$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	25-Nov-2016	12:33:08.233	Surface Ops	Optode Message	0.140000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],369.312,AirSaturation[%],126.404,Temperature[Deg.C],18.574,CalPhase[Deg],26.959,TCPhase[Deg],27.780,C1RPh[Deg],35.781,C2RPh[Deg],8.001,C1Amp[mV],492.6,C2Amp[mV],757.5,RawTemp[mV],289.5							
$	25-Nov-2016	12:33:08.234	Surface Ops	Optode Message	0.140000																											#							
$	25-Nov-2016	12:34:32.933	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,50,700,376,539								
$	25-Nov-2016	12:34:33.040	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,51,700,186,539								
$	25-Nov-2016	12:34:33.145	Surface Ops	FLBB Message	0.130000																																		
$	25-Nov-2016	12:34:33.251	Surface Ops	FLBB Message	0.130000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:34:33.358	Surface Ops	FLBB Message	0.130000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:34:33.465	Surface Ops	FLBB Message	0.130000																										Ave 3								
$	25-Nov-2016	12:34:33.569	Surface Ops	FLBB Message	0.130000																										Pkt 2								
$	25-Nov-2016	12:34:33.463	Surface Ops	Optode Message	0.130000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],304.955,AirSaturation[%],103.035,Temperature[Deg.C],17.943,CalPhase[Deg],29.240,TCPhase[Deg],30.061,C1RPh[Deg],38.001,C2RPh[Deg],7.940,C1Amp[mV],556.1,C2Amp[mV],762.9,RawTemp[mV],309.6							
$	25-Nov-2016	12:34:33.463	Surface Ops	Optode Message	0.130000																											#							
$	25-Nov-2016	12:34:34.108	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,51,700,249,539								
$	25-Nov-2016	12:34:34.215	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,46,700,315,539								
$	25-Nov-2016	12:34:34.322	Surface Ops	FLBB Message	0.130000																																		
$	25-Nov-2016	12:34:34.427	Surface Ops	FLBB Message	0.130000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:34:34.531	Surface Ops	FLBB Message	0.130000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:34:34.635	Surface Ops	FLBB Message	0.130000																										Ave 3								
$	25-Nov-2016	12:34:34.741	Surface Ops	FLBB Message	0.130000																										Pkt 2								
$	25-Nov-2016	12:37:54.265	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,48,700,132,539								
$	25-Nov-2016	12:37:54.101	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],303.156,AirSaturation[%],101.446,Temperature[Deg.C],17.478,CalPhase[Deg],29.498,TCPhase[Deg],30.319,C1RPh[Deg],38.315,C2RPh[Deg],7.996,C1Amp[mV],564.1,C2Amp[mV],766.8,RawTemp[mV],324.4							
$	25-Nov-2016	12:37:54.104	Anchor	Optode Message	9.500000																											#							
$	25-Nov-2016	12:37:54.605	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,47,700,133,539								
$	25-Nov-2016	12:37:54.712	Anchor	FLBB Message	9.500000																																		
$	25-Nov-2016	12:37:54.816	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:37:54.922	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:37:55.026	Anchor	FLBB Message	9.500000																										Ave 3								
$	25-Nov-2016	12:37:55.131	Anchor	FLBB Message	9.500000																										Pkt 2								
$	25-Nov-2016	12:37:55.822	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:38:07.413	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:38:53.823	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:38:55.264	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,45,700,130,539								
$	25-Nov-2016	12:38:54.097	Anchor	Optode Message	9.500000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],358.360,AirSaturation[%],122.649,Temperature[Deg.C],18.572,CalPhase[Deg],27.268,TCPhase[Deg],28.089,C1RPh[Deg],36.089,C2RPh[Deg],8.000,C1Amp[mV],499.7,C2Amp[mV],757.9,RawTemp[mV],289.5							
$	25-Nov-2016	12:38:54.100	Anchor	Optode Message	9.500000																											#							
$	25-Nov-2016	12:38:55.604	Anchor	FLBB Message	9.500000																										99/99/99,99:99:99,695,49,700,134,539								
$	25-Nov-2016	12:38:55.709	Anchor	FLBB Message	9.500000																																		
$	25-Nov-2016	12:38:55.813	Anchor	FLBB Message	9.500000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:38:55.917	Anchor	FLBB Message	9.500000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:38:56.023	Anchor	FLBB Message	9.500000																										Ave 3								
$	25-Nov-2016	12:38:56.129	Anchor	FLBB Message	9.500000																										Pkt 2								
$	25-Nov-2016	12:39:05.413	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:39:56.018	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,54,700,138,538								
$	25-Nov-2016	12:39:54.005	Anchor	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],362.844,AirSaturation[%],124.198,Temperature[Deg.C],18.578,CalPhase[Deg],27.138,TCPhase[Deg],27.959,C1RPh[Deg],35.960,C2RPh[Deg],8.001,C1Amp[mV],496.0,C2Amp[mV],757.3,RawTemp[mV],289.4							
$	25-Nov-2016	12:39:54.008	Anchor	Optode Message	9.510000																											#							
$	25-Nov-2016	12:39:56.357	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,47,700,131,539								
$	25-Nov-2016	12:39:56.462	Anchor	FLBB Message	9.510000																																		
$	25-Nov-2016	12:39:56.569	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:39:56.673	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:39:56.778	Anchor	FLBB Message	9.510000																										Ave 3								
$	25-Nov-2016	12:39:56.882	Anchor	FLBB Message	9.510000																										Pkt 2								
$	25-Nov-2016	12:39:54.823	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:40:06.413	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:40:56.918	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,51,700,131,539								
$	25-Nov-2016	12:40:54.126	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.257,AirSaturation[%],124.698,Temperature[Deg.C],18.584,CalPhase[Deg],27.096,TCPhase[Deg],27.917,C1RPh[Deg],35.919,C2RPh[Deg],8.002,C1Amp[mV],494.9,C2Amp[mV],757.0,RawTemp[mV],289.2							
$	25-Nov-2016	12:40:54.130	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	12:40:57.259	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,46,700,132,539								
$	25-Nov-2016	12:40:57.366	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	12:40:57.469	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:40:57.576	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:40:57.680	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	12:40:57.784	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	12:40:55.824	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:41:07.413	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:41:53.823	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:41:54.966	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,51,700,131,539								
$	25-Nov-2016	12:41:54.101	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.632,AirSaturation[%],124.825,Temperature[Deg.C],18.584,CalPhase[Deg],27.086,TCPhase[Deg],27.907,C1RPh[Deg],35.908,C2RPh[Deg],8.002,C1Amp[mV],494.5,C2Amp[mV],756.9,RawTemp[mV],289.2							
$	25-Nov-2016	12:41:54.104	Anchor	Optode Message	9.530000																											#							
$	25-Nov-2016	12:41:55.304	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,51,700,134,539								
$	25-Nov-2016	12:41:55.409	Anchor	FLBB Message	9.530000																																		
$	25-Nov-2016	12:41:55.515	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:41:55.619	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:41:55.725	Anchor	FLBB Message	9.530000																										Ave 3								
$	25-Nov-2016	12:41:55.829	Anchor	FLBB Message	9.530000																										Pkt 2								
$	25-Nov-2016	12:42:05.414	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:42:56.026	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,135,538								
$	25-Nov-2016	12:42:54.142	Start CP Mode	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.632,AirSaturation[%],125.179,Temperature[Deg.C],18.588,CalPhase[Deg],27.056,TCPhase[Deg],27.877,C1RPh[Deg],35.880,C2RPh[Deg],8.004,C1Amp[mV],493.8,C2Amp[mV],756.8,RawTemp[mV],289.0							
$	25-Nov-2016	12:42:54.145	Start CP Mode	Optode Message	9.510000																											#							
$	25-Nov-2016	12:42:56.367	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,132,538								
$	25-Nov-2016	12:42:56.473	Start CP Mode	FLBB Message	9.510000																																		
$	25-Nov-2016	12:42:56.578	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:42:56.683	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:42:56.787	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	25-Nov-2016	12:42:56.893	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	25-Nov-2016	12:42:54.823	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	12:42:55.188	Start CP Mode	Optode Message	9.510000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.144,AirSaturation[%],125.375,Temperature[Deg.C],18.596,CalPhase[Deg],27.039,TCPhase[Deg],27.860,C1RPh[Deg],35.861,C2RPh[Deg],8.002,C1Amp[mV],493.4,C2Amp[mV],756.8,RawTemp[mV],288.8							
$	25-Nov-2016	12:42:55.191	Start CP Mode	Optode Message	9.510000																											#							
$	25-Nov-2016	12:42:57.444	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,132,538								
$	25-Nov-2016	12:42:57.549	Start CP Mode	FLBB Message	9.510000																										99/99/99,99:99:99,695,53,700,134,538								
$	25-Nov-2016	12:42:57.656	Start CP Mode	FLBB Message	9.510000																																		
$	25-Nov-2016	12:42:57.760	Start CP Mode	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:42:57.864	Start CP Mode	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:42:57.971	Start CP Mode	FLBB Message	9.510000																										Ave 3								
$	25-Nov-2016	12:42:58.077	Start CP Mode	FLBB Message	9.510000																										Pkt 2								
$	25-Nov-2016	12:43:06.413	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:43:10.953	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:43:17.043	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	25-Nov-2016	12:43:17.044	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	25-Nov-2016	12:43:25.170	DeAnchor	Unparsed SBE41 Message						S>    9.44, 18.5601, 35.1900  																													
$	25-Nov-2016	12:45:35.861	Ascend	FLBB Message	5.070000																										99/99/99,99:99:99,695,49,700,125,539								
$	25-Nov-2016	12:45:35.970	Ascend	Optode Message	5.070000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.409,AirSaturation[%],125.466,Temperature[Deg.C],18.596,CalPhase[Deg],27.031,TCPhase[Deg],27.852,C1RPh[Deg],35.841,C2RPh[Deg],7.989,C1Amp[mV],493.5,C2Amp[mV],753.2,RawTemp[mV],288.8							
$	25-Nov-2016	12:45:35.970	Ascend	Optode Message	5.070000																											#							
$	25-Nov-2016	12:45:36.406	Ascend	FLBB Message	5.070000																										99/99/99,99:99:99,695,50,700,125,538								
$	25-Nov-2016	12:45:36.513	Ascend	FLBB Message	5.070000																																		
$	25-Nov-2016	12:45:36.619	Ascend	FLBB Message	5.070000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:45:36.723	Ascend	FLBB Message	5.070000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:45:36.827	Ascend	FLBB Message	5.070000																										Ave 3								
$	25-Nov-2016	12:45:36.931	Ascend	FLBB Message	5.070000																										Pkt 2								
$	25-Nov-2016	12:45:49.349	Ascend	FLBB Message	3.100000																										99/99/99,99:99:99,695,50,700,132,539								
$	25-Nov-2016	12:45:48.728	Ascend	Optode Message	3.120000																											MEASUREMENT,4330,2592,O2Concentration[uM],368.504,AirSaturation[%],126.129,Temperature[Deg.C],18.575,CalPhase[Deg],26.981,TCPhase[Deg],27.802,C1RPh[Deg],35.804,C2RPh[Deg],8.002,C1Amp[mV],492.4,C2Amp[mV],757.3,RawTemp[mV],289.5							
$	25-Nov-2016	12:45:48.728	Ascend	Optode Message	3.120000																											#							
$	25-Nov-2016	12:45:49.934	Ascend	FLBB Message	3.120000																										99/99/99,99:99:99,695,48,700,126,538								
$	25-Nov-2016	12:45:50.039	Ascend	FLBB Message	3.120000																																		
$	25-Nov-2016	12:45:50.143	Ascend	FLBB Message	3.120000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:45:50.248	Ascend	FLBB Message	3.120000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:45:50.354	Ascend	FLBB Message	3.120000																										Ave 3								
$	25-Nov-2016	12:45:50.463	Ascend	FLBB Message	3.120000																										Pkt 2								
$	25-Nov-2016	12:46:05.744	Ascend	FLBB Message	1.150000																										99/99/99,99:99:99,695,50,700,124,538								
$	25-Nov-2016	12:46:05.519	Ascend	Optode Message	1.150000																											MEASUREMENT,4330,2592,O2Concentration[uM],368.948,AirSaturation[%],126.280,Temperature[Deg.C],18.575,CalPhase[Deg],26.969,TCPhase[Deg],27.790,C1RPh[Deg],35.791,C2RPh[Deg],8.001,C1Amp[mV],492.5,C2Amp[mV],757.4,RawTemp[mV],289.5							
$	25-Nov-2016	12:46:05.971	Ascend	FLBB Message	1.150000																										99/99/99,99:99:99,695,45,700,121,539								
$	25-Nov-2016	12:46:06.076	Ascend	FLBB Message	1.150000																																		
$	25-Nov-2016	12:46:06.180	Ascend	FLBB Message	1.150000																										Ser FLBB2K-3310								
$	25-Nov-2016	12:46:06.286	Ascend	FLBB Message	1.150000																										Ver FLNTU 4.08								
$	25-Nov-2016	12:46:06.393	Ascend	FLBB Message	1.150000																										Ave 3								
$	25-Nov-2016	12:46:06.497	Ascend	FLBB Message	1.150000																										Pkt 2								
$	25-Nov-2016	12:46:05.633	Ascend	Optode Message	1.150000																											#							
$	25-Nov-2016	12:46:07.724	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	25-Nov-2016	12:46:07.837	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	25-Nov-2016	12:46:10.633	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	25-Nov-2016	12:46:11.813	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	25-Nov-2016	12:46:11.835	Init Profile	Unparsed SBE41 Message						powering down  																													
$	25-Nov-2016	12:46:36.484	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	25-Nov-2016	12:47:39.273	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	12:47:45.351	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	25-Nov-2016	12:47:45.505	Dump CP Data	Unparsed SBE41 Message						samples = 167, maxPress = 9.59  																													
$	25-Nov-2016	12:47:53.455	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	25-Nov-2016	12:47:54.141	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	25-Nov-2016	12:47:54.349	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	25-Nov-2016	12:47:54.482	Dump CP Data	Unparsed SBE41 Message						  																													
$	25-Nov-2016	12:47:54.500	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	25-Nov-2016	12:48:00.556	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	25-Nov-2016	12:48:00.557	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:00.594	Dump CP Data	Unparsed SBE41 Message						    1.95,  18.5594,  35.1893, 19  																													
$	25-Nov-2016	12:48:00.644	Dump CP Data	Unparsed SBE41 Message						    3.94,  18.5607,  35.1894, 14  																													
$	25-Nov-2016	12:48:00.693	Dump CP Data	Unparsed SBE41 Message						    5.96,  18.5597,  35.1894, 13  																													
$	25-Nov-2016	12:48:00.743	Dump CP Data	Unparsed SBE41 Message						    8.12,  18.5604,  35.1894, 19  																													
$	25-Nov-2016	12:48:00.864	Dump CP Data	Unparsed SBE41 Message						    9.47,  18.5607,  35.1897, 102  																													
$	25-Nov-2016	12:48:00.865	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:00.975	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:00.975	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:01.081	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:01.082	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	12:48:01.082	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	12:48:06.588	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	25-Nov-2016	12:48:06.589	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.689	Dump CP Data	Unparsed SBE41 Message						0013487F897513  																													
$	25-Nov-2016	12:48:06.690	Dump CP Data	Unparsed SBE41 Message						0027488189750E  																													
$	25-Nov-2016	12:48:06.690	Dump CP Data	Unparsed SBE41 Message						003C488089750D  																													
$	25-Nov-2016	12:48:06.690	Dump CP Data	Unparsed SBE41 Message						00514880897513  																													
$	25-Nov-2016	12:48:06.705	Dump CP Data	Unparsed SBE41 Message						005F4881897666  																													
$	25-Nov-2016	12:48:06.731	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.756	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.781	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.807	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.842	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	12:48:06.862	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	12:48:12.511	Dump CP Data	Unparsed SBE41 Message						S>  																													
