$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	23-Nov-2016	09:33:21.451	Surface Ops	Optode Message	0.480000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.309,AirSaturation[%],125.576,Temperature[Deg.C],18.652,CalPhase[Deg],27.013,TCPhase[Deg],27.834,C1RPh[Deg],35.875,C2RPh[Deg],8.041,C1Amp[mV],508.0,C2Amp[mV],767.7,RawTemp[mV],287.0							
$	23-Nov-2016	09:33:22.004	Surface Ops	Optode Message	0.480000																											#							
$	23-Nov-2016	09:34:45.672	Surface Ops	FLBB Message	0.140000																										99/99/99,99:99:99,695,48,700,156,539								
$	23-Nov-2016	09:34:45.781	Surface Ops	FLBB Message	0.140000																										99/99/99,99:99:99,695,50,700,177,539								
$	23-Nov-2016	09:34:45.886	Surface Ops	FLBB Message	0.140000																																		
$	23-Nov-2016	09:34:45.993	Surface Ops	FLBB Message	0.140000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:34:46.101	Surface Ops	FLBB Message	0.140000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:34:46.208	Surface Ops	FLBB Message	0.140000																										Ave 3								
$	23-Nov-2016	09:34:46.312	Surface Ops	FLBB Message	0.140000																										Pkt 2								
$	23-Nov-2016	09:34:46.207	Surface Ops	Optode Message	0.120000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.432,AirSaturation[%],102.369,Temperature[Deg.C],18.193,CalPhase[Deg],29.267,TCPhase[Deg],30.088,C1RPh[Deg],38.127,C2RPh[Deg],8.039,C1Amp[mV],575.5,C2Amp[mV],773.6,RawTemp[mV],301.6							
$	23-Nov-2016	09:34:46.207	Surface Ops	Optode Message	0.120000																											#							
$	23-Nov-2016	09:34:46.767	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,51,700,245,539								
$	23-Nov-2016	09:34:46.874	Surface Ops	FLBB Message	0.120000																										99/99/99,99:99:99,695,51,700,271,539								
$	23-Nov-2016	09:34:46.981	Surface Ops	FLBB Message	0.120000																																		
$	23-Nov-2016	09:34:47.084	Surface Ops	FLBB Message	0.120000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:34:47.189	Surface Ops	FLBB Message	0.120000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:34:47.293	Surface Ops	FLBB Message	0.120000																										Ave 3								
$	23-Nov-2016	09:34:47.399	Surface Ops	FLBB Message	0.120000																										Pkt 2								
$	23-Nov-2016	09:37:12.367	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,170,539								
$	23-Nov-2016	09:37:11.915	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],300.190,AirSaturation[%],101.148,Temperature[Deg.C],17.811,CalPhase[Deg],29.469,TCPhase[Deg],30.290,C1RPh[Deg],38.325,C2RPh[Deg],8.035,C1Amp[mV],582.1,C2Amp[mV],776.6,RawTemp[mV],313.8							
$	23-Nov-2016	09:37:11.918	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	09:37:12.706	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,47,700,170,539								
$	23-Nov-2016	09:37:12.812	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	09:37:12.915	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:37:13.021	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:37:13.127	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	09:37:13.231	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	09:37:12.163	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:37:23.636	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:38:13.361	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,48,700,172,539								
$	23-Nov-2016	09:38:12.090	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],352.360,AirSaturation[%],120.783,Temperature[Deg.C],18.648,CalPhase[Deg],27.414,TCPhase[Deg],28.235,C1RPh[Deg],36.274,C2RPh[Deg],8.040,C1Amp[mV],510.8,C2Amp[mV],763.1,RawTemp[mV],287.1							
$	23-Nov-2016	09:38:12.093	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	09:38:13.699	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,46,700,166,539								
$	23-Nov-2016	09:38:13.804	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	09:38:13.908	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:38:14.012	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:38:14.118	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	09:38:14.224	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	09:38:13.157	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:38:24.636	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:39:14.221	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,171,539								
$	23-Nov-2016	09:39:12.038	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],358.347,AirSaturation[%],122.851,Temperature[Deg.C],18.655,CalPhase[Deg],27.237,TCPhase[Deg],28.058,C1RPh[Deg],36.040,C2RPh[Deg],7.982,C1Amp[mV],505.7,C2Amp[mV],762.4,RawTemp[mV],286.9							
$	23-Nov-2016	09:39:12.041	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	09:39:14.559	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,45,700,171,538								
$	23-Nov-2016	09:39:14.665	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	09:39:14.770	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:39:14.876	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:39:14.980	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	09:39:15.084	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	09:39:14.046	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:39:25.636	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:40:12.067	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,168,539								
$	23-Nov-2016	09:40:12.006	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.881,AirSaturation[%],123.732,Temperature[Deg.C],18.660,CalPhase[Deg],27.163,TCPhase[Deg],27.984,C1RPh[Deg],36.025,C2RPh[Deg],8.041,C1Amp[mV],503.6,C2Amp[mV],762.0,RawTemp[mV],286.7							
$	23-Nov-2016	09:40:12.009	Anchor	Optode Message	9.530000																											#							
$	23-Nov-2016	09:40:12.045	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:40:12.625	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,49,700,168,538								
$	23-Nov-2016	09:40:12.730	Anchor	FLBB Message	9.530000																																		
$	23-Nov-2016	09:40:12.834	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:40:12.940	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:40:13.046	Anchor	FLBB Message	9.530000																										Ave 3								
$	23-Nov-2016	09:40:13.150	Anchor	FLBB Message	9.530000																										Pkt 2								
$	23-Nov-2016	09:40:23.636	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:41:12.971	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,172,538								
$	23-Nov-2016	09:41:12.012	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],362.250,AirSaturation[%],124.209,Temperature[Deg.C],18.662,CalPhase[Deg],27.123,TCPhase[Deg],27.944,C1RPh[Deg],35.985,C2RPh[Deg],8.040,C1Amp[mV],502.5,C2Amp[mV],762.0,RawTemp[mV],286.7							
$	23-Nov-2016	09:41:12.016	Anchor	Optode Message	9.520000																											#							
$	23-Nov-2016	09:41:13.316	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,53,700,166,538								
$	23-Nov-2016	09:41:13.423	Anchor	FLBB Message	9.520000																																		
$	23-Nov-2016	09:41:13.527	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:41:13.631	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:41:13.735	Anchor	FLBB Message	9.520000																										Ave 3								
$	23-Nov-2016	09:41:13.841	Anchor	FLBB Message	9.520000																										Pkt 2								
$	23-Nov-2016	09:41:13.200	Anchor	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:41:24.636	Anchor	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:42:14.250	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,49,700,171,538								
$	23-Nov-2016	09:42:12.051	Start CP Mode	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.725,AirSaturation[%],124.389,Temperature[Deg.C],18.669,CalPhase[Deg],27.107,TCPhase[Deg],27.928,C1RPh[Deg],35.968,C2RPh[Deg],8.040,C1Amp[mV],502.0,C2Amp[mV],761.8,RawTemp[mV],286.4							
$	23-Nov-2016	09:42:12.054	Start CP Mode	Optode Message	9.540000																											#							
$	23-Nov-2016	09:42:14.589	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,47,700,171,538								
$	23-Nov-2016	09:42:14.696	Start CP Mode	FLBB Message	9.540000																																		
$	23-Nov-2016	09:42:14.801	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:42:14.906	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:42:15.010	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	09:42:15.114	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	09:42:13.096	Start CP Mode	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.004,AirSaturation[%],124.491,Temperature[Deg.C],18.672,CalPhase[Deg],27.098,TCPhase[Deg],27.919,C1RPh[Deg],35.960,C2RPh[Deg],8.041,C1Amp[mV],501.6,C2Amp[mV],761.7,RawTemp[mV],286.4							
$	23-Nov-2016	09:42:13.100	Start CP Mode	Optode Message	9.540000																											#							
$	23-Nov-2016	09:42:15.451	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,166,538								
$	23-Nov-2016	09:42:15.560	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,166,538								
$	23-Nov-2016	09:42:15.665	Start CP Mode	FLBB Message	9.540000																																		
$	23-Nov-2016	09:42:15.769	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:42:15.873	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:42:15.981	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	23-Nov-2016	09:42:16.085	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	23-Nov-2016	09:42:14.063	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	23-Nov-2016	09:43:05.353	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:43:11.441	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	23-Nov-2016	09:43:11.442	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	23-Nov-2016	09:43:19.760	DeAnchor	Unparsed SBE41 Message						S>    9.57, 18.6385, 35.1820  																													
$	23-Nov-2016	09:45:33.253	Ascend	FLBB Message	5.050000																										99/99/99,99:99:99,695,45,700,147,538								
$	23-Nov-2016	09:45:32.827	Ascend	Optode Message	5.050000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.613,AirSaturation[%],124.701,Temperature[Deg.C],18.672,CalPhase[Deg],27.081,TCPhase[Deg],27.902,C1RPh[Deg],35.929,C2RPh[Deg],8.027,C1Amp[mV],501.8,C2Amp[mV],758.2,RawTemp[mV],286.3							
$	23-Nov-2016	09:45:32.830	Ascend	Optode Message	5.050000																											#							
$	23-Nov-2016	09:45:33.593	Ascend	FLBB Message	5.050000																										99/99/99,99:99:99,695,50,700,136,539								
$	23-Nov-2016	09:45:33.700	Ascend	FLBB Message	5.050000																																		
$	23-Nov-2016	09:45:33.806	Ascend	FLBB Message	5.050000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:45:33.910	Ascend	FLBB Message	5.050000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:45:34.014	Ascend	FLBB Message	5.050000																										Ave 3								
$	23-Nov-2016	09:45:34.118	Ascend	FLBB Message	5.050000																										Pkt 2								
$	23-Nov-2016	09:45:46.208	Ascend	FLBB Message	3.290000																										99/99/99,99:99:99,695,50,700,151,539								
$	23-Nov-2016	09:45:45.537	Ascend	Optode Message	3.290000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.414,AirSaturation[%],125.269,Temperature[Deg.C],18.652,CalPhase[Deg],27.038,TCPhase[Deg],27.859,C1RPh[Deg],35.900,C2RPh[Deg],8.041,C1Amp[mV],503.8,C2Amp[mV],764.6,RawTemp[mV],287.0							
$	23-Nov-2016	09:45:45.537	Ascend	Optode Message	3.290000																											#							
$	23-Nov-2016	09:45:46.549	Ascend	FLBB Message	3.290000																										99/99/99,99:99:99,695,50,700,150,539								
$	23-Nov-2016	09:45:46.656	Ascend	FLBB Message	3.290000																																		
$	23-Nov-2016	09:45:46.759	Ascend	FLBB Message	3.290000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:45:46.863	Ascend	FLBB Message	3.290000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:45:46.991	Ascend	FLBB Message	3.290000																										Ave 3								
$	23-Nov-2016	09:45:47.095	Ascend	FLBB Message	3.290000																										Pkt 2								
$	23-Nov-2016	09:46:01.957	Ascend	FLBB Message	1.530000																										99/99/99,99:99:99,695,48,700,122,538								
$	23-Nov-2016	09:46:02.062	Ascend	Optode Message	1.420000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.835,AirSaturation[%],125.415,Temperature[Deg.C],18.653,CalPhase[Deg],27.026,TCPhase[Deg],27.847,C1RPh[Deg],35.888,C2RPh[Deg],8.041,C1Amp[mV],505.7,C2Amp[mV],765.8,RawTemp[mV],287.0							
$	23-Nov-2016	09:46:02.062	Ascend	Optode Message	1.420000																											#							
$	23-Nov-2016	09:46:02.539	Ascend	FLBB Message	1.420000																										99/99/99,99:99:99,695,46,700,123,538								
$	23-Nov-2016	09:46:02.646	Ascend	FLBB Message	1.420000																																		
$	23-Nov-2016	09:46:02.750	Ascend	FLBB Message	1.420000																										Ser FLBB2K-3310								
$	23-Nov-2016	09:46:02.855	Ascend	FLBB Message	1.420000																										Ver FLNTU 4.08								
$	23-Nov-2016	09:46:02.959	Ascend	FLBB Message	1.420000																										Ave 3								
$	23-Nov-2016	09:46:03.065	Ascend	FLBB Message	1.420000																										Pkt 2								
$	23-Nov-2016	09:46:04.202	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	23-Nov-2016	09:46:07.148	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	23-Nov-2016	09:46:08.328	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	23-Nov-2016	09:46:08.350	Init Profile	Unparsed SBE41 Message						powering down  																													
$	23-Nov-2016	09:46:32.333	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	23-Nov-2016	09:47:29.313	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	23-Nov-2016	09:47:35.383	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	23-Nov-2016	09:47:35.536	Dump CP Data	Unparsed SBE41 Message						samples = 169, maxPress = 9.59  																													
$	23-Nov-2016	09:47:43.486	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	23-Nov-2016	09:47:44.171	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	23-Nov-2016	09:47:44.381	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	23-Nov-2016	09:47:44.519	Dump CP Data	Unparsed SBE41 Message						  																													
$	23-Nov-2016	09:47:44.537	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	23-Nov-2016	09:47:50.565	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	23-Nov-2016	09:47:50.566	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:50.602	Dump CP Data	Unparsed SBE41 Message						    2.05,  18.6374,  35.1816, 17  																													
$	23-Nov-2016	09:47:50.652	Dump CP Data	Unparsed SBE41 Message						    3.93,  18.6385,  35.1817, 15  																													
$	23-Nov-2016	09:47:50.702	Dump CP Data	Unparsed SBE41 Message						    5.98,  18.6378,  35.1818, 13  																													
$	23-Nov-2016	09:47:50.751	Dump CP Data	Unparsed SBE41 Message						    8.08,  18.6387,  35.1818, 19  																													
$	23-Nov-2016	09:47:50.872	Dump CP Data	Unparsed SBE41 Message						    9.49,  18.6384,  35.1819, 105  																													
$	23-Nov-2016	09:47:50.873	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:50.978	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:50.979	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:51.087	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:51.087	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	23-Nov-2016	09:47:51.088	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	09:47:56.694	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	23-Nov-2016	09:47:56.694	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:56.695	Dump CP Data	Unparsed SBE41 Message						001548CD896E11  																													
$	23-Nov-2016	09:47:56.716	Dump CP Data	Unparsed SBE41 Message						002748CE896E0F  																													
$	23-Nov-2016	09:47:56.742	Dump CP Data	Unparsed SBE41 Message						003C48CE896E0D  																													
$	23-Nov-2016	09:47:56.767	Dump CP Data	Unparsed SBE41 Message						005148CF896E13  																													
$	23-Nov-2016	09:47:56.801	Dump CP Data	Unparsed SBE41 Message						005F48CE896E69  																													
$	23-Nov-2016	09:47:56.826	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:56.852	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:56.877	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:57.001	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:57.002	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	23-Nov-2016	09:47:57.002	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	23-Nov-2016	09:48:02.523	Dump CP Data	Unparsed SBE41 Message						S>  																													
