$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	24-Nov-2016	13:22:39.549	Surface Ops	Optode Message	0.210000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],368.131,AirSaturation[%],126.129,Temperature[Deg.C],18.625,CalPhase[Deg],26.973,TCPhase[Deg],27.794,C1RPh[Deg],35.809,C2RPh[Deg],8.015,C1Amp[mV],498.9,C2Amp[mV],761.9,RawTemp[mV],287.9							
$	24-Nov-2016	13:22:39.549	Surface Ops	Optode Message	0.210000																											#							
$	24-Nov-2016	13:24:04.326	Surface Ops	FLBB Message	0.240000																										99/99/99,99:99:99,695,47,700,109,539								
$	24-Nov-2016	13:24:04.433	Surface Ops	FLBB Message	0.240000																										99/99/99,99:99:99,695,52,700,108,538								
$	24-Nov-2016	13:24:04.541	Surface Ops	FLBB Message	0.240000																																		
$	24-Nov-2016	13:24:04.649	Surface Ops	FLBB Message	0.240000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:24:04.754	Surface Ops	FLBB Message	0.240000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:24:04.858	Surface Ops	FLBB Message	0.240000																										Ave 3								
$	24-Nov-2016	13:24:04.963	Surface Ops	FLBB Message	0.240000																										Pkt 2								
$	24-Nov-2016	13:24:04.648	Surface Ops	Optode Message	0.240000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.031,AirSaturation[%],101.732,Temperature[Deg.C],17.954,CalPhase[Deg],29.379,TCPhase[Deg],30.200,C1RPh[Deg],38.214,C2RPh[Deg],8.014,C1Amp[mV],567.6,C2Amp[mV],766.2,RawTemp[mV],309.3							
$	24-Nov-2016	13:24:04.753	Surface Ops	Optode Message	0.240000																											#							
$	24-Nov-2016	13:24:05.303	Surface Ops	FLBB Message	0.240000																										99/99/99,99:99:99,695,49,700,114,539								
$	24-Nov-2016	13:24:05.412	Surface Ops	FLBB Message	0.240000																										99/99/99,99:99:99,695,51,700,108,539								
$	24-Nov-2016	13:24:05.518	Surface Ops	FLBB Message	0.240000																																		
$	24-Nov-2016	13:24:05.622	Surface Ops	FLBB Message	0.240000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:24:05.726	Surface Ops	FLBB Message	0.240000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:24:05.831	Surface Ops	FLBB Message	0.240000																										Ave 3								
$	24-Nov-2016	13:24:05.942	Surface Ops	FLBB Message	0.240000																										Pkt 2								
$	24-Nov-2016	13:27:04.167	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,46,700,112,539								
$	24-Nov-2016	13:27:03.959	Anchor	Optode Message	9.560000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.327,AirSaturation[%],101.100,Temperature[Deg.C],17.605,CalPhase[Deg],29.512,TCPhase[Deg],30.333,C1RPh[Deg],38.344,C2RPh[Deg],8.010,C1Amp[mV],572.1,C2Amp[mV],770.0,RawTemp[mV],320.4							
$	24-Nov-2016	13:27:03.961	Anchor	Optode Message	9.560000																											#							
$	24-Nov-2016	13:27:04.508	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,49,700,112,539								
$	24-Nov-2016	13:27:04.613	Anchor	FLBB Message	9.560000																																		
$	24-Nov-2016	13:27:04.718	Anchor	FLBB Message	9.560000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:27:04.824	Anchor	FLBB Message	9.560000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:27:04.929	Anchor	FLBB Message	9.560000																										Ave 3								
$	24-Nov-2016	13:27:05.033	Anchor	FLBB Message	9.560000																										Pkt 2								
$	24-Nov-2016	13:27:05.906	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:27:17.496	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:28:04.948	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,50,700,109,539								
$	24-Nov-2016	13:28:03.895	Anchor	Optode Message	9.580000																											MEASUREMENT,4330,2592,O2Concentration[uM],354.728,AirSaturation[%],121.527,Temperature[Deg.C],18.621,CalPhase[Deg],27.355,TCPhase[Deg],28.176,C1RPh[Deg],36.188,C2RPh[Deg],8.012,C1Amp[mV],505.3,C2Amp[mV],760.4,RawTemp[mV],288.0							
$	24-Nov-2016	13:28:03.898	Anchor	Optode Message	9.580000																											#							
$	24-Nov-2016	13:28:03.906	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:28:05.497	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,46,700,115,539								
$	24-Nov-2016	13:28:05.604	Anchor	FLBB Message	9.580000																																		
$	24-Nov-2016	13:28:05.708	Anchor	FLBB Message	9.580000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:28:05.812	Anchor	FLBB Message	9.580000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:28:05.918	Anchor	FLBB Message	9.580000																										Ave 3								
$	24-Nov-2016	13:28:06.022	Anchor	FLBB Message	9.580000																										Pkt 2								
$	24-Nov-2016	13:28:15.496	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:29:05.882	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,51,700,112,539								
$	24-Nov-2016	13:29:03.891	Anchor	Optode Message	9.580000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.812,AirSaturation[%],123.633,Temperature[Deg.C],18.630,CalPhase[Deg],27.176,TCPhase[Deg],27.997,C1RPh[Deg],36.013,C2RPh[Deg],8.016,C1Amp[mV],500.2,C2Amp[mV],759.6,RawTemp[mV],287.7							
$	24-Nov-2016	13:29:03.894	Anchor	Optode Message	9.580000																											#							
$	24-Nov-2016	13:29:06.220	Anchor	FLBB Message	9.580000																										99/99/99,99:99:99,695,45,700,111,539								
$	24-Nov-2016	13:29:06.325	Anchor	FLBB Message	9.580000																																		
$	24-Nov-2016	13:29:06.433	Anchor	FLBB Message	9.580000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:29:06.537	Anchor	FLBB Message	9.580000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:29:06.641	Anchor	FLBB Message	9.580000																										Ave 3								
$	24-Nov-2016	13:29:06.745	Anchor	FLBB Message	9.580000																										Pkt 2								
$	24-Nov-2016	13:29:04.904	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:29:16.496	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:30:04.061	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,51,700,112,538								
$	24-Nov-2016	13:30:03.888	Anchor	Optode Message	9.560000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.626,AirSaturation[%],124.610,Temperature[Deg.C],18.635,CalPhase[Deg],27.095,TCPhase[Deg],27.916,C1RPh[Deg],35.931,C2RPh[Deg],8.015,C1Amp[mV],497.9,C2Amp[mV],759.2,RawTemp[mV],287.5							
$	24-Nov-2016	13:30:03.889	Anchor	Optode Message	9.560000																											#							
$	24-Nov-2016	13:30:04.402	Anchor	FLBB Message	9.560000																										99/99/99,99:99:99,695,46,700,117,538								
$	24-Nov-2016	13:30:04.507	Anchor	FLBB Message	9.560000																																		
$	24-Nov-2016	13:30:04.610	Anchor	FLBB Message	9.560000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:30:04.716	Anchor	FLBB Message	9.560000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:30:04.823	Anchor	FLBB Message	9.560000																										Ave 3								
$	24-Nov-2016	13:30:04.927	Anchor	FLBB Message	9.560000																										Pkt 2								
$	24-Nov-2016	13:30:05.905	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:30:17.600	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:31:04.645	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,49,700,112,538								
$	24-Nov-2016	13:31:03.890	Anchor	Optode Message	9.590000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.318,AirSaturation[%],124.856,Temperature[Deg.C],18.638,CalPhase[Deg],27.074,TCPhase[Deg],27.895,C1RPh[Deg],35.910,C2RPh[Deg],8.015,C1Amp[mV],497.3,C2Amp[mV],759.0,RawTemp[mV],287.4							
$	24-Nov-2016	13:31:03.894	Anchor	Optode Message	9.590000																											#							
$	24-Nov-2016	13:31:03.905	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:31:05.196	Anchor	FLBB Message	9.590000																										99/99/99,99:99:99,695,48,700,110,538								
$	24-Nov-2016	13:31:05.301	Anchor	FLBB Message	9.590000																																		
$	24-Nov-2016	13:31:05.405	Anchor	FLBB Message	9.590000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:31:05.511	Anchor	FLBB Message	9.590000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:31:05.617	Anchor	FLBB Message	9.590000																										Ave 3								
$	24-Nov-2016	13:31:05.721	Anchor	FLBB Message	9.590000																										Pkt 2								
$	24-Nov-2016	13:31:15.495	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:32:05.813	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,46,700,114,538								
$	24-Nov-2016	13:32:03.936	Start CP Mode	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.107,AirSaturation[%],125.128,Temperature[Deg.C],18.639,CalPhase[Deg],27.052,TCPhase[Deg],27.873,C1RPh[Deg],35.889,C2RPh[Deg],8.016,C1Amp[mV],496.7,C2Amp[mV],759.0,RawTemp[mV],287.4							
$	24-Nov-2016	13:32:03.938	Start CP Mode	Optode Message	9.570000																											#							
$	24-Nov-2016	13:32:06.159	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,53,700,111,539								
$	24-Nov-2016	13:32:06.265	Start CP Mode	FLBB Message	9.570000																																		
$	24-Nov-2016	13:32:06.370	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:32:06.475	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:32:06.579	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	24-Nov-2016	13:32:06.685	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	24-Nov-2016	13:32:06.791	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,50,700,111,538								
$	24-Nov-2016	13:32:04.981	Start CP Mode	Optode Message	9.570000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.611,AirSaturation[%],125.309,Temperature[Deg.C],18.642,CalPhase[Deg],27.036,TCPhase[Deg],27.857,C1RPh[Deg],35.875,C2RPh[Deg],8.017,C1Amp[mV],496.2,C2Amp[mV],758.8,RawTemp[mV],287.3							
$	24-Nov-2016	13:32:04.984	Start CP Mode	Optode Message	9.570000																											#							
$	24-Nov-2016	13:32:07.130	Start CP Mode	FLBB Message	9.570000																										99/99/99,99:99:99,695,48,700,109,538								
$	24-Nov-2016	13:32:07.235	Start CP Mode	FLBB Message	9.570000																																		
$	24-Nov-2016	13:32:07.339	Start CP Mode	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:32:07.445	Start CP Mode	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:32:07.555	Start CP Mode	FLBB Message	9.570000																										Ave 3								
$	24-Nov-2016	13:32:07.659	Start CP Mode	FLBB Message	9.570000																										Pkt 2								
$	24-Nov-2016	13:32:06.043	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	13:32:57.124	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:33:03.214	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	24-Nov-2016	13:33:03.214	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	24-Nov-2016	13:33:12.139	DeAnchor	Unparsed SBE41 Message						S>    9.55, 18.6130, 35.1856  																													
$	24-Nov-2016	13:35:25.231	Ascend	FLBB Message	5.190000																										99/99/99,99:99:99,695,47,700,113,539								
$	24-Nov-2016	13:35:24.803	Ascend	Optode Message	5.190000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.981,AirSaturation[%],125.437,Temperature[Deg.C],18.642,CalPhase[Deg],27.026,TCPhase[Deg],27.847,C1RPh[Deg],35.850,C2RPh[Deg],8.003,C1Amp[mV],496.3,C2Amp[mV],755.2,RawTemp[mV],287.3							
$	24-Nov-2016	13:35:24.803	Ascend	Optode Message	5.190000																											#							
$	24-Nov-2016	13:35:25.575	Ascend	FLBB Message	5.190000																										99/99/99,99:99:99,695,48,700,112,539								
$	24-Nov-2016	13:35:25.680	Ascend	FLBB Message	5.190000																																		
$	24-Nov-2016	13:35:25.786	Ascend	FLBB Message	5.190000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:35:25.890	Ascend	FLBB Message	5.190000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:35:25.997	Ascend	FLBB Message	5.190000																										Ave 3								
$	24-Nov-2016	13:35:26.101	Ascend	FLBB Message	5.190000																										Pkt 2								
$	24-Nov-2016	13:35:38.433	Ascend	FLBB Message	3.100000																										99/99/99,99:99:99,695,48,700,108,539								
$	24-Nov-2016	13:35:37.765	Ascend	Optode Message	3.100000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.259,AirSaturation[%],125.835,Temperature[Deg.C],18.627,CalPhase[Deg],26.996,TCPhase[Deg],27.817,C1RPh[Deg],35.834,C2RPh[Deg],8.017,C1Amp[mV],496.9,C2Amp[mV],760.3,RawTemp[mV],287.8							
$	24-Nov-2016	13:35:37.768	Ascend	Optode Message	3.100000																											#							
$	24-Nov-2016	13:35:38.779	Ascend	FLBB Message	3.100000																										99/99/99,99:99:99,695,48,700,113,539								
$	24-Nov-2016	13:35:38.886	Ascend	FLBB Message	3.100000																																		
$	24-Nov-2016	13:35:38.990	Ascend	FLBB Message	3.100000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:35:39.096	Ascend	FLBB Message	3.100000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:35:39.200	Ascend	FLBB Message	3.100000																										Ave 3								
$	24-Nov-2016	13:35:39.304	Ascend	FLBB Message	3.100000																										Pkt 2								
$	24-Nov-2016	13:35:51.383	Ascend	FLBB Message	1.670000																										99/99/99,99:99:99,695,46,700,112,538								
$	24-Nov-2016	13:35:51.267	Ascend	Optode Message	1.670000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.637,AirSaturation[%],125.956,Temperature[Deg.C],18.623,CalPhase[Deg],26.987,TCPhase[Deg],27.808,C1RPh[Deg],35.825,C2RPh[Deg],8.017,C1Amp[mV],497.8,C2Amp[mV],760.9,RawTemp[mV],287.9							
$	24-Nov-2016	13:35:51.267	Ascend	Optode Message	1.670000																											#							
$	24-Nov-2016	13:35:51.722	Ascend	FLBB Message	1.670000																										99/99/99,99:99:99,695,47,700,112,539								
$	24-Nov-2016	13:35:51.827	Ascend	FLBB Message	1.670000																																		
$	24-Nov-2016	13:35:51.931	Ascend	FLBB Message	1.670000																										Ser FLBB2K-3310								
$	24-Nov-2016	13:35:52.035	Ascend	FLBB Message	1.670000																										Ver FLNTU 4.08								
$	24-Nov-2016	13:35:52.141	Ascend	FLBB Message	1.670000																										Ave 3								
$	24-Nov-2016	13:35:52.247	Ascend	FLBB Message	1.670000																										Pkt 2								
$	24-Nov-2016	13:35:59.599	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	24-Nov-2016	13:36:00.268	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	13:36:02.659	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	13:36:03.841	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	24-Nov-2016	13:36:03.863	Init Profile	Unparsed SBE41 Message						powering down  																													
$	24-Nov-2016	13:36:28.802	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	24-Nov-2016	13:37:28.029	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	13:37:34.105	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	24-Nov-2016	13:37:34.260	Dump CP Data	Unparsed SBE41 Message						samples = 172, maxPress = 9.64  																													
$	24-Nov-2016	13:37:42.209	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	24-Nov-2016	13:37:42.895	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	24-Nov-2016	13:37:43.208	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	24-Nov-2016	13:37:43.314	Dump CP Data	Unparsed SBE41 Message						  																													
$	24-Nov-2016	13:37:43.314	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	24-Nov-2016	13:37:49.289	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	24-Nov-2016	13:37:49.290	Dump CP Data	Unparsed SBE41 Message						    0.85,  18.6060,  35.1856, 2  																													
$	24-Nov-2016	13:37:49.327	Dump CP Data	Unparsed SBE41 Message						    1.91,  18.6083,  35.1857, 20  																													
$	24-Nov-2016	13:37:49.378	Dump CP Data	Unparsed SBE41 Message						    3.95,  18.6093,  35.1856, 13  																													
$	24-Nov-2016	13:37:49.427	Dump CP Data	Unparsed SBE41 Message						    6.00,  18.6118,  35.1856, 13  																													
$	24-Nov-2016	13:37:49.477	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.6125,  35.1857, 18  																													
$	24-Nov-2016	13:37:49.596	Dump CP Data	Unparsed SBE41 Message						    9.53,  18.6129,  35.1857, 106  																													
$	24-Nov-2016	13:37:49.596	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	13:37:49.702	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	13:37:49.703	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	13:37:49.811	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	13:37:49.811	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	13:37:49.812	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	13:37:55.415	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	24-Nov-2016	13:37:55.415	Dump CP Data	Unparsed SBE41 Message						000948AE897202  																													
$	24-Nov-2016	13:37:55.416	Dump CP Data	Unparsed SBE41 Message						001348B0897214  																													
$	24-Nov-2016	13:37:55.438	Dump CP Data	Unparsed SBE41 Message						002748B189720D  																													
$	24-Nov-2016	13:37:55.464	Dump CP Data	Unparsed SBE41 Message						003C48B489720D  																													
$	24-Nov-2016	13:37:55.489	Dump CP Data	Unparsed SBE41 Message						005148B4897212  																													
$	24-Nov-2016	13:37:55.523	Dump CP Data	Unparsed SBE41 Message						005F48B589726A  																													
$	24-Nov-2016	13:37:55.548	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	13:37:55.574	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	13:37:55.599	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	13:37:55.722	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	13:37:55.723	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	13:37:55.723	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	13:38:01.246	Dump CP Data	Unparsed SBE41 Message						S>  																													
