$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	24-Nov-2016	19:24:49.966	Surface Ops	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:24:53.458	Surface Ops	Optode Message	1.090000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.538,AirSaturation[%],125.855,Temperature[Deg.C],18.597,CalPhase[Deg],26.999,TCPhase[Deg],27.820,C1RPh[Deg],35.832,C2RPh[Deg],8.012,C1Amp[mV],497.1,C2Amp[mV],760.0,RawTemp[mV],288.7							
$	24-Nov-2016	19:24:53.458	Surface Ops	Optode Message	1.090000																											#							
$	24-Nov-2016	19:25:01.467	Surface Ops	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:26:20.555	Surface Ops	FLBB Message	0.170000																										99/99/99,99:99:99,695,47,700,114,539								
$	24-Nov-2016	19:26:20.666	Surface Ops	FLBB Message	0.170000																										99/99/99,99:99:99,695,50,700,112,539								
$	24-Nov-2016	19:26:20.771	Surface Ops	FLBB Message	0.170000																																		
$	24-Nov-2016	19:26:20.875	Surface Ops	FLBB Message	0.170000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:26:20.983	Surface Ops	FLBB Message	0.170000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:26:21.091	Surface Ops	FLBB Message	0.170000																										Ave 3								
$	24-Nov-2016	19:26:21.199	Surface Ops	FLBB Message	0.170000																										Pkt 2								
$	24-Nov-2016	19:26:21.197	Surface Ops	Optode Message	0.170000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.379,AirSaturation[%],101.748,Temperature[Deg.C],17.745,CalPhase[Deg],29.416,TCPhase[Deg],30.237,C1RPh[Deg],38.246,C2RPh[Deg],8.010,C1Amp[mV],565.5,C2Amp[mV],764.0,RawTemp[mV],315.9							
$	24-Nov-2016	19:26:21.198	Surface Ops	Optode Message	0.170000																											#							
$	24-Nov-2016	19:26:21.654	Surface Ops	FLBB Message	-0.100000																										99/99/99,99:99:99,695,50,700,111,539								
$	24-Nov-2016	19:26:21.760	Surface Ops	FLBB Message	-0.100000																										99/99/99,99:99:99,695,52,700,113,539								
$	24-Nov-2016	19:26:21.869	Surface Ops	FLBB Message	-0.100000																																		
$	24-Nov-2016	19:26:21.973	Surface Ops	FLBB Message	-0.100000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:26:22.080	Surface Ops	FLBB Message	-0.100000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:26:22.184	Surface Ops	FLBB Message	-0.100000																										Ave 3								
$	24-Nov-2016	19:26:22.288	Surface Ops	FLBB Message	-0.100000																										Pkt 2								
$	24-Nov-2016	19:29:29.915	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,116,538								
$	24-Nov-2016	19:29:29.750	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.993,AirSaturation[%],101.051,Temperature[Deg.C],17.316,CalPhase[Deg],29.571,TCPhase[Deg],30.392,C1RPh[Deg],38.399,C2RPh[Deg],8.007,C1Amp[mV],570.8,C2Amp[mV],768.9,RawTemp[mV],329.6							
$	24-Nov-2016	19:29:29.752	Anchor	Optode Message	9.540000																											#							
$	24-Nov-2016	19:29:30.255	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,52,700,111,539								
$	24-Nov-2016	19:29:30.364	Anchor	FLBB Message	9.540000																																		
$	24-Nov-2016	19:29:30.468	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:29:30.572	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:29:30.676	Anchor	FLBB Message	9.540000																										Ave 3								
$	24-Nov-2016	19:29:29.878	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:29:31.026	Anchor	FLBB Message	9.540000																										Pkt 2								
$	24-Nov-2016	19:29:41.469	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:30:30.817	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,48,700,111,539								
$	24-Nov-2016	19:30:29.746	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],355.233,AirSaturation[%],121.627,Temperature[Deg.C],18.591,CalPhase[Deg],27.351,TCPhase[Deg],28.172,C1RPh[Deg],36.121,C2RPh[Deg],7.949,C1Amp[mV],503.6,C2Amp[mV],759.4,RawTemp[mV],288.9							
$	24-Nov-2016	19:30:29.749	Anchor	Optode Message	9.550000																											#							
$	24-Nov-2016	19:30:31.163	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,46,700,111,539								
$	24-Nov-2016	19:30:31.268	Anchor	FLBB Message	9.550000																																		
$	24-Nov-2016	19:30:31.371	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:30:31.478	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:30:31.582	Anchor	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:30:31.688	Anchor	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:31:31.631	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,51,700,113,539								
$	24-Nov-2016	19:31:29.752	Anchor	Optode Message	9.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.263,AirSaturation[%],123.370,Temperature[Deg.C],18.600,CalPhase[Deg],27.203,TCPhase[Deg],28.024,C1RPh[Deg],35.974,C2RPh[Deg],7.950,C1Amp[mV],499.3,C2Amp[mV],758.5,RawTemp[mV],288.7							
$	24-Nov-2016	19:31:29.755	Anchor	Optode Message	9.570000																											#							
$	24-Nov-2016	19:31:31.970	Anchor	FLBB Message	9.570000																										99/99/99,99:99:99,695,49,700,118,539								
$	24-Nov-2016	19:31:32.082	Anchor	FLBB Message	9.570000																																		
$	24-Nov-2016	19:31:32.186	Anchor	FLBB Message	9.570000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:31:32.290	Anchor	FLBB Message	9.570000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:31:32.396	Anchor	FLBB Message	9.570000																										Ave 3								
$	24-Nov-2016	19:31:32.500	Anchor	FLBB Message	9.570000																										Pkt 2								
$	24-Nov-2016	19:31:32.080	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:31:43.468	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:32:32.239	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,52,700,112,538								
$	24-Nov-2016	19:32:29.751	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.971,AirSaturation[%],124.308,Temperature[Deg.C],18.604,CalPhase[Deg],27.125,TCPhase[Deg],27.946,C1RPh[Deg],35.898,C2RPh[Deg],7.953,C1Amp[mV],497.2,C2Amp[mV],758.3,RawTemp[mV],288.5							
$	24-Nov-2016	19:32:29.753	Anchor	Optode Message	9.550000																											#							
$	24-Nov-2016	19:32:32.615	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,51,700,115,539								
$	24-Nov-2016	19:32:32.722	Anchor	FLBB Message	9.550000																																		
$	24-Nov-2016	19:32:32.825	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:32:32.931	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:32:33.035	Anchor	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:32:29.878	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:32:33.351	Anchor	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:32:41.468	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:33:30.416	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,47,700,114,539								
$	24-Nov-2016	19:33:29.747	Anchor	Optode Message	9.550000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.256,AirSaturation[%],124.755,Temperature[Deg.C],18.606,CalPhase[Deg],27.088,TCPhase[Deg],27.909,C1RPh[Deg],35.920,C2RPh[Deg],8.011,C1Amp[mV],496.0,C2Amp[mV],758.1,RawTemp[mV],288.4							
$	24-Nov-2016	19:33:29.750	Anchor	Optode Message	9.550000																											#							
$	24-Nov-2016	19:33:30.755	Anchor	FLBB Message	9.550000																										99/99/99,99:99:99,695,50,700,115,538								
$	24-Nov-2016	19:33:30.865	Anchor	FLBB Message	9.550000																																		
$	24-Nov-2016	19:33:30.971	Anchor	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:33:31.075	Anchor	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:33:31.179	Anchor	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:33:31.285	Anchor	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:33:30.969	Anchor	Unparsed SBE41 Message						S>pts  																													
$	24-Nov-2016	19:33:42.546	Anchor	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:34:31.638	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,50,700,108,538								
$	24-Nov-2016	19:34:29.756	Start CP Mode	Optode Message	9.550000																											MEASUREMENT,4330,2592,O2Concentration[uM],364.921,AirSaturation[%],124.988,Temperature[Deg.C],18.609,CalPhase[Deg],27.068,TCPhase[Deg],27.889,C1RPh[Deg],35.900,C2RPh[Deg],8.011,C1Amp[mV],495.6,C2Amp[mV],757.9,RawTemp[mV],288.4							
$	24-Nov-2016	19:34:29.759	Start CP Mode	Optode Message	9.550000																											#							
$	24-Nov-2016	19:34:31.981	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,45,700,112,538								
$	24-Nov-2016	19:34:32.088	Start CP Mode	FLBB Message	9.550000																																		
$	24-Nov-2016	19:34:32.192	Start CP Mode	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:34:32.298	Start CP Mode	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:34:32.402	Start CP Mode	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:34:32.506	Start CP Mode	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:34:32.725	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,51,700,115,538								
$	24-Nov-2016	19:34:30.802	Start CP Mode	Optode Message	9.550000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.262,AirSaturation[%],125.104,Temperature[Deg.C],18.608,CalPhase[Deg],27.059,TCPhase[Deg],27.880,C1RPh[Deg],35.891,C2RPh[Deg],8.011,C1Amp[mV],495.2,C2Amp[mV],757.9,RawTemp[mV],288.4							
$	24-Nov-2016	19:34:30.805	Start CP Mode	Optode Message	9.550000																											#							
$	24-Nov-2016	19:34:33.066	Start CP Mode	FLBB Message	9.550000																										99/99/99,99:99:99,695,52,700,112,539								
$	24-Nov-2016	19:34:33.171	Start CP Mode	FLBB Message	9.550000																																		
$	24-Nov-2016	19:34:33.275	Start CP Mode	FLBB Message	9.550000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:34:33.379	Start CP Mode	FLBB Message	9.550000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:34:33.486	Start CP Mode	FLBB Message	9.550000																										Ave 3								
$	24-Nov-2016	19:34:33.592	Start CP Mode	FLBB Message	9.550000																										Pkt 2								
$	24-Nov-2016	19:34:37.443	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:34:43.530	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	24-Nov-2016	19:34:43.531	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	24-Nov-2016	19:34:52.487	DeAnchor	Unparsed SBE41 Message						S>    9.60, 18.5853, 35.1865  																													
$	24-Nov-2016	19:37:07.871	Ascend	FLBB Message	5.150000																										99/99/99,99:99:99,695,52,700,112,539								
$	24-Nov-2016	19:37:07.978	Ascend	Optode Message	5.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.930,AirSaturation[%],125.334,Temperature[Deg.C],18.609,CalPhase[Deg],27.040,TCPhase[Deg],27.861,C1RPh[Deg],35.860,C2RPh[Deg],8.000,C1Amp[mV],495.4,C2Amp[mV],754.4,RawTemp[mV],288.4							
$	24-Nov-2016	19:37:07.978	Ascend	Optode Message	5.150000																											#							
$	24-Nov-2016	19:37:08.216	Ascend	FLBB Message	5.150000																										99/99/99,99:99:99,695,47,700,115,539								
$	24-Nov-2016	19:37:08.322	Ascend	FLBB Message	5.150000																																		
$	24-Nov-2016	19:37:08.425	Ascend	FLBB Message	5.150000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:37:08.533	Ascend	FLBB Message	5.150000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:37:08.637	Ascend	FLBB Message	5.150000																										Ave 3								
$	24-Nov-2016	19:37:08.741	Ascend	FLBB Message	5.150000																										Pkt 2								
$	24-Nov-2016	19:37:20.854	Ascend	FLBB Message	3.280000																										99/99/99,99:99:99,695,53,700,117,539								
$	24-Nov-2016	19:37:20.183	Ascend	Optode Message	3.280000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.139,AirSaturation[%],125.718,Temperature[Deg.C],18.597,CalPhase[Deg],27.011,TCPhase[Deg],27.832,C1RPh[Deg],35.845,C2RPh[Deg],8.013,C1Amp[mV],495.5,C2Amp[mV],759.0,RawTemp[mV],288.8							
$	24-Nov-2016	19:37:20.294	Ascend	Optode Message	3.280000																											#							
$	24-Nov-2016	19:37:21.194	Ascend	FLBB Message	3.280000																										99/99/99,99:99:99,695,48,700,117,538								
$	24-Nov-2016	19:37:21.299	Ascend	FLBB Message	3.280000																																		
$	24-Nov-2016	19:37:21.405	Ascend	FLBB Message	3.280000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:37:21.509	Ascend	FLBB Message	3.280000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:37:21.613	Ascend	FLBB Message	3.280000																										Ave 3								
$	24-Nov-2016	19:37:21.718	Ascend	FLBB Message	3.280000																										Pkt 2								
$	24-Nov-2016	19:37:36.860	Ascend	FLBB Message	1.440000																										99/99/99,99:99:99,695,47,700,113,538								
$	24-Nov-2016	19:37:36.746	Ascend	Optode Message	1.460000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.422,AirSaturation[%],125.814,Temperature[Deg.C],18.597,CalPhase[Deg],27.003,TCPhase[Deg],27.824,C1RPh[Deg],35.835,C2RPh[Deg],8.011,C1Amp[mV],496.0,C2Amp[mV],759.3,RawTemp[mV],288.8							
$	24-Nov-2016	19:37:36.746	Ascend	Optode Message	1.460000																											#							
$	24-Nov-2016	19:37:37.440	Ascend	FLBB Message	1.460000																										99/99/99,99:99:99,695,47,700,112,539								
$	24-Nov-2016	19:37:37.548	Ascend	FLBB Message	1.460000																																		
$	24-Nov-2016	19:37:37.654	Ascend	FLBB Message	1.460000																										Ser FLBB2K-3310								
$	24-Nov-2016	19:37:37.758	Ascend	FLBB Message	1.460000																										Ver FLNTU 4.08								
$	24-Nov-2016	19:37:37.862	Ascend	FLBB Message	1.460000																										Ave 3								
$	24-Nov-2016	19:37:37.967	Ascend	FLBB Message	1.460000																										Pkt 2								
$	24-Nov-2016	19:37:38.979	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	24-Nov-2016	19:37:39.405	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	19:37:41.997	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	24-Nov-2016	19:37:43.179	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	24-Nov-2016	19:37:43.201	Init Profile	Unparsed SBE41 Message						powering down  																													
$	24-Nov-2016	19:38:07.525	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	24-Nov-2016	19:39:07.030	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	24-Nov-2016	19:39:13.106	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	24-Nov-2016	19:39:13.259	Dump CP Data	Unparsed SBE41 Message						samples = 171, maxPress = 9.62  																													
$	24-Nov-2016	19:39:21.209	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	24-Nov-2016	19:39:21.895	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	24-Nov-2016	19:39:22.208	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	24-Nov-2016	19:39:22.258	Dump CP Data	Unparsed SBE41 Message						  																													
$	24-Nov-2016	19:39:22.276	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	24-Nov-2016	19:39:28.301	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	24-Nov-2016	19:39:28.302	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.338	Dump CP Data	Unparsed SBE41 Message						    2.08,  18.5813,  35.1868, 18  																													
$	24-Nov-2016	19:39:28.388	Dump CP Data	Unparsed SBE41 Message						    3.99,  18.5828,  35.1870, 13  																													
$	24-Nov-2016	19:39:28.438	Dump CP Data	Unparsed SBE41 Message						    5.95,  18.5831,  35.1869, 13  																													
$	24-Nov-2016	19:39:28.487	Dump CP Data	Unparsed SBE41 Message						    8.09,  18.5838,  35.1869, 19  																													
$	24-Nov-2016	19:39:28.608	Dump CP Data	Unparsed SBE41 Message						    9.51,  18.5848,  35.1868, 108  																													
$	24-Nov-2016	19:39:28.608	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.714	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.715	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.823	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.823	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	24-Nov-2016	19:39:28.824	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	19:39:34.429	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	24-Nov-2016	19:39:34.429	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.430	Dump CP Data	Unparsed SBE41 Message						00154895897312  																													
$	24-Nov-2016	19:39:34.451	Dump CP Data	Unparsed SBE41 Message						0028489789730D  																													
$	24-Nov-2016	19:39:34.477	Dump CP Data	Unparsed SBE41 Message						003C489789730D  																													
$	24-Nov-2016	19:39:34.502	Dump CP Data	Unparsed SBE41 Message						00514898897313  																													
$	24-Nov-2016	19:39:34.536	Dump CP Data	Unparsed SBE41 Message						005F489989736C  																													
$	24-Nov-2016	19:39:34.561	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.588	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.613	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.736	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.737	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	24-Nov-2016	19:39:34.737	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	24-Nov-2016	19:39:40.255	Dump CP Data	Unparsed SBE41 Message						S>  																													
