,99:99:99,695,49,700,110,538								
$	23-Nov-2016	13:21:21.804	Ascend	Optode Message	1.610000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.109,AirSaturation[%],125.519,Temperature[Deg.C],18.657,CalPhase[Deg],27.017,TCPhase[Deg],27.838,C1RPh[Deg],35.875,C2RPh[Deg],8.037,C1Amp[mV],504.9,C2Amp[mV],765.2,RawTemp[mV],286.8							
$	23-Nov-2016	13:21:21.804	Ascend	Optode Message	1.610000																											#							
$	23-Nov-2016	13:21:22.257	Ascend	FLBB Message	1.610000																										99/99/99,99:99:99,695,50,700,118,539								
$	23-Nov-2016	13:21:22.363	Ascend	FLBB Message	1.610000																																		
$	23-Nov-2016	13:21:22.467	Ascend	FLBB Message	1.610000																										Ser FLBB2K-3310								
$	23-Nov-2016	13:21:22.571	Ascend	FLBB Message	1.610000																										Ver FLNTU 4.08								
$	23-Nov-2016	13:21:22.677	Ascend	FLBB Message	1.610000																										Ave 3								
$	23-Nov-2016	13:21:22.786	Ascend	FLBB Message	1.610000																										Pkt 2								
$	23-Nov-2016	13:21:23.799	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	23-Nov-2016	13:21:24.237	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	13:21:26.617	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	13:21:27.796	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	23-Nov-2016	13:21:27.819	Init Profile	Unparsed SBE41 Message						powering down  																													
$	23-Nov-2016	13:21:31.382	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	23-Nov-2016	13:22:53.014	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	13:22:59.088	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	23-Nov-2016	13:22:59.243	Dump CP Data	Unparsed SBE41 Message						samples = 164, maxPress = 9.61  																													
$	23-Nov-2016	13:23:07.192	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	23-Nov-2016	13:23:07.878	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	23-Nov-2016	13:23:08.106	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	23-Nov-2016	13:23:08.203	Dump CP Data	Unparsed SBE41 Message						  																													
$	23-Nov-2016	13:23:08.221	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	23-Nov-2016	13:23:14.287	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	23-Nov-2016	13:23:14.288	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.325	Dump CP Data	Unparsed SBE41 Message						    2.16,  18.6422,  35.1823, 15  																													
$	23-Nov-2016	13:23:14.374	Dump CP Data	Unparsed SBE41 Message						    4.01,  18.6429,  35.1822, 13  																													
$	23-Nov-2016	13:23:14.423	Dump CP Data	Unparsed SBE41 Message						    6.05,  18.6444,  35.1823, 13  																													
$	23-Nov-2016	13:23:14.473	Dump CP Data	Unparsed SBE41 Message						    8.09,  18.6444,  35.1823, 17  																													
$	23-Nov-2016	13:23:14.595	Dump CP Data	Unparsed SBE41 Message						    9.51,  18.6442,  35.1824, 106  																													
$	23-Nov-2016	13:23:14.596	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.703	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.704	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.811	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.812	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	13:23:14.812	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	13:23:20.416	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	23-Nov-2016	13:23:20.416	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.417	Dump CP Data	Unparsed SBE41 Message						001648D2896E0F  																													
$	23-Nov-2016	13:23:20.439	Dump CP Data	Unparsed SBE41 Message						002848D3896E0D  																													
$	23-Nov-2016	13:23:20.464	Dump CP Data	Unparsed SBE41 Message						003D48D4896E0D  																													
$	23-Nov-2016	13:23:20.490	Dump CP Data	Unparsed SBE41 Message						005148D4896E11  																													
$	23-Nov-2016	13:23:20.524	Dump CP Data	Unparsed SBE41 Message						005F48D4896E6A  																													
$	23-Nov-2016	13:23:20.549	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.575	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.600	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.723	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.724	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	13:23:20.724	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	13:23:26.248	Dump CP Data	Unparsed SBE41 Message						S>  																													
