$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	23-Nov-2016	02:33:28.007	Surface Ops	Optode Message	0.490000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.050,AirSaturation[%],125.152,Temperature[Deg.C],18.656,CalPhase[Deg],27.047,TCPhase[Deg],27.868,C1RPh[Deg],35.916,C2RPh[Deg],8.048,C1Amp[mV],509.9,C2Amp[mV],767.6,RawTemp[mV],286.9							
$	23-Nov-2016	02:33:28.007	Surface Ops	Optode Message	0.490000																											#							
$	23-Nov-2016	02:37:08.874	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,50,700,110,539								
$	23-Nov-2016	02:37:09.185	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,51,700,117,539								
$	23-Nov-2016	02:37:09.291	Surface Ops	FLBB Message	0.150000																																		
$	23-Nov-2016	02:37:09.397	Surface Ops	FLBB Message	0.150000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:37:09.504	Surface Ops	FLBB Message	0.150000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:37:09.609	Surface Ops	FLBB Message	0.150000																										Ave 3								
$	23-Nov-2016	02:37:09.715	Surface Ops	FLBB Message	0.150000																										Pkt 2								
$	23-Nov-2016	02:37:09.608	Surface Ops	Optode Message	0.270000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],297.601,AirSaturation[%],101.215,Temperature[Deg.C],18.264,CalPhase[Deg],29.379,TCPhase[Deg],30.200,C1RPh[Deg],38.250,C2RPh[Deg],8.050,C1Amp[mV],580.5,C2Amp[mV],772.2,RawTemp[mV],299.4							
$	23-Nov-2016	02:37:09.608	Surface Ops	Optode Message	0.270000																											#							
$	23-Nov-2016	02:37:10.172	Surface Ops	FLBB Message	0.270000																										99/99/99,99:99:99,695,51,700,115,539								
$	23-Nov-2016	02:37:10.279	Surface Ops	FLBB Message	0.270000																										99/99/99,99:99:99,695,51,700,113,539								
$	23-Nov-2016	02:37:10.392	Surface Ops	FLBB Message	0.270000																																		
$	23-Nov-2016	02:37:10.495	Surface Ops	FLBB Message	0.270000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:37:10.600	Surface Ops	FLBB Message	0.270000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:37:10.704	Surface Ops	FLBB Message	0.270000																										Ave 3								
$	23-Nov-2016	02:37:10.810	Surface Ops	FLBB Message	0.270000																										Pkt 2								
$	23-Nov-2016	02:40:30.250	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],297.308,AirSaturation[%],100.378,Temperature[Deg.C],17.908,CalPhase[Deg],29.537,TCPhase[Deg],30.358,C1RPh[Deg],38.403,C2RPh[Deg],8.045,C1Amp[mV],586.2,C2Amp[mV],776.6,RawTemp[mV],310.7							
$	23-Nov-2016	02:40:30.251	Anchor	Optode Message	9.540000																											#							
$	23-Nov-2016	02:40:30.251	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:40:30.901	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,52,700,117,539								
$	23-Nov-2016	02:40:31.008	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,47,700,114,539								
$	23-Nov-2016	02:40:31.114	Anchor	FLBB Message	9.540000																																		
$	23-Nov-2016	02:40:31.219	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:40:31.323	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:40:31.428	Anchor	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	02:40:31.533	Anchor	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	02:40:41.686	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:41:31.472	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,114,539								
$	23-Nov-2016	02:41:30.106	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],350.130,AirSaturation[%],120.028,Temperature[Deg.C],18.652,CalPhase[Deg],27.478,TCPhase[Deg],28.299,C1RPh[Deg],36.349,C2RPh[Deg],8.050,C1Amp[mV],512.4,C2Amp[mV],762.3,RawTemp[mV],287.0							
$	23-Nov-2016	02:41:30.109	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	02:41:31.812	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,113,539								
$	23-Nov-2016	02:41:31.917	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	02:41:32.038	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:41:32.142	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:41:32.246	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	02:41:32.353	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	02:41:31.160	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:41:42.686	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:42:32.393	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,112,539								
$	23-Nov-2016	02:42:30.203	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],354.912,AirSaturation[%],121.683,Temperature[Deg.C],18.658,CalPhase[Deg],27.335,TCPhase[Deg],28.156,C1RPh[Deg],36.206,C2RPh[Deg],8.049,C1Amp[mV],508.3,C2Amp[mV],761.7,RawTemp[mV],286.8							
$	23-Nov-2016	02:42:30.206	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	02:42:32.731	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,49,700,111,538								
$	23-Nov-2016	02:42:32.838	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	02:42:32.944	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:42:33.048	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:42:33.152	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	02:42:33.256	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	02:42:32.189	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:42:43.687	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:43:30.096	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:43:30.472	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,49,700,111,538								
$	23-Nov-2016	02:43:30.201	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],357.612,AirSaturation[%],122.623,Temperature[Deg.C],18.664,CalPhase[Deg],27.255,TCPhase[Deg],28.076,C1RPh[Deg],36.125,C2RPh[Deg],8.049,C1Amp[mV],506.0,C2Amp[mV],761.5,RawTemp[mV],286.6							
$	23-Nov-2016	02:43:30.204	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	02:43:30.810	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,114,538								
$	23-Nov-2016	02:43:30.915	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	02:43:31.021	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:43:31.127	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:43:31.231	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	02:43:31.335	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	02:43:41.686	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:44:31.164	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,46,700,116,538								
$	23-Nov-2016	02:44:30.200	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],359.823,AirSaturation[%],123.373,Temperature[Deg.C],18.661,CalPhase[Deg],27.193,TCPhase[Deg],28.014,C1RPh[Deg],36.063,C2RPh[Deg],8.050,C1Amp[mV],504.2,C2Amp[mV],761.4,RawTemp[mV],286.7							
$	23-Nov-2016	02:44:30.203	Anchor	Optode Message	9.540000																											#							
$	23-Nov-2016	02:44:31.504	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,114,538								
$	23-Nov-2016	02:44:31.609	Anchor	FLBB Message	9.540000																																		
$	23-Nov-2016	02:44:31.713	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:44:31.817	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:44:31.923	Anchor	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	02:44:32.029	Anchor	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	02:44:31.096	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:44:42.686	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:45:32.437	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,47,700,116,538								
$	23-Nov-2016	02:45:30.127	Start CP Mode	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],361.437,AirSaturation[%],123.937,Temperature[Deg.C],18.665,CalPhase[Deg],27.145,TCPhase[Deg],27.966,C1RPh[Deg],36.016,C2RPh[Deg],8.050,C1Amp[mV],503.0,C2Amp[mV],761.4,RawTemp[mV],286.6							
$	23-Nov-2016	02:45:30.130	Start CP Mode	Optode Message	9.540000																											#							
$	23-Nov-2016	02:45:32.778	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,54,700,113,538								
$	23-Nov-2016	02:45:32.883	Start CP Mode	FLBB Message	9.540000																																		
$	23-Nov-2016	02:45:32.989	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:45:33.093	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:45:33.200	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	02:45:33.304	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	02:45:31.173	Start CP Mode	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],362.274,AirSaturation[%],124.243,Temperature[Deg.C],18.672,CalPhase[Deg],27.119,TCPhase[Deg],27.940,C1RPh[Deg],35.989,C2RPh[Deg],8.049,C1Amp[mV],502.1,C2Amp[mV],761.2,RawTemp[mV],286.3							
$	23-Nov-2016	02:45:31.176	Start CP Mode	Optode Message	9.540000																											#							
$	23-Nov-2016	02:45:33.641	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,115,538								
$	23-Nov-2016	02:45:33.748	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,48,700,111,538								
$	23-Nov-2016	02:45:33.853	Start CP Mode	FLBB Message	9.540000																																		
$	23-Nov-2016	02:45:33.959	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:45:34.063	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:45:34.169	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	02:45:34.273	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	02:45:32.234	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	02:46:23.545	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:46:29.635	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	23-Nov-2016	02:46:29.636	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	23-Nov-2016	02:46:38.373	DeAnchor	Unparsed SBE41 Message						S>    9.58, 18.6410, 35.1818  																													
$	23-Nov-2016	02:48:51.214	Ascend	FLBB Message	5.250000																										99/99/99,99:99:99,695,48,700,109,539								
$	23-Nov-2016	02:48:51.032	Ascend	Optode Message	5.250000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.741,AirSaturation[%],124.405,Temperature[Deg.C],18.673,CalPhase[Deg],27.105,TCPhase[Deg],27.926,C1RPh[Deg],35.903,C2RPh[Deg],7.977,C1Amp[mV],502.2,C2Amp[mV],757.7,RawTemp[mV],286.3							
$	23-Nov-2016	02:48:51.035	Ascend	Optode Message	5.250000																											#							
$	23-Nov-2016	02:48:51.554	Ascend	FLBB Message	5.250000																										99/99/99,99:99:99,695,44,700,113,539								
$	23-Nov-2016	02:48:51.659	Ascend	FLBB Message	5.250000																																		
$	23-Nov-2016	02:48:51.764	Ascend	FLBB Message	5.250000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:48:51.870	Ascend	FLBB Message	5.250000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:48:51.975	Ascend	FLBB Message	5.250000																										Ave 3								
$	23-Nov-2016	02:48:52.078	Ascend	FLBB Message	5.250000																										Pkt 2								
$	23-Nov-2016	02:49:04.074	Ascend	FLBB Message	3.300000																										99/99/99,99:99:99,695,46,700,114,539								
$	23-Nov-2016	02:49:03.643	Ascend	Optode Message	3.300000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.401,AirSaturation[%],124.928,Temperature[Deg.C],18.655,CalPhase[Deg],27.065,TCPhase[Deg],27.886,C1RPh[Deg],35.936,C2RPh[Deg],8.050,C1Amp[mV],504.9,C2Amp[mV],764.3,RawTemp[mV],286.9							
$	23-Nov-2016	02:49:03.647	Ascend	Optode Message	3.300000																											#							
$	23-Nov-2016	02:49:04.412	Ascend	FLBB Message	3.300000																										99/99/99,99:99:99,695,50,700,112,538								
$	23-Nov-2016	02:49:04.519	Ascend	FLBB Message	3.300000																																		
$	23-Nov-2016	02:49:04.622	Ascend	FLBB Message	3.300000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:49:04.727	Ascend	FLBB Message	3.300000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:49:04.831	Ascend	FLBB Message	3.300000																										Ave 3								
$	23-Nov-2016	02:49:04.937	Ascend	FLBB Message	3.300000																										Pkt 2								
$	23-Nov-2016	02:49:19.842	Ascend	FLBB Message	1.410000																										99/99/99,99:99:99,695,51,700,106,538								
$	23-Nov-2016	02:49:19.728	Ascend	Optode Message	1.410000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.729,AirSaturation[%],125.040,Temperature[Deg.C],18.655,CalPhase[Deg],27.056,TCPhase[Deg],27.877,C1RPh[Deg],35.926,C2RPh[Deg],8.048,C1Amp[mV],507.1,C2Amp[mV],765.7,RawTemp[mV],286.9							
$	23-Nov-2016	02:49:19.728	Ascend	Optode Message	1.410000																											#							
$	23-Nov-2016	02:49:20.180	Ascend	FLBB Message	1.410000																										99/99/99,99:99:99,695,53,700,113,539								
$	23-Nov-2016	02:49:20.285	Ascend	FLBB Message	1.410000																																		
$	23-Nov-2016	02:49:20.389	Ascend	FLBB Message	1.410000																										Ser FLBB2K-3310								
$	23-Nov-2016	02:49:20.501	Ascend	FLBB Message	1.410000																										Ver FLNTU 4.08								
$	23-Nov-2016	02:49:20.606	Ascend	FLBB Message	1.410000																										Ave 3								
$	23-Nov-2016	02:49:20.711	Ascend	FLBB Message	1.410000																										Pkt 2								
$	23-Nov-2016	02:49:21.937	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	23-Nov-2016	02:49:24.733	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	02:49:25.913	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	23-Nov-2016	02:49:25.935	Init Profile	Unparsed SBE41 Message						powering down  																													
$	23-Nov-2016	02:49:54.736	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	23-Nov-2016	02:50:51.200	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	02:50:57.175	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	23-Nov-2016	02:50:57.362	Dump CP Data	Unparsed SBE41 Message						samples = 168, maxPress = 9.59  																													
$	23-Nov-2016	02:51:05.295	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	23-Nov-2016	02:51:05.981	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	23-Nov-2016	02:51:06.268	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	23-Nov-2016	02:51:06.325	Dump CP Data	Unparsed SBE41 Message						  																													
$	23-Nov-2016	02:51:06.344	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	23-Nov-2016	02:51:12.350	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	23-Nov-2016	02:51:12.351	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.451	Dump CP Data	Unparsed SBE41 Message						    2.01,  18.6392,  35.1806, 17  																													
$	23-Nov-2016	02:51:12.452	Dump CP Data	Unparsed SBE41 Message						    3.92,  18.6405,  35.1806, 14  																													
$	23-Nov-2016	02:51:12.495	Dump CP Data	Unparsed SBE41 Message						    6.06,  18.6402,  35.1807, 14  																													
$	23-Nov-2016	02:51:12.544	Dump CP Data	Unparsed SBE41 Message						    8.13,  18.6403,  35.1808, 17  																													
$	23-Nov-2016	02:51:12.604	Dump CP Data	Unparsed SBE41 Message						    9.49,  18.6407,  35.1810, 106  																													
$	23-Nov-2016	02:51:12.649	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.759	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.759	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.865	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.866	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	02:51:12.866	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	02:51:18.481	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	23-Nov-2016	02:51:18.481	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.482	Dump CP Data	Unparsed SBE41 Message						001448CF896D11  																													
$	23-Nov-2016	02:51:18.504	Dump CP Data	Unparsed SBE41 Message						002748D1896D0E  																													
$	23-Nov-2016	02:51:18.530	Dump CP Data	Unparsed SBE41 Message						003D48D0896D0E  																													
$	23-Nov-2016	02:51:18.555	Dump CP Data	Unparsed SBE41 Message						005148D0896D11  																													
$	23-Nov-2016	02:51:18.589	Dump CP Data	Unparsed SBE41 Message						005F48D1896D6A  																													
$	23-Nov-2016	02:51:18.614	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.640	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.665	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.788	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.789	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	02:51:18.789	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	02:51:24.309	Dump CP Data	Unparsed SBE41 Message						S>  																													
