$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	23-Nov-2016	16:42:10.416	Surface Ops	Optode Message	0.220000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.246,AirSaturation[%],125.536,Temperature[Deg.C],18.646,CalPhase[Deg],27.017,TCPhase[Deg],27.838,C1RPh[Deg],35.872,C2RPh[Deg],8.034,C1Amp[mV],507.1,C2Amp[mV],767.1,RawTemp[mV],287.2							
$	23-Nov-2016	16:42:10.416	Surface Ops	Optode Message	0.220000																											#							
$	23-Nov-2016	16:43:34.449	Surface Ops	FLBB Message	0.200000																										99/99/99,99:99:99,695,50,700,115,539								
$	23-Nov-2016	16:43:34.696	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,46,700,110,539								
$	23-Nov-2016	16:43:34.802	Surface Ops	FLBB Message	-0.080000																																		
$	23-Nov-2016	16:43:34.908	Surface Ops	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:43:35.024	Surface Ops	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:43:35.130	Surface Ops	FLBB Message	-0.080000																										Ave 3								
$	23-Nov-2016	16:43:35.235	Surface Ops	FLBB Message	-0.080000																										Pkt 2								
$	23-Nov-2016	16:43:35.129	Surface Ops	Optode Message	-0.080000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.763,AirSaturation[%],101.671,Temperature[Deg.C],17.968,CalPhase[Deg],29.383,TCPhase[Deg],30.204,C1RPh[Deg],38.237,C2RPh[Deg],8.033,C1Amp[mV],577.0,C2Amp[mV],771.4,RawTemp[mV],308.8							
$	23-Nov-2016	16:43:35.129	Surface Ops	Optode Message	-0.080000																											#							
$	23-Nov-2016	16:43:35.779	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,49,700,115,539								
$	23-Nov-2016	16:43:35.886	Surface Ops	FLBB Message	-0.080000																										99/99/99,99:99:99,695,49,700,110,539								
$	23-Nov-2016	16:43:35.993	Surface Ops	FLBB Message	-0.080000																																		
$	23-Nov-2016	16:43:36.097	Surface Ops	FLBB Message	-0.080000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:43:36.201	Surface Ops	FLBB Message	-0.080000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:43:36.305	Surface Ops	FLBB Message	-0.080000																										Ave 3								
$	23-Nov-2016	16:43:36.411	Surface Ops	FLBB Message	-0.080000																										Pkt 2								
$	23-Nov-2016	16:46:34.649	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,115,539								
$	23-Nov-2016	16:46:34.480	Anchor	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.036,AirSaturation[%],101.010,Temperature[Deg.C],17.769,CalPhase[Deg],29.492,TCPhase[Deg],30.313,C1RPh[Deg],38.282,C2RPh[Deg],7.968,C1Amp[mV],580.6,C2Amp[mV],774.8,RawTemp[mV],315.2							
$	23-Nov-2016	16:46:34.483	Anchor	Optode Message	9.570000																											#							
$	23-Nov-2016	16:46:34.991	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,45,700,115,539								
$	23-Nov-2016	16:46:35.097	Anchor	FLBB Message	9.570000																																		
$	23-Nov-2016	16:46:35.201	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:46:35.308	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:46:35.413	Anchor	FLBB Message	9.570000																										Ave 3								
$	23-Nov-2016	16:46:35.517	Anchor	FLBB Message	9.570000																										Pkt 2								
$	23-Nov-2016	16:46:36.963	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:46:48.553	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:47:35.236	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,50,700,109,539								
$	23-Nov-2016	16:47:34.486	Anchor	Optode Message	9.560000																											MEASUREMENT,4330,2592,O2Concentration[uM],354.111,AirSaturation[%],121.372,Temperature[Deg.C],18.644,CalPhase[Deg],27.364,TCPhase[Deg],28.185,C1RPh[Deg],36.216,C2RPh[Deg],8.031,C1Amp[mV],509.2,C2Amp[mV],762.9,RawTemp[mV],287.3							
$	23-Nov-2016	16:47:34.489	Anchor	Optode Message	9.560000																											#							
$	23-Nov-2016	16:47:35.580	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,49,700,113,538								
$	23-Nov-2016	16:47:35.685	Anchor	FLBB Message	9.560000																																		
$	23-Nov-2016	16:47:35.789	Anchor	FLBB Message	9.560000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:47:35.893	Anchor	FLBB Message	9.560000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:47:35.999	Anchor	FLBB Message	9.560000																										Ave 3								
$	23-Nov-2016	16:47:36.105	Anchor	FLBB Message	9.560000																										Pkt 2								
$	23-Nov-2016	16:47:35.578	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:47:47.052	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:48:36.188	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,48,700,116,538								
$	23-Nov-2016	16:48:34.507	Anchor	Optode Message	9.560000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],359.830,AirSaturation[%],123.347,Temperature[Deg.C],18.649,CalPhase[Deg],27.197,TCPhase[Deg],28.018,C1RPh[Deg],36.050,C2RPh[Deg],8.033,C1Amp[mV],504.4,C2Amp[mV],762.3,RawTemp[mV],287.1							
$	23-Nov-2016	16:48:34.510	Anchor	Optode Message	9.560000																											#							
$	23-Nov-2016	16:48:36.526	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,48,700,115,539								
$	23-Nov-2016	16:48:36.636	Anchor	FLBB Message	9.560000																																		
$	23-Nov-2016	16:48:36.742	Anchor	FLBB Message	9.560000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:48:36.846	Anchor	FLBB Message	9.560000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:48:36.952	Anchor	FLBB Message	9.560000																										Ave 3								
$	23-Nov-2016	16:48:37.056	Anchor	FLBB Message	9.560000																										Pkt 2								
$	23-Nov-2016	16:48:36.635	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:48:48.053	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:49:36.974	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,50,700,115,538								
$	23-Nov-2016	16:49:34.462	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:49:34.485	Anchor	Optode Message	9.580000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],361.965,AirSaturation[%],124.089,Temperature[Deg.C],18.654,CalPhase[Deg],27.134,TCPhase[Deg],27.955,C1RPh[Deg],35.930,C2RPh[Deg],7.974,C1Amp[mV],502.6,C2Amp[mV],762.0,RawTemp[mV],286.9							
$	23-Nov-2016	16:49:34.489	Anchor	Optode Message	9.580000																											#							
$	23-Nov-2016	16:49:37.525	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,47,700,110,538								
$	23-Nov-2016	16:49:37.630	Anchor	FLBB Message	9.580000																																		
$	23-Nov-2016	16:49:37.735	Anchor	FLBB Message	9.580000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:49:37.842	Anchor	FLBB Message	9.580000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:49:37.946	Anchor	FLBB Message	9.580000																										Ave 3								
$	23-Nov-2016	16:49:38.050	Anchor	FLBB Message	9.580000																										Pkt 2								
$	23-Nov-2016	16:49:46.052	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:50:34.847	Anchor	FLBB Message	9.600000																										99/99/99,99:99:99,695,50,700,115,538								
$	23-Nov-2016	16:50:34.484	Anchor	Optode Message	9.600000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.042,AirSaturation[%],124.475,Temperature[Deg.C],18.660,CalPhase[Deg],27.102,TCPhase[Deg],27.923,C1RPh[Deg],35.956,C2RPh[Deg],8.033,C1Amp[mV],501.6,C2Amp[mV],761.8,RawTemp[mV],286.7							
$	23-Nov-2016	16:50:34.487	Anchor	Optode Message	9.600000																											#							
$	23-Nov-2016	16:50:35.188	Anchor	FLBB Message	9.600000																										99/99/99,99:99:99,695,50,700,115,538								
$	23-Nov-2016	16:50:35.295	Anchor	FLBB Message	9.600000																																		
$	23-Nov-2016	16:50:35.399	Anchor	FLBB Message	9.600000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:50:35.508	Anchor	FLBB Message	9.600000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:50:35.613	Anchor	FLBB Message	9.600000																										Ave 3								
$	23-Nov-2016	16:50:35.718	Anchor	FLBB Message	9.600000																										Pkt 2								
$	23-Nov-2016	16:50:35.611	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:50:47.053	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:51:35.933	Start CP Mode	FLBB Message	9.590000																										99/99/99,99:99:99,695,48,700,111,538								
$	23-Nov-2016	16:51:34.534	Start CP Mode	Optode Message	9.590000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.775,AirSaturation[%],124.724,Temperature[Deg.C],18.659,CalPhase[Deg],27.081,TCPhase[Deg],27.902,C1RPh[Deg],35.936,C2RPh[Deg],8.034,C1Amp[mV],501.1,C2Amp[mV],761.7,RawTemp[mV],286.8							
$	23-Nov-2016	16:51:34.537	Start CP Mode	Optode Message	9.590000																											#							
$	23-Nov-2016	16:51:36.272	Start CP Mode	FLBB Message	9.590000																										99/99/99,99:99:99,695,47,700,105,539								
$	23-Nov-2016	16:51:36.381	Start CP Mode	FLBB Message	9.590000																																		
$	23-Nov-2016	16:51:36.490	Start CP Mode	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:51:36.594	Start CP Mode	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:51:36.699	Start CP Mode	FLBB Message	9.590000																										Ave 3								
$	23-Nov-2016	16:51:36.803	Start CP Mode	FLBB Message	9.590000																										Pkt 2								
$	23-Nov-2016	16:51:37.113	Start CP Mode	FLBB Message	9.590000																										99/99/99,99:99:99,695,48,700,108,538								
$	23-Nov-2016	16:51:35.580	Start CP Mode	Optode Message	9.590000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.425,AirSaturation[%],124.941,Temperature[Deg.C],18.657,CalPhase[Deg],27.064,TCPhase[Deg],27.885,C1RPh[Deg],35.917,C2RPh[Deg],8.032,C1Amp[mV],500.5,C2Amp[mV],761.6,RawTemp[mV],286.8							
$	23-Nov-2016	16:51:35.583	Start CP Mode	Optode Message	9.590000																											#							
$	23-Nov-2016	16:51:37.452	Start CP Mode	FLBB Message	9.590000																										99/99/99,99:99:99,695,48,700,112,538								
$	23-Nov-2016	16:51:37.557	Start CP Mode	FLBB Message	9.590000																																		
$	23-Nov-2016	16:51:37.661	Start CP Mode	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:51:37.765	Start CP Mode	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:51:37.876	Start CP Mode	FLBB Message	9.590000																										Ave 3								
$	23-Nov-2016	16:51:37.980	Start CP Mode	FLBB Message	9.590000																										Pkt 2								
$	23-Nov-2016	16:51:36.593	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	16:52:27.245	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:52:33.335	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	23-Nov-2016	16:52:33.336	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	23-Nov-2016	16:52:41.484	DeAnchor	Unparsed SBE41 Message						S>    9.63, 18.6317, 35.1829  																													
$	23-Nov-2016	16:55:00.887	Ascend	FLBB Message	5.470000																										99/99/99,99:99:99,695,49,700,109,539								
$	23-Nov-2016	16:55:00.993	Ascend	Optode Message	5.470000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.914,AirSaturation[%],125.110,Temperature[Deg.C],18.658,CalPhase[Deg],27.050,TCPhase[Deg],27.871,C1RPh[Deg],35.891,C2RPh[Deg],8.020,C1Amp[mV],500.6,C2Amp[mV],758.1,RawTemp[mV],286.8							
$	23-Nov-2016	16:55:00.993	Ascend	Optode Message	5.470000																											#							
$	23-Nov-2016	16:55:01.232	Ascend	FLBB Message	5.470000																										99/99/99,99:99:99,695,52,700,113,539								
$	23-Nov-2016	16:55:01.337	Ascend	FLBB Message	5.470000																																		
$	23-Nov-2016	16:55:01.443	Ascend	FLBB Message	5.470000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:55:01.547	Ascend	FLBB Message	5.470000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:55:01.651	Ascend	FLBB Message	5.470000																										Ave 3								
$	23-Nov-2016	16:55:01.757	Ascend	FLBB Message	5.470000																										Pkt 2								
$	23-Nov-2016	16:55:16.901	Ascend	FLBB Message	3.180000																										99/99/99,99:99:99,695,51,700,115,539								
$	23-Nov-2016	16:55:16.675	Ascend	Optode Message	3.180000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.184,AirSaturation[%],125.519,Temperature[Deg.C],18.647,CalPhase[Deg],27.018,TCPhase[Deg],27.839,C1RPh[Deg],35.873,C2RPh[Deg],8.034,C1Amp[mV],502.3,C2Amp[mV],764.1,RawTemp[mV],287.1							
$	23-Nov-2016	16:55:16.676	Ascend	Optode Message	3.180000																											#							
$	23-Nov-2016	16:55:17.481	Ascend	FLBB Message	3.030000																										99/99/99,99:99:99,695,47,700,117,539								
$	23-Nov-2016	16:55:17.589	Ascend	FLBB Message	3.030000																																		
$	23-Nov-2016	16:55:17.693	Ascend	FLBB Message	3.030000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:55:17.798	Ascend	FLBB Message	3.030000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:55:17.902	Ascend	FLBB Message	3.030000																										Ave 3								
$	23-Nov-2016	16:55:18.006	Ascend	FLBB Message	3.030000																										Pkt 2								
$	23-Nov-2016	16:55:33.139	Ascend	FLBB Message	1.410000																										99/99/99,99:99:99,695,50,700,109,538								
$	23-Nov-2016	16:55:32.716	Ascend	Optode Message	1.410000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.525,AirSaturation[%],125.635,Temperature[Deg.C],18.647,CalPhase[Deg],27.009,TCPhase[Deg],27.830,C1RPh[Deg],35.864,C2RPh[Deg],8.034,C1Amp[mV],504.5,C2Amp[mV],765.6,RawTemp[mV],287.2							
$	23-Nov-2016	16:55:32.719	Ascend	Optode Message	1.410000																											#							
$	23-Nov-2016	16:55:33.478	Ascend	FLBB Message	1.410000																										99/99/99,99:99:99,695,50,700,112,538								
$	23-Nov-2016	16:55:33.583	Ascend	FLBB Message	1.410000																																		
$	23-Nov-2016	16:55:33.688	Ascend	FLBB Message	1.410000																										Ser FLBB2K-3310								
$	23-Nov-2016	16:55:33.793	Ascend	FLBB Message	1.410000																										Ver FLNTU 4.08								
$	23-Nov-2016	16:55:33.898	Ascend	FLBB Message	1.410000																										Ave 3								
$	23-Nov-2016	16:55:34.002	Ascend	FLBB Message	1.410000																										Pkt 2								
$	23-Nov-2016	16:55:35.696	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	23-Nov-2016	16:55:38.123	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	16:55:39.303	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	23-Nov-2016	16:55:39.325	Init Profile	Unparsed SBE41 Message						powering down  																													
$	23-Nov-2016	16:56:04.537	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	23-Nov-2016	16:57:04.345	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	16:57:10.422	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	23-Nov-2016	16:57:10.576	Dump CP Data	Unparsed SBE41 Message						samples = 179, maxPress = 9.66  																													
$	23-Nov-2016	16:57:18.526	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	23-Nov-2016	16:57:19.206	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	23-Nov-2016	16:57:19.457	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	23-Nov-2016	16:57:19.606	Dump CP Data	Unparsed SBE41 Message						  																													
$	23-Nov-2016	16:57:19.624	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	23-Nov-2016	16:57:25.625	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	23-Nov-2016	16:57:25.626	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:25.660	Dump CP Data	Unparsed SBE41 Message						    1.97,  18.6304,  35.1830, 19  																													
$	23-Nov-2016	16:57:25.710	Dump CP Data	Unparsed SBE41 Message						    3.97,  18.6326,  35.1826, 15  																													
$	23-Nov-2016	16:57:25.759	Dump CP Data	Unparsed SBE41 Message						    6.05,  18.6332,  35.1826, 13  																													
$	23-Nov-2016	16:57:25.809	Dump CP Data	Unparsed SBE41 Message						    8.16,  18.6330,  35.1827, 19  																													
$	23-Nov-2016	16:57:25.932	Dump CP Data	Unparsed SBE41 Message						    9.54,  18.6319,  35.1830, 113  																													
$	23-Nov-2016	16:57:25.933	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:26.038	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:26.039	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:26.145	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:26.145	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	16:57:26.146	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	16:57:31.747	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	23-Nov-2016	16:57:31.748	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:31.748	Dump CP Data	Unparsed SBE41 Message						001448C6896F13  																													
$	23-Nov-2016	16:57:31.768	Dump CP Data	Unparsed SBE41 Message						002848C9896F0F  																													
$	23-Nov-2016	16:57:31.794	Dump CP Data	Unparsed SBE41 Message						003C48C9896F0D  																													
$	23-Nov-2016	16:57:31.819	Dump CP Data	Unparsed SBE41 Message						005248C9896F13  																													
$	23-Nov-2016	16:57:31.854	Dump CP Data	Unparsed SBE41 Message						005F48C8896F71  																													
$	23-Nov-2016	16:57:31.879	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:31.905	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:31.930	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:32.054	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:32.055	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	16:57:32.055	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	16:57:37.574	Dump CP Data	Unparsed SBE41 Message						S>  																													
