$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	23-Nov-2016	06:39:48.131	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:39:51.983	Surface Ops	Optode Message	1.190000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.749,AirSaturation[%],125.029,Temperature[Deg.C],18.648,CalPhase[Deg],27.058,TCPhase[Deg],27.879,C1RPh[Deg],35.924,C2RPh[Deg],8.044,C1Amp[mV],509.7,C2Amp[mV],767.8,RawTemp[mV],287.1							
$	23-Nov-2016	06:39:51.983	Surface Ops	Optode Message	1.190000																											#							
$	23-Nov-2016	06:39:59.632	Surface Ops	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:43:35.795	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,51,700,322,539								
$	23-Nov-2016	06:43:35.904	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,48,700,278,539								
$	23-Nov-2016	06:43:36.011	Surface Ops	FLBB Message	0.150000																																		
$	23-Nov-2016	06:43:36.116	Surface Ops	FLBB Message	0.150000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:43:36.222	Surface Ops	FLBB Message	0.150000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:43:36.327	Surface Ops	FLBB Message	0.150000																										Ave 3								
$	23-Nov-2016	06:43:36.433	Surface Ops	FLBB Message	0.150000																										Pkt 2								
$	23-Nov-2016	06:43:36.326	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],299.035,AirSaturation[%],101.914,Temperature[Deg.C],18.365,CalPhase[Deg],29.285,TCPhase[Deg],30.106,C1RPh[Deg],38.150,C2RPh[Deg],8.045,C1Amp[mV],576.5,C2Amp[mV],772.2,RawTemp[mV],296.1							
$	23-Nov-2016	06:43:36.326	Surface Ops	Optode Message	0.150000																											#							
$	23-Nov-2016	06:43:36.774	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,54,700,321,539								
$	23-Nov-2016	06:43:36.887	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,46,700,208,539								
$	23-Nov-2016	06:43:36.992	Surface Ops	FLBB Message	0.150000																																		
$	23-Nov-2016	06:43:37.100	Surface Ops	FLBB Message	0.150000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:43:37.206	Surface Ops	FLBB Message	0.150000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:43:37.312	Surface Ops	FLBB Message	0.150000																										Ave 3								
$	23-Nov-2016	06:43:37.416	Surface Ops	FLBB Message	0.150000																										Pkt 2								
$	23-Nov-2016	06:46:54.067	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,46,700,112,539								
$	23-Nov-2016	06:46:53.894	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],298.287,AirSaturation[%],100.335,Temperature[Deg.C],17.728,CalPhase[Deg],29.575,TCPhase[Deg],30.396,C1RPh[Deg],38.432,C2RPh[Deg],8.037,C1Amp[mV],586.1,C2Amp[mV],778.0,RawTemp[mV],316.5							
$	23-Nov-2016	06:46:53.897	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	06:46:54.408	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,113,539								
$	23-Nov-2016	06:46:54.515	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	06:46:54.619	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:46:54.724	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:46:54.828	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	06:46:54.934	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	06:46:55.150	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:47:06.631	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:47:54.970	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,48,700,114,539								
$	23-Nov-2016	06:47:53.901	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],351.650,AirSaturation[%],120.535,Temperature[Deg.C],18.646,CalPhase[Deg],27.435,TCPhase[Deg],28.256,C1RPh[Deg],36.300,C2RPh[Deg],8.043,C1Amp[mV],511.3,C2Amp[mV],762.8,RawTemp[mV],287.2							
$	23-Nov-2016	06:47:53.904	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	06:47:55.310	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,116,539								
$	23-Nov-2016	06:47:55.415	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	06:47:55.519	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:47:55.626	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:47:55.733	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	06:47:55.837	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	06:47:56.068	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:48:07.631	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:48:55.786	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,47,700,112,538								
$	23-Nov-2016	06:48:53.913	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],356.513,AirSaturation[%],122.217,Temperature[Deg.C],18.653,CalPhase[Deg],27.291,TCPhase[Deg],28.112,C1RPh[Deg],36.157,C2RPh[Deg],8.045,C1Amp[mV],507.2,C2Amp[mV],762.2,RawTemp[mV],287.0							
$	23-Nov-2016	06:48:53.916	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	06:48:56.126	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,48,700,118,539								
$	23-Nov-2016	06:48:56.231	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	06:48:56.335	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:48:56.439	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:48:56.543	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	06:48:54.041	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:48:56.862	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	06:49:05.631	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:49:56.494	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,46,700,110,538								
$	23-Nov-2016	06:49:53.900	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],358.245,AirSaturation[%],122.824,Temperature[Deg.C],18.658,CalPhase[Deg],27.239,TCPhase[Deg],28.060,C1RPh[Deg],36.104,C2RPh[Deg],8.044,C1Amp[mV],505.6,C2Amp[mV],761.8,RawTemp[mV],286.8							
$	23-Nov-2016	06:49:53.903	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	06:49:56.832	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,114,539								
$	23-Nov-2016	06:49:56.939	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	06:49:57.043	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:49:57.147	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:49:57.251	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	06:49:57.357	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	06:49:55.040	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:50:06.631	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:50:54.669	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,51,700,117,538								
$	23-Nov-2016	06:50:53.800	Anchor	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],359.769,AirSaturation[%],123.344,Temperature[Deg.C],18.657,CalPhase[Deg],27.196,TCPhase[Deg],28.017,C1RPh[Deg],36.061,C2RPh[Deg],8.044,C1Amp[mV],504.4,C2Amp[mV],761.7,RawTemp[mV],286.8							
$	23-Nov-2016	06:50:53.803	Anchor	Optode Message	9.540000																											#							
$	23-Nov-2016	06:50:55.007	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,116,538								
$	23-Nov-2016	06:50:55.114	Anchor	FLBB Message	9.540000																																		
$	23-Nov-2016	06:50:55.217	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:50:55.323	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:50:55.427	Anchor	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	06:50:55.531	Anchor	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	06:50:56.041	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:51:07.631	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:51:55.818	Start CP Mode	FLBB Message	9.520000																										99/99/99,99:99:99,695,52,700,112,538								
$	23-Nov-2016	06:51:53.935	Start CP Mode	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.725,AirSaturation[%],123.683,Temperature[Deg.C],18.661,CalPhase[Deg],27.167,TCPhase[Deg],27.988,C1RPh[Deg],36.033,C2RPh[Deg],8.045,C1Amp[mV],503.5,C2Amp[mV],761.6,RawTemp[mV],286.7							
$	23-Nov-2016	06:51:53.938	Start CP Mode	Optode Message	9.520000																											#							
$	23-Nov-2016	06:51:56.161	Start CP Mode	FLBB Message	9.520000																										99/99/99,99:99:99,695,48,700,115,538								
$	23-Nov-2016	06:51:56.266	Start CP Mode	FLBB Message	9.520000																																		
$	23-Nov-2016	06:51:54.040	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	06:51:56.584	Start CP Mode	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:51:56.690	Start CP Mode	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:51:56.795	Start CP Mode	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	06:51:56.899	Start CP Mode	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	06:51:54.981	Start CP Mode	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],361.667,AirSaturation[%],124.013,Temperature[Deg.C],18.664,CalPhase[Deg],27.139,TCPhase[Deg],27.960,C1RPh[Deg],36.005,C2RPh[Deg],8.045,C1Amp[mV],502.6,C2Amp[mV],761.6,RawTemp[mV],286.6							
$	23-Nov-2016	06:51:54.984	Start CP Mode	Optode Message	9.520000																											#							
$	23-Nov-2016	06:51:57.238	Start CP Mode	FLBB Message	9.520000																										99/99/99,99:99:99,695,51,700,114,538								
$	23-Nov-2016	06:51:57.343	Start CP Mode	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,117,538								
$	23-Nov-2016	06:51:57.448	Start CP Mode	FLBB Message	9.520000																																		
$	23-Nov-2016	06:51:57.554	Start CP Mode	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:51:57.658	Start CP Mode	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:51:57.765	Start CP Mode	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	06:51:57.869	Start CP Mode	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	06:52:05.631	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:52:10.645	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:52:16.746	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	23-Nov-2016	06:52:16.747	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	23-Nov-2016	06:52:25.094	DeAnchor	Unparsed SBE41 Message						S>    9.46, 18.6336, 35.1814  																													
$	23-Nov-2016	06:54:38.695	Ascend	FLBB Message	4.990000																										99/99/99,99:99:99,695,49,700,113,538								
$	23-Nov-2016	06:54:38.269	Ascend	Optode Message	4.990000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.057,AirSaturation[%],124.148,Temperature[Deg.C],18.664,CalPhase[Deg],27.128,TCPhase[Deg],27.949,C1RPh[Deg],35.981,C2RPh[Deg],8.032,C1Amp[mV],502.7,C2Amp[mV],758.0,RawTemp[mV],286.6							
$	23-Nov-2016	06:54:38.272	Ascend	Optode Message	4.990000																											#							
$	23-Nov-2016	06:54:39.038	Ascend	FLBB Message	4.990000																										99/99/99,99:99:99,695,51,700,110,538								
$	23-Nov-2016	06:54:39.144	Ascend	FLBB Message	4.990000																																		
$	23-Nov-2016	06:54:39.249	Ascend	FLBB Message	4.990000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:54:39.353	Ascend	FLBB Message	4.990000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:54:39.459	Ascend	FLBB Message	4.990000																										Ave 3								
$	23-Nov-2016	06:54:39.563	Ascend	FLBB Message	4.990000																										Pkt 2								
$	23-Nov-2016	06:54:51.447	Ascend	FLBB Message	3.250000																										99/99/99,99:99:99,695,47,700,112,539								
$	23-Nov-2016	06:54:51.552	Ascend	FLBB Message	3.250000																										99/99/99,99:99:99,695,48,700,119,539								
$	23-Nov-2016	06:54:51.658	Ascend	FLBB Message	3.250000																																		
$	23-Nov-2016	06:54:51.764	Ascend	FLBB Message	3.250000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:54:51.870	Ascend	FLBB Message	3.250000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:54:51.974	Ascend	FLBB Message	3.250000																										Ave 3								
$	23-Nov-2016	06:54:52.078	Ascend	FLBB Message	3.250000																										Pkt 2								
$	23-Nov-2016	06:54:51.335	Ascend	Optode Message	3.250000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.084,AirSaturation[%],124.802,Temperature[Deg.C],18.648,CalPhase[Deg],27.077,TCPhase[Deg],27.898,C1RPh[Deg],35.884,C2RPh[Deg],7.986,C1Amp[mV],505.0,C2Amp[mV],764.6,RawTemp[mV],287.1							
$	23-Nov-2016	06:54:51.335	Ascend	Optode Message	3.250000																											#							
$	23-Nov-2016	06:55:07.274	Ascend	FLBB Message	1.480000																										99/99/99,99:99:99,695,47,700,110,538								
$	23-Nov-2016	06:55:07.379	Ascend	FLBB Message	1.480000																										99/99/99,99:99:99,695,49,700,111,538								
$	23-Nov-2016	06:55:07.486	Ascend	FLBB Message	1.480000																																		
$	23-Nov-2016	06:55:07.050	Ascend	Optode Message	1.480000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.611,AirSaturation[%],124.981,Temperature[Deg.C],18.647,CalPhase[Deg],27.062,TCPhase[Deg],27.883,C1RPh[Deg],35.869,C2RPh[Deg],7.986,C1Amp[mV],506.9,C2Amp[mV],765.9,RawTemp[mV],287.1							
$	23-Nov-2016	06:55:07.050	Ascend	Optode Message	1.480000																											#							
$	23-Nov-2016	06:55:07.822	Ascend	FLBB Message	1.480000																										Ser FLBB2K-3310								
$	23-Nov-2016	06:55:07.926	Ascend	FLBB Message	1.480000																										Ver FLNTU 4.08								
$	23-Nov-2016	06:55:08.031	Ascend	FLBB Message	1.480000																										Ave 3								
$	23-Nov-2016	06:55:08.136	Ascend	FLBB Message	1.480000																										Pkt 2								
$	23-Nov-2016	06:55:09.539	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	23-Nov-2016	06:55:12.457	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	06:55:13.639	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	23-Nov-2016	06:55:13.661	Init Profile	Unparsed SBE41 Message						powering down  																													
$	23-Nov-2016	06:55:41.336	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	23-Nov-2016	06:56:38.326	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	06:56:44.297	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	23-Nov-2016	06:56:44.467	Dump CP Data	Unparsed SBE41 Message						samples = 169, maxPress = 9.59  																													
$	23-Nov-2016	06:56:52.416	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	23-Nov-2016	06:56:53.177	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	23-Nov-2016	06:56:53.392	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	23-Nov-2016	06:56:53.524	Dump CP Data	Unparsed SBE41 Message						  																													
$	23-Nov-2016	06:56:53.543	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	23-Nov-2016	06:56:59.475	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	23-Nov-2016	06:56:59.476	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.577	Dump CP Data	Unparsed SBE41 Message						    2.03,  18.6345,  35.1811, 17  																													
$	23-Nov-2016	06:56:59.577	Dump CP Data	Unparsed SBE41 Message						    3.98,  18.6333,  35.1813, 16  																													
$	23-Nov-2016	06:56:59.620	Dump CP Data	Unparsed SBE41 Message						    6.07,  18.6332,  35.1814, 13  																													
$	23-Nov-2016	06:56:59.669	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.6338,  35.1814, 19  																													
$	23-Nov-2016	06:56:59.729	Dump CP Data	Unparsed SBE41 Message						    9.49,  18.6338,  35.1814, 104  																													
$	23-Nov-2016	06:56:59.773	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.886	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.887	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.991	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.992	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	06:56:59.993	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	06:57:05.703	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	23-Nov-2016	06:57:05.704	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:05.704	Dump CP Data	Unparsed SBE41 Message						001448CB896D11  																													
$	23-Nov-2016	06:57:05.726	Dump CP Data	Unparsed SBE41 Message						002848C9896D10  																													
$	23-Nov-2016	06:57:05.752	Dump CP Data	Unparsed SBE41 Message						003D48C9896D0D  																													
$	23-Nov-2016	06:57:05.777	Dump CP Data	Unparsed SBE41 Message						005148CA896D13  																													
$	23-Nov-2016	06:57:05.812	Dump CP Data	Unparsed SBE41 Message						005F48CA896D68  																													
$	23-Nov-2016	06:57:05.837	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:05.864	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:05.888	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:06.012	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:06.013	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	06:57:06.013	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	06:57:11.535	Dump CP Data	Unparsed SBE41 Message						S>  																													
