$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	24-Nov-2016	19:11:03.457	Surface Ops	Optode Message	0.190000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],368.011,AirSaturation[%],126.012,Temperature[Deg.C],18.595,CalPhase[Deg],26.987,TCPhase[Deg],27.808,C1RPh[Deg],35.820,C2RPh[Deg],8.012,C1Amp[mV],496.8,C2Amp[mV],760.1,RawTemp[mV],288.8							
$	24-Nov-2016	19:11:03.458	Surface Ops	Optode Message	0.190000																											#							
$	24-Nov-2016	19:12:06.781	Surface Ops	FLBB Message	-0.030000																										99/99/99,99:99:99,695,52,700,113,539								
$	24-Nov-2016	19:12:06.891	Surface Ops	FLBB Message	-0.030000																										99/99/99,99:99:99,695,52,700,110,539								
$	24-Nov-2016	19:12:06.999	Surface Ops	FLBB Message	-0.030000																																		
$	24-Nov-2016	19:12:07.107	Surface Ops	FLBB Message	-0.030000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:12:07.211	Surface Ops	FLBB Message	-0.030000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:12:07.315	Surface Ops	FLBB Message	-0.030000																										Ave 3								
$	24-Nov-2016	19:12:07.420	Surface Ops	FLBB Message	-0.030000																										Pkt 2								
$	24-Nov-2016	19:12:07.105	Surface Ops	Optode Message	-0.030000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.503,AirSaturation[%],101.722,Temperature[Deg.C],17.713,CalPhase[Deg],29.424,TCPhase[Deg],30.245,C1RPh[Deg],38.256,C2RPh[Deg],8.010,C1Amp[mV],565.9,C2Amp[mV],764.1,RawTemp[mV],317.0							
$	24-Nov-2016	19:12:07.105	Surface Ops	Optode Message	-0.030000																											#							
$	24-Nov-2016	19:12:07.760	Surface Ops	FLBB Message	-0.030000																										99/99/99,99:99:99,695,51,700,112,539								
$	24-Nov-2016	19:12:07.869	Surface Ops	FLBB Message	-0.030000																										99/99/99,99:99:99,695,50,700,108,539								
$	24-Nov-2016	19:12:07.974	Surface Ops	FLBB Message	-0.030000																																		
$	24-Nov-2016	19:12:08.078	Surface Ops	FLBB Message	-0.030000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:12:08.183	Surface Ops	FLBB Message	-0.030000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:12:08.289	Surface Ops	FLBB Message	-0.030000																										Ave 3								
$	24-Nov-2016	19:12:08.394	Surface Ops	FLBB Message	-0.030000																										Pkt 2								
$	24-Nov-2016	19:14:38.512	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,48,700,108,539								
$	24-Nov-2016	19:14:35.635	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],303.137,AirSaturation[%],101.172,Temperature[Deg.C],17.350,CalPhase[Deg],29.551,TCPhase[Deg],30.372,C1RPh[Deg],38.378,C2RPh[Deg],8.006,C1Amp[mV],570.1,C2Amp[mV],768.0,RawTemp[mV],328.5							
$	24-Nov-2016	19:14:35.638	Anchor	Optode Message	9.550000																											#							
$	24-Nov-2016	19:14:38.897	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,51,700,118,539								
$	24-Nov-2016	19:14:39.002	Anchor	FLBB Message	9.550000																																		
$	24-Nov-2016	19:14:39.106	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:14:39.212	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:14:39.316	Anchor	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:14:39.422	Anchor	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:14:36.376	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:14:47.967	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:15:36.506	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,51,700,113,539								
$	24-Nov-2016	19:15:35.636	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],354.345,AirSaturation[%],121.328,Temperature[Deg.C],18.593,CalPhase[Deg],27.376,TCPhase[Deg],28.197,C1RPh[Deg],36.205,C2RPh[Deg],8.008,C1Amp[mV],504.4,C2Amp[mV],759.5,RawTemp[mV],288.9							
$	24-Nov-2016	19:15:35.640	Anchor	Optode Message	9.540000																											#							
$	24-Nov-2016	19:15:36.844	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,52,700,113,539								
$	24-Nov-2016	19:15:36.949	Anchor	FLBB Message	9.540000																																		
$	24-Nov-2016	19:15:37.055	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:15:37.159	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:15:37.264	Anchor	FLBB Message	9.540000																										Ave 3								
$	24-Nov-2016	19:15:37.370	Anchor	FLBB Message	9.540000																										Pkt 2								
$	24-Nov-2016	19:15:37.477	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:15:48.967	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:16:35.376	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:16:37.432	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,43,700,113,538								
$	24-Nov-2016	19:16:35.657	Anchor	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.214,AirSaturation[%],123.359,Temperature[Deg.C],18.602,CalPhase[Deg],27.204,TCPhase[Deg],28.025,C1RPh[Deg],36.034,C2RPh[Deg],8.009,C1Amp[mV],499.5,C2Amp[mV],758.6,RawTemp[mV],288.6							
$	24-Nov-2016	19:16:35.660	Anchor	Optode Message	9.570000																											#							
$	24-Nov-2016	19:16:37.770	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,54,700,108,539								
$	24-Nov-2016	19:16:37.879	Anchor	FLBB Message	9.570000																																		
$	24-Nov-2016	19:16:37.982	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:16:38.087	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:16:38.191	Anchor	FLBB Message	9.570000																										Ave 3								
$	24-Nov-2016	19:16:38.297	Anchor	FLBB Message	9.570000																										Pkt 2								
$	24-Nov-2016	19:16:46.967	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:17:38.123	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,48,700,108,538								
$	24-Nov-2016	19:17:35.635	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.478,AirSaturation[%],124.155,Temperature[Deg.C],18.610,CalPhase[Deg],27.136,TCPhase[Deg],27.957,C1RPh[Deg],35.969,C2RPh[Deg],8.011,C1Amp[mV],497.5,C2Amp[mV],758.3,RawTemp[mV],288.3							
$	24-Nov-2016	19:17:35.638	Anchor	Optode Message	9.550000																											#							
$	24-Nov-2016	19:17:38.461	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,50,700,111,539								
$	24-Nov-2016	19:17:38.566	Anchor	FLBB Message	9.550000																																		
$	24-Nov-2016	19:17:38.671	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:17:38.778	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:17:38.882	Anchor	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:17:38.986	Anchor	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:17:36.377	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:17:47.968	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:18:36.306	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,112,539								
$	24-Nov-2016	19:18:35.634	Anchor	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.107,AirSaturation[%],124.396,Temperature[Deg.C],18.620,CalPhase[Deg],27.115,TCPhase[Deg],27.936,C1RPh[Deg],35.946,C2RPh[Deg],8.010,C1Amp[mV],496.8,C2Amp[mV],758.1,RawTemp[mV],288.0							
$	24-Nov-2016	19:18:35.637	Anchor	Optode Message	9.540000																											#							
$	24-Nov-2016	19:18:36.645	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,49,700,117,538								
$	24-Nov-2016	19:18:36.750	Anchor	FLBB Message	9.540000																																		
$	24-Nov-2016	19:18:36.853	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:18:36.958	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:18:37.065	Anchor	FLBB Message	9.540000																										Ave 3								
$	24-Nov-2016	19:18:37.170	Anchor	FLBB Message	9.540000																										Pkt 2								
$	24-Nov-2016	19:18:37.377	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:18:48.967	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:19:37.387	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,53,700,112,538								
$	24-Nov-2016	19:19:35.677	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:19:35.706	Start CP Mode	Optode Message	9.550000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.752,AirSaturation[%],124.621,Temperature[Deg.C],18.622,CalPhase[Deg],27.096,TCPhase[Deg],27.917,C1RPh[Deg],35.929,C2RPh[Deg],8.012,C1Amp[mV],496.4,C2Amp[mV],758.0,RawTemp[mV],288.0							
$	24-Nov-2016	19:19:35.710	Start CP Mode	Optode Message	9.550000																											#							
$	24-Nov-2016	19:19:37.938	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,52,700,111,538								
$	24-Nov-2016	19:19:38.045	Start CP Mode	FLBB Message	9.550000																																		
$	24-Nov-2016	19:19:38.149	Start CP Mode	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:19:38.255	Start CP Mode	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:19:38.360	Start CP Mode	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:19:38.466	Start CP Mode	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:19:36.752	Start CP Mode	Optode Message	9.550000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.316,AirSaturation[%],124.818,Temperature[Deg.C],18.623,CalPhase[Deg],27.080,TCPhase[Deg],27.901,C1RPh[Deg],35.854,C2RPh[Deg],7.953,C1Amp[mV],496.0,C2Amp[mV],757.9,RawTemp[mV],287.9							
$	24-Nov-2016	19:19:38.891	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,49,700,112,538								
$	24-Nov-2016	19:19:36.756	Start CP Mode	Optode Message	9.550000																											#							
$	24-Nov-2016	19:19:39.111	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,48,700,112,539								
$	24-Nov-2016	19:19:39.218	Start CP Mode	FLBB Message	9.550000																																		
$	24-Nov-2016	19:19:39.322	Start CP Mode	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:19:39.429	Start CP Mode	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:19:39.533	Start CP Mode	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:19:39.637	Start CP Mode	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:19:46.966	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:19:52.506	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:19:58.596	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	24-Nov-2016	19:19:58.597	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	24-Nov-2016	19:20:06.956	DeAnchor	Unparsed SBE41 Message						S>    9.61, 18.5862, 35.1866  																													
$	24-Nov-2016	19:22:17.312	Ascend	FLBB Message	4.980000																										99/99/99,99:99:99,695,49,700,110,539								
$	24-Nov-2016	19:22:17.085	Ascend	Optode Message	4.980000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.765,AirSaturation[%],124.973,Temperature[Deg.C],18.624,CalPhase[Deg],27.067,TCPhase[Deg],27.888,C1RPh[Deg],35.887,C2RPh[Deg],7.999,C1Amp[mV],496.1,C2Amp[mV],754.3,RawTemp[mV],287.9							
$	24-Nov-2016	19:22:17.085	Ascend	Optode Message	4.980000																											#							
$	24-Nov-2016	19:22:17.652	Ascend	FLBB Message	4.980000																										99/99/99,99:99:99,695,50,700,115,539								
$	24-Nov-2016	19:22:17.760	Ascend	FLBB Message	4.980000																																		
$	24-Nov-2016	19:22:17.864	Ascend	FLBB Message	4.980000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:22:17.968	Ascend	FLBB Message	4.980000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:22:18.073	Ascend	FLBB Message	4.980000																										Ave 3								
$	24-Nov-2016	19:22:18.178	Ascend	FLBB Message	4.980000																										Pkt 2								
$	24-Nov-2016	19:22:26.887	Ascend	Optode Message	3.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.672,AirSaturation[%],125.566,Temperature[Deg.C],18.600,CalPhase[Deg],27.022,TCPhase[Deg],27.843,C1RPh[Deg],35.856,C2RPh[Deg],8.012,C1Amp[mV],496.0,C2Amp[mV],759.1,RawTemp[mV],288.6							
$	24-Nov-2016	19:22:27.254	Ascend	FLBB Message	3.520000																										99/99/99,99:99:99,695,48,700,110,539								
$	24-Nov-2016	19:22:26.888	Ascend	Optode Message	3.520000																											#							
$	24-Nov-2016	19:22:27.474	Ascend	FLBB Message	3.520000																										99/99/99,99:99:99,695,49,700,111,539								
$	24-Nov-2016	19:22:27.580	Ascend	FLBB Message	3.520000																																		
$	24-Nov-2016	19:22:27.684	Ascend	FLBB Message	3.520000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:22:27.788	Ascend	FLBB Message	3.520000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:22:27.894	Ascend	FLBB Message	3.520000																										Ave 3								
$	24-Nov-2016	19:22:28.000	Ascend	FLBB Message	3.520000																										Pkt 2								
$	24-Nov-2016	19:22:46.510	Ascend	FLBB Message	1.300000																										99/99/99,99:99:99,695,49,700,118,538								
$	24-Nov-2016	19:22:45.880	Ascend	Optode Message	1.300000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.069,AirSaturation[%],125.677,Temperature[Deg.C],18.590,CalPhase[Deg],27.015,TCPhase[Deg],27.836,C1RPh[Deg],35.847,C2RPh[Deg],8.011,C1Amp[mV],496.3,C2Amp[mV],759.2,RawTemp[mV],289.0							
$	24-Nov-2016	19:22:45.880	Ascend	Optode Message	1.300000																											#							
$	24-Nov-2016	19:22:46.857	Ascend	FLBB Message	1.300000																										99/99/99,99:99:99,695,52,700,112,538								
$	24-Nov-2016	19:22:46.965	Ascend	FLBB Message	1.300000																																		
$	24-Nov-2016	19:22:47.069	Ascend	FLBB Message	1.300000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:22:47.176	Ascend	FLBB Message	1.300000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:22:47.281	Ascend	FLBB Message	1.300000																										Ave 3								
$	24-Nov-2016	19:22:47.385	Ascend	FLBB Message	1.300000																										Pkt 2								
$	24-Nov-2016	19:22:48.532	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	24-Nov-2016	19:22:49.068	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	19:22:51.459	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	19:22:52.641	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	24-Nov-2016	19:22:52.663	Init Profile	Unparsed SBE41 Message						powering down  																													
$	24-Nov-2016	19:23:15.240	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	24-Nov-2016	19:24:15.027	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:24:21.104	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	24-Nov-2016	19:24:21.259	Dump CP Data	Unparsed SBE41 Message						samples = 166, maxPress = 9.63  																													
$	24-Nov-2016	19:24:29.209	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	24-Nov-2016	19:24:29.894	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	24-Nov-2016	19:24:30.104	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	24-Nov-2016	19:24:30.233	Dump CP Data	Unparsed SBE41 Message						  																													
$	24-Nov-2016	19:24:30.252	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	24-Nov-2016	19:24:36.285	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	24-Nov-2016	19:24:36.286	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.322	Dump CP Data	Unparsed SBE41 Message						    1.92,  18.5822,  35.1870, 18  																													
$	24-Nov-2016	19:24:36.372	Dump CP Data	Unparsed SBE41 Message						    3.90,  18.5803,  35.1868, 15  																													
$	24-Nov-2016	19:24:36.421	Dump CP Data	Unparsed SBE41 Message						    6.02,  18.5851,  35.1868, 13  																													
$	24-Nov-2016	19:24:36.471	Dump CP Data	Unparsed SBE41 Message						    8.10,  18.5855,  35.1868, 18  																													
$	24-Nov-2016	19:24:36.595	Dump CP Data	Unparsed SBE41 Message						    9.51,  18.5856,  35.1867, 102  																													
$	24-Nov-2016	19:24:36.595	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.703	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.704	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.810	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.810	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:24:36.811	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	19:24:42.441	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	24-Nov-2016	19:24:42.441	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.442	Dump CP Data	Unparsed SBE41 Message						00134896897312  																													
$	24-Nov-2016	19:24:42.464	Dump CP Data	Unparsed SBE41 Message						0027489489730F  																													
$	24-Nov-2016	19:24:42.490	Dump CP Data	Unparsed SBE41 Message						003C489989730D  																													
$	24-Nov-2016	19:24:42.515	Dump CP Data	Unparsed SBE41 Message						00514899897312  																													
$	24-Nov-2016	19:24:42.549	Dump CP Data	Unparsed SBE41 Message						005F489A897366  																													
$	24-Nov-2016	19:24:42.574	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.601	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.626	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.750	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.750	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:24:42.751	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	19:24:48.270	Dump CP Data	Unparsed SBE41 Message						S>  																													
