$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	24-Nov-2016	10:34:31.710	Surface Ops	Optode Message	0.220000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.265,AirSaturation[%],125.837,Temperature[Deg.C],18.627,CalPhase[Deg],26.996,TCPhase[Deg],27.817,C1RPh[Deg],35.835,C2RPh[Deg],8.018,C1Amp[mV],500.4,C2Amp[mV],762.5,RawTemp[mV],287.8							
$	24-Nov-2016	10:34:31.710	Surface Ops	Optode Message	0.220000																											#							
$	24-Nov-2016	10:35:56.300	Surface Ops	FLBB Message	0.210000																										99/99/99,99:99:99,695,54,700,266,539								
$	24-Nov-2016	10:35:56.408	Surface Ops	FLBB Message	0.210000																										99/99/99,99:99:99,695,51,700,232,539								
$	24-Nov-2016	10:35:56.514	Surface Ops	FLBB Message	0.210000																																		
$	24-Nov-2016	10:35:56.624	Surface Ops	FLBB Message	0.210000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:35:56.729	Surface Ops	FLBB Message	0.210000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:35:56.833	Surface Ops	FLBB Message	0.210000																										Ave 3								
$	24-Nov-2016	10:35:56.938	Surface Ops	FLBB Message	0.210000																										Pkt 2								
$	24-Nov-2016	10:35:56.623	Surface Ops	Optode Message	0.210000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],305.226,AirSaturation[%],103.677,Temperature[Deg.C],18.202,CalPhase[Deg],29.125,TCPhase[Deg],29.946,C1RPh[Deg],37.962,C2RPh[Deg],8.016,C1Amp[mV],561.5,C2Amp[mV],766.7,RawTemp[mV],301.4							
$	24-Nov-2016	10:35:56.727	Surface Ops	Optode Message	0.210000																											#							
$	24-Nov-2016	10:35:57.279	Surface Ops	FLBB Message	0.210000																										99/99/99,99:99:99,695,52,700,234,539								
$	24-Nov-2016	10:35:57.388	Surface Ops	FLBB Message	0.210000																										99/99/99,99:99:99,695,52,700,208,539								
$	24-Nov-2016	10:35:57.493	Surface Ops	FLBB Message	0.210000																																		
$	24-Nov-2016	10:35:57.597	Surface Ops	FLBB Message	0.210000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:35:57.701	Surface Ops	FLBB Message	0.210000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:35:57.806	Surface Ops	FLBB Message	0.210000																										Ave 3								
$	24-Nov-2016	10:35:57.914	Surface Ops	FLBB Message	0.210000																										Pkt 2								
$	24-Nov-2016	10:38:57.258	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,52,700,141,539								
$	24-Nov-2016	10:38:55.585	Anchor	Optode Message	9.580000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],303.602,AirSaturation[%],102.148,Temperature[Deg.C],17.740,CalPhase[Deg],29.373,TCPhase[Deg],30.194,C1RPh[Deg],38.208,C2RPh[Deg],8.015,C1Amp[mV],569.5,C2Amp[mV],770.6,RawTemp[mV],316.1							
$	24-Nov-2016	10:38:55.589	Anchor	Optode Message	9.580000																											#							
$	24-Nov-2016	10:38:57.598	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,49,700,142,539								
$	24-Nov-2016	10:38:57.703	Anchor	FLBB Message	9.580000																																		
$	24-Nov-2016	10:38:58.011	Anchor	FLBB Message	9.580000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:38:58.117	Anchor	FLBB Message	9.580000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:38:58.226	Anchor	FLBB Message	9.580000																										Ave 3								
$	24-Nov-2016	10:38:58.330	Anchor	FLBB Message	9.580000																										Pkt 2								
$	24-Nov-2016	10:38:58.330	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:39:09.780	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:39:56.745	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,52,700,149,539								
$	24-Nov-2016	10:39:55.587	Anchor	Optode Message	9.590000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],355.160,AirSaturation[%],121.679,Temperature[Deg.C],18.622,CalPhase[Deg],27.342,TCPhase[Deg],28.163,C1RPh[Deg],36.178,C2RPh[Deg],8.015,C1Amp[mV],505.6,C2Amp[mV],760.5,RawTemp[mV],287.9							
$	24-Nov-2016	10:39:55.590	Anchor	Optode Message	9.590000																											#							
$	24-Nov-2016	10:39:57.285	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,51,700,146,538								
$	24-Nov-2016	10:39:57.390	Anchor	FLBB Message	9.590000																																		
$	24-Nov-2016	10:39:57.494	Anchor	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:39:57.598	Anchor	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:39:57.702	Anchor	FLBB Message	9.590000																										Ave 3								
$	24-Nov-2016	10:39:57.810	Anchor	FLBB Message	9.590000																										Pkt 2								
$	24-Nov-2016	10:39:56.246	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:40:07.779	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:40:57.778	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,46,700,142,539								
$	24-Nov-2016	10:40:55.587	Anchor	Optode Message	9.590000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],358.951,AirSaturation[%],123.004,Temperature[Deg.C],18.633,CalPhase[Deg],27.228,TCPhase[Deg],28.049,C1RPh[Deg],36.066,C2RPh[Deg],8.016,C1Amp[mV],502.2,C2Amp[mV],759.9,RawTemp[mV],287.6							
$	24-Nov-2016	10:40:55.590	Anchor	Optode Message	9.590000																											#							
$	24-Nov-2016	10:40:58.319	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,50,700,138,539								
$	24-Nov-2016	10:40:58.424	Anchor	FLBB Message	9.590000																																		
$	24-Nov-2016	10:40:58.530	Anchor	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:40:58.634	Anchor	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:40:58.741	Anchor	FLBB Message	9.590000																										Ave 3								
$	24-Nov-2016	10:40:58.844	Anchor	FLBB Message	9.590000																										Pkt 2								
$	24-Nov-2016	10:40:57.189	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:41:08.779	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:41:55.648	Anchor	Optode Message	9.580000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.609,AirSaturation[%],123.570,Temperature[Deg.C],18.632,CalPhase[Deg],27.181,TCPhase[Deg],28.002,C1RPh[Deg],36.020,C2RPh[Deg],8.017,C1Amp[mV],500.7,C2Amp[mV],759.6,RawTemp[mV],287.6							
$	24-Nov-2016	10:41:55.649	Anchor	Optode Message	9.580000																											#							
$	24-Nov-2016	10:41:56.090	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,50,700,145,538								
$	24-Nov-2016	10:41:56.311	Anchor	FLBB Message	9.600000																										99/99/99,99:99:99,695,50,700,147,539								
$	24-Nov-2016	10:41:56.659	Anchor	FLBB Message	9.600000																																		
$	24-Nov-2016	10:41:56.762	Anchor	FLBB Message	9.600000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:41:56.870	Anchor	FLBB Message	9.600000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:41:56.974	Anchor	FLBB Message	9.600000																										Ave 3								
$	24-Nov-2016	10:41:57.078	Anchor	FLBB Message	9.600000																										Pkt 2								
$	24-Nov-2016	10:41:58.189	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:42:09.780	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:42:56.464	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,50,700,141,538								
$	24-Nov-2016	10:42:55.610	Anchor	Optode Message	9.580000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],361.903,AirSaturation[%],124.031,Temperature[Deg.C],18.639,CalPhase[Deg],27.142,TCPhase[Deg],27.963,C1RPh[Deg],35.981,C2RPh[Deg],8.018,C1Amp[mV],499.7,C2Amp[mV],759.4,RawTemp[mV],287.4							
$	24-Nov-2016	10:42:55.613	Anchor	Optode Message	9.580000																											#							
$	24-Nov-2016	10:42:57.006	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,52,700,148,538								
$	24-Nov-2016	10:42:57.111	Anchor	FLBB Message	9.580000																																		
$	24-Nov-2016	10:42:57.215	Anchor	FLBB Message	9.580000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:42:57.321	Anchor	FLBB Message	9.580000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:42:57.425	Anchor	FLBB Message	9.580000																										Ave 3								
$	24-Nov-2016	10:42:57.531	Anchor	FLBB Message	9.580000																										Pkt 2								
$	24-Nov-2016	10:42:56.267	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:43:07.779	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:43:57.710	Start CP Mode	FLBB Message	9.600000																										99/99/99,99:99:99,695,49,700,141,538								
$	24-Nov-2016	10:43:55.588	Start CP Mode	Optode Message	9.600000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.663,AirSaturation[%],124.314,Temperature[Deg.C],18.648,CalPhase[Deg],27.117,TCPhase[Deg],27.938,C1RPh[Deg],35.955,C2RPh[Deg],8.017,C1Amp[mV],499.0,C2Amp[mV],759.3,RawTemp[mV],287.1							
$	24-Nov-2016	10:43:55.591	Start CP Mode	Optode Message	9.600000																											#							
$	24-Nov-2016	10:43:58.049	Start CP Mode	FLBB Message	9.600000																										99/99/99,99:99:99,695,46,700,146,538								
$	24-Nov-2016	10:43:58.154	Start CP Mode	FLBB Message	9.600000																																		
$	24-Nov-2016	10:43:58.260	Start CP Mode	FLBB Message	9.600000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:43:58.366	Start CP Mode	FLBB Message	9.600000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:43:58.470	Start CP Mode	FLBB Message	9.600000																										Ave 3								
$	24-Nov-2016	10:43:58.575	Start CP Mode	FLBB Message	9.600000																										Pkt 2								
$	24-Nov-2016	10:43:58.883	Start CP Mode	FLBB Message	9.600000																										99/99/99,99:99:99,695,50,700,144,538								
$	24-Nov-2016	10:43:56.728	Start CP Mode	Optode Message	9.600000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.235,AirSaturation[%],124.504,Temperature[Deg.C],18.645,CalPhase[Deg],27.102,TCPhase[Deg],27.923,C1RPh[Deg],35.940,C2RPh[Deg],8.017,C1Amp[mV],498.5,C2Amp[mV],759.2,RawTemp[mV],287.2							
$	24-Nov-2016	10:43:56.732	Start CP Mode	Optode Message	9.600000																											#							
$	24-Nov-2016	10:43:59.222	Start CP Mode	FLBB Message	9.600000																										99/99/99,99:99:99,695,47,700,140,538								
$	24-Nov-2016	10:43:59.329	Start CP Mode	FLBB Message	9.600000																																		
$	24-Nov-2016	10:43:59.433	Start CP Mode	FLBB Message	9.600000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:43:59.538	Start CP Mode	FLBB Message	9.600000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:43:59.644	Start CP Mode	FLBB Message	9.600000																										Ave 3								
$	24-Nov-2016	10:43:59.748	Start CP Mode	FLBB Message	9.600000																										Pkt 2								
$	24-Nov-2016	10:43:57.814	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	10:44:49.119	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:44:55.201	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	24-Nov-2016	10:44:55.202	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	24-Nov-2016	10:45:04.205	DeAnchor	Unparsed SBE41 Message						S>    9.52, 18.6114, 35.1854  																													
$	24-Nov-2016	10:47:01.928	Ascend	FLBB Message	5.410000																										99/99/99,99:99:99,695,46,700,129,539								
$	24-Nov-2016	10:47:01.716	Ascend	Optode Message	5.410000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.594,AirSaturation[%],124.628,Temperature[Deg.C],18.646,CalPhase[Deg],27.091,TCPhase[Deg],27.912,C1RPh[Deg],35.923,C2RPh[Deg],8.010,C1Amp[mV],498.6,C2Amp[mV],755.9,RawTemp[mV],287.2							
$	24-Nov-2016	10:47:01.812	Ascend	Optode Message	5.410000																											#							
$	24-Nov-2016	10:47:02.471	Ascend	FLBB Message	5.410000																										99/99/99,99:99:99,695,44,700,131,539								
$	24-Nov-2016	10:47:02.576	Ascend	FLBB Message	5.410000																																		
$	24-Nov-2016	10:47:02.686	Ascend	FLBB Message	5.410000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:47:02.790	Ascend	FLBB Message	5.410000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:47:02.894	Ascend	FLBB Message	5.410000																										Ave 3								
$	24-Nov-2016	10:47:03.000	Ascend	FLBB Message	5.410000																										Pkt 2								
$	24-Nov-2016	10:47:17.737	Ascend	FLBB Message	3.260000																										99/99/99,99:99:99,695,50,700,124,538								
$	24-Nov-2016	10:47:17.623	Ascend	Optode Message	3.260000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.584,AirSaturation[%],125.604,Temperature[Deg.C],18.627,CalPhase[Deg],27.015,TCPhase[Deg],27.836,C1RPh[Deg],35.854,C2RPh[Deg],8.018,C1Amp[mV],498.4,C2Amp[mV],760.8,RawTemp[mV],287.8							
$	24-Nov-2016	10:47:17.623	Ascend	Optode Message	3.260000																											#							
$	24-Nov-2016	10:47:18.085	Ascend	FLBB Message	3.260000																										99/99/99,99:99:99,695,50,700,127,539								
$	24-Nov-2016	10:47:18.193	Ascend	FLBB Message	3.260000																																		
$	24-Nov-2016	10:47:18.297	Ascend	FLBB Message	3.260000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:47:18.401	Ascend	FLBB Message	3.260000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:47:18.505	Ascend	FLBB Message	3.260000																										Ave 3								
$	24-Nov-2016	10:47:18.611	Ascend	FLBB Message	3.260000																										Pkt 2								
$	24-Nov-2016	10:47:36.417	Ascend	FLBB Message	1.360000																										99/99/99,99:99:99,695,47,700,124,538								
$	24-Nov-2016	10:47:36.525	Ascend	FLBB Message	1.360000																										99/99/99,99:99:99,695,50,700,127,538								
$	24-Nov-2016	10:47:36.631	Ascend	FLBB Message	1.360000																																		
$	24-Nov-2016	10:47:36.735	Ascend	FLBB Message	1.360000																										Ser FLBB2K-3310								
$	24-Nov-2016	10:47:36.839	Ascend	FLBB Message	1.360000																										Ver FLNTU 4.08								
$	24-Nov-2016	10:47:36.947	Ascend	FLBB Message	1.360000																										Ave 3								
$	24-Nov-2016	10:47:37.051	Ascend	FLBB Message	1.360000																										Pkt 2								
$	24-Nov-2016	10:47:36.305	Ascend	Optode Message	1.360000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.991,AirSaturation[%],125.743,Temperature[Deg.C],18.627,CalPhase[Deg],27.004,TCPhase[Deg],27.825,C1RPh[Deg],35.841,C2RPh[Deg],8.017,C1Amp[mV],499.2,C2Amp[mV],761.6,RawTemp[mV],287.8							
$	24-Nov-2016	10:47:36.305	Ascend	Optode Message	1.360000																											#							
$	24-Nov-2016	10:47:38.880	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	24-Nov-2016	10:47:41.611	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	10:47:42.791	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	24-Nov-2016	10:47:42.813	Init Profile	Unparsed SBE41 Message						powering down  																													
$	24-Nov-2016	10:48:07.491	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	24-Nov-2016	10:49:07.158	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	10:49:13.232	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	24-Nov-2016	10:49:13.427	Dump CP Data	Unparsed SBE41 Message						samples = 159, maxPress = 9.65  																													
$	24-Nov-2016	10:49:21.336	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	24-Nov-2016	10:49:22.027	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	24-Nov-2016	10:49:22.228	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	24-Nov-2016	10:49:22.323	Dump CP Data	Unparsed SBE41 Message						  																													
$	24-Nov-2016	10:49:22.341	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	24-Nov-2016	10:49:28.411	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	24-Nov-2016	10:49:28.411	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.448	Dump CP Data	Unparsed SBE41 Message						    2.05,  18.6121,  35.1850, 21  																													
$	24-Nov-2016	10:49:28.498	Dump CP Data	Unparsed SBE41 Message						    3.97,  18.6121,  35.1852, 14  																													
$	24-Nov-2016	10:49:28.547	Dump CP Data	Unparsed SBE41 Message						    5.98,  18.6118,  35.1852, 15  																													
$	24-Nov-2016	10:49:28.597	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.6111,  35.1853, 17  																													
$	24-Nov-2016	10:49:28.718	Dump CP Data	Unparsed SBE41 Message						    9.53,  18.6114,  35.1853, 92  																													
$	24-Nov-2016	10:49:28.718	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.826	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.827	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.932	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.933	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	10:49:28.934	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	10:49:34.442	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	24-Nov-2016	10:49:34.442	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.443	Dump CP Data	Unparsed SBE41 Message						001548B4897115  																													
$	24-Nov-2016	10:49:34.465	Dump CP Data	Unparsed SBE41 Message						002848B489710E  																													
$	24-Nov-2016	10:49:34.491	Dump CP Data	Unparsed SBE41 Message						003C48B489710F  																													
$	24-Nov-2016	10:49:34.516	Dump CP Data	Unparsed SBE41 Message						005148B3897111  																													
$	24-Nov-2016	10:49:34.550	Dump CP Data	Unparsed SBE41 Message						005F48B389715C  																													
$	24-Nov-2016	10:49:34.575	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.601	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.626	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.749	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.750	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	10:49:34.750	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	10:49:40.297	Dump CP Data	Unparsed SBE41 Message						S>  																													
