$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	25-Nov-2016	06:59:01.627	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.596,AirSaturation[%],125.797,Temperature[Deg.C],18.567,CalPhase[Deg],27.009,TCPhase[Deg],27.830,C1RPh[Deg],35.834,C2RPh[Deg],8.004,C1Amp[mV],494.1,C2Amp[mV],757.8,RawTemp[mV],289.7							
$	25-Nov-2016	06:59:01.627	Surface Ops	Optode Message	0.150000																											#							
$	25-Nov-2016	07:00:25.844	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,53,700,127,539								
$	25-Nov-2016	07:00:25.980	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,51,700,131,539								
$	25-Nov-2016	07:00:26.085	Surface Ops	FLBB Message	0.150000																																		
$	25-Nov-2016	07:00:26.195	Surface Ops	FLBB Message	0.150000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:00:26.301	Surface Ops	FLBB Message	0.150000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:00:26.407	Surface Ops	FLBB Message	0.150000																										Ave 3								
$	25-Nov-2016	07:00:26.511	Surface Ops	FLBB Message	0.150000																										Pkt 2								
$	25-Nov-2016	07:00:26.405	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],303.969,AirSaturation[%],103.088,Temperature[Deg.C],18.126,CalPhase[Deg],29.201,TCPhase[Deg],30.022,C1RPh[Deg],38.025,C2RPh[Deg],8.002,C1Amp[mV],555.0,C2Amp[mV],761.2,RawTemp[mV],303.8							
$	25-Nov-2016	07:00:26.406	Surface Ops	Optode Message	0.150000																											#							
$	25-Nov-2016	07:00:27.055	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,53,700,129,539								
$	25-Nov-2016	07:00:27.165	Surface Ops	FLBB Message	0.150000																										99/99/99,99:99:99,695,52,700,132,539								
$	25-Nov-2016	07:00:27.270	Surface Ops	FLBB Message	0.150000																																		
$	25-Nov-2016	07:00:27.374	Surface Ops	FLBB Message	0.150000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:00:27.478	Surface Ops	FLBB Message	0.150000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:00:27.582	Surface Ops	FLBB Message	0.150000																										Ave 3								
$	25-Nov-2016	07:00:27.690	Surface Ops	FLBB Message	0.150000																										Pkt 2								
$	25-Nov-2016	07:03:44.196	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,47,700,137,539								
$	25-Nov-2016	07:03:44.026	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],301.798,AirSaturation[%],101.528,Temperature[Deg.C],17.734,CalPhase[Deg],29.442,TCPhase[Deg],30.263,C1RPh[Deg],38.264,C2RPh[Deg],8.001,C1Amp[mV],562.6,C2Amp[mV],765.1,RawTemp[mV],316.3							
$	25-Nov-2016	07:03:44.028	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	07:03:44.536	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,47,700,136,539								
$	25-Nov-2016	07:03:44.642	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	07:03:44.745	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:03:44.853	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:03:44.957	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	07:03:45.061	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	07:03:46.032	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:03:57.623	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:04:44.784	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,136,539								
$	25-Nov-2016	07:04:44.032	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:04:44.033	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],355.514,AirSaturation[%],121.655,Temperature[Deg.C],18.564,CalPhase[Deg],27.353,TCPhase[Deg],28.174,C1RPh[Deg],36.176,C2RPh[Deg],8.002,C1Amp[mV],501.8,C2Amp[mV],757.8,RawTemp[mV],289.8							
$	25-Nov-2016	07:04:44.035	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	07:04:45.333	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,138,539								
$	25-Nov-2016	07:04:45.438	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	07:04:45.544	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:04:45.648	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:04:45.754	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	07:04:45.858	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	07:04:55.622	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:05:45.722	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,51,700,135,539								
$	25-Nov-2016	07:05:44.047	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],360.012,AirSaturation[%],123.215,Temperature[Deg.C],18.572,CalPhase[Deg],27.221,TCPhase[Deg],28.042,C1RPh[Deg],35.986,C2RPh[Deg],7.945,C1Amp[mV],498.2,C2Amp[mV],757.3,RawTemp[mV],289.5							
$	25-Nov-2016	07:05:44.050	Anchor	Optode Message	9.530000																											#							
$	25-Nov-2016	07:05:46.061	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,49,700,143,539								
$	25-Nov-2016	07:05:46.166	Anchor	FLBB Message	9.530000																																		
$	25-Nov-2016	07:05:46.271	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:05:46.375	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:05:46.482	Anchor	FLBB Message	9.530000																										Ave 3								
$	25-Nov-2016	07:05:46.586	Anchor	FLBB Message	9.530000																										Pkt 2								
$	25-Nov-2016	07:05:45.032	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:05:56.623	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:06:46.621	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,49,700,138,539								
$	25-Nov-2016	07:06:44.030	Anchor	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],361.852,AirSaturation[%],123.858,Temperature[Deg.C],18.577,CalPhase[Deg],27.166,TCPhase[Deg],27.987,C1RPh[Deg],35.992,C2RPh[Deg],8.005,C1Amp[mV],496.6,C2Amp[mV],757.1,RawTemp[mV],289.4							
$	25-Nov-2016	07:06:44.033	Anchor	Optode Message	9.540000																											#							
$	25-Nov-2016	07:06:46.961	Anchor	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,138,539								
$	25-Nov-2016	07:06:47.066	Anchor	FLBB Message	9.540000																																		
$	25-Nov-2016	07:06:47.170	Anchor	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:06:47.276	Anchor	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:06:47.386	Anchor	FLBB Message	9.540000																										Ave 3								
$	25-Nov-2016	07:06:47.490	Anchor	FLBB Message	9.540000																										Pkt 2								
$	25-Nov-2016	07:06:46.032	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:06:57.635	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:07:44.479	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,55,700,139,539								
$	25-Nov-2016	07:07:44.027	Anchor	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],362.889,AirSaturation[%],124.240,Temperature[Deg.C],18.588,CalPhase[Deg],27.133,TCPhase[Deg],27.954,C1RPh[Deg],35.959,C2RPh[Deg],8.005,C1Amp[mV],495.8,C2Amp[mV],757.0,RawTemp[mV],289.0							
$	25-Nov-2016	07:07:44.031	Anchor	Optode Message	9.530000																											#							
$	25-Nov-2016	07:07:44.032	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:07:45.032	Anchor	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,142,539								
$	25-Nov-2016	07:07:45.137	Anchor	FLBB Message	9.530000																																		
$	25-Nov-2016	07:07:45.241	Anchor	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:07:45.345	Anchor	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:07:45.453	Anchor	FLBB Message	9.530000																										Ave 3								
$	25-Nov-2016	07:07:45.557	Anchor	FLBB Message	9.530000																										Pkt 2								
$	25-Nov-2016	07:07:55.622	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:08:45.753	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,132,538								
$	25-Nov-2016	07:08:44.074	Start CP Mode	Optode Message	9.540000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],363.320,AirSaturation[%],124.379,Temperature[Deg.C],18.585,CalPhase[Deg],27.122,TCPhase[Deg],27.943,C1RPh[Deg],35.889,C2RPh[Deg],7.946,C1Amp[mV],495.3,C2Amp[mV],756.8,RawTemp[mV],289.1							
$	25-Nov-2016	07:08:44.077	Start CP Mode	Optode Message	9.540000																											#							
$	25-Nov-2016	07:08:46.092	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,56,700,138,539								
$	25-Nov-2016	07:08:46.197	Start CP Mode	FLBB Message	9.540000																																		
$	25-Nov-2016	07:08:46.305	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:08:46.409	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:08:46.513	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	25-Nov-2016	07:08:46.617	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	25-Nov-2016	07:08:45.032	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	07:08:45.119	Start CP Mode	Optode Message	9.540000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.724,AirSaturation[%],124.536,Temperature[Deg.C],18.592,CalPhase[Deg],27.108,TCPhase[Deg],27.929,C1RPh[Deg],35.935,C2RPh[Deg],8.006,C1Amp[mV],495.0,C2Amp[mV],756.8,RawTemp[mV],288.9							
$	25-Nov-2016	07:08:45.122	Start CP Mode	Optode Message	9.540000																											#							
$	25-Nov-2016	07:08:47.375	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,49,700,136,538								
$	25-Nov-2016	07:08:47.481	Start CP Mode	FLBB Message	9.540000																										99/99/99,99:99:99,695,50,700,141,539								
$	25-Nov-2016	07:08:47.586	Start CP Mode	FLBB Message	9.540000																																		
$	25-Nov-2016	07:08:47.690	Start CP Mode	FLBB Message	9.540000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:08:47.798	Start CP Mode	FLBB Message	9.540000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:08:47.903	Start CP Mode	FLBB Message	9.540000																										Ave 3								
$	25-Nov-2016	07:08:48.007	Start CP Mode	FLBB Message	9.540000																										Pkt 2								
$	25-Nov-2016	07:08:56.629	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:09:00.886	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:09:07.002	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	25-Nov-2016	07:09:07.003	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	25-Nov-2016	07:09:15.752	DeAnchor	Unparsed SBE41 Message						S>    9.54, 18.5528, 35.1881  																													
$	25-Nov-2016	07:11:29.375	Ascend	FLBB Message	4.910000																										99/99/99,99:99:99,695,48,700,120,539								
$	25-Nov-2016	07:11:29.484	Ascend	FLBB Message	4.910000																										99/99/99,99:99:99,695,50,700,121,539								
$	25-Nov-2016	07:11:29.590	Ascend	FLBB Message	4.910000																																		
$	25-Nov-2016	07:11:29.261	Ascend	Optode Message	4.910000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.085,AirSaturation[%],124.660,Temperature[Deg.C],18.592,CalPhase[Deg],27.098,TCPhase[Deg],27.919,C1RPh[Deg],35.910,C2RPh[Deg],7.992,C1Amp[mV],495.1,C2Amp[mV],753.2,RawTemp[mV],288.9							
$	25-Nov-2016	07:11:29.261	Ascend	Optode Message	4.910000																											#							
$	25-Nov-2016	07:11:29.930	Ascend	FLBB Message	4.910000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:11:30.034	Ascend	FLBB Message	4.910000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:11:30.140	Ascend	FLBB Message	4.910000																										Ave 3								
$	25-Nov-2016	07:11:30.244	Ascend	FLBB Message	4.910000																										Pkt 2								
$	25-Nov-2016	07:11:42.544	Ascend	FLBB Message	3.250000																										99/99/99,99:99:99,695,49,700,114,539								
$	25-Nov-2016	07:11:42.118	Ascend	Optode Message	3.250000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.359,AirSaturation[%],125.371,Temperature[Deg.C],18.566,CalPhase[Deg],27.044,TCPhase[Deg],27.865,C1RPh[Deg],35.869,C2RPh[Deg],8.004,C1Amp[mV],494.1,C2Amp[mV],757.5,RawTemp[mV],289.7							
$	25-Nov-2016	07:11:42.227	Ascend	Optode Message	3.250000																											#							
$	25-Nov-2016	07:11:42.882	Ascend	FLBB Message	3.250000																										99/99/99,99:99:99,695,52,700,121,539								
$	25-Nov-2016	07:11:42.989	Ascend	FLBB Message	3.250000																																		
$	25-Nov-2016	07:11:43.093	Ascend	FLBB Message	3.250000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:11:43.445	Ascend	FLBB Message	2.990000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:11:43.549	Ascend	FLBB Message	2.990000																										Ave 3								
$	25-Nov-2016	07:11:43.657	Ascend	FLBB Message	2.990000																										Pkt 2								
$	25-Nov-2016	07:11:58.383	Ascend	FLBB Message	1.610000																										99/99/99,99:99:99,695,43,700,124,539								
$	25-Nov-2016	07:11:58.158	Ascend	Optode Message	1.610000																											MEASUREMENT,4330,2592,O2Concentration[uM],366.714,AirSaturation[%],125.493,Temperature[Deg.C],18.566,CalPhase[Deg],27.034,TCPhase[Deg],27.855,C1RPh[Deg],35.859,C2RPh[Deg],8.004,C1Amp[mV],494.2,C2Amp[mV],757.5,RawTemp[mV],289.7							
$	25-Nov-2016	07:11:58.268	Ascend	Optode Message	1.610000																											#							
$	25-Nov-2016	07:11:58.721	Ascend	FLBB Message	1.610000																										99/99/99,99:99:99,695,51,700,123,539								
$	25-Nov-2016	07:11:58.826	Ascend	FLBB Message	1.610000																																		
$	25-Nov-2016	07:11:58.931	Ascend	FLBB Message	1.610000																										Ser FLBB2K-3310								
$	25-Nov-2016	07:11:59.037	Ascend	FLBB Message	1.610000																										Ver FLNTU 4.08								
$	25-Nov-2016	07:11:59.143	Ascend	FLBB Message	1.610000																										Ave 3								
$	25-Nov-2016	07:11:59.247	Ascend	FLBB Message	1.610000																										Pkt 2								
$	25-Nov-2016	07:12:00.283	Ascend	Unparsed SBE41 Message						profile stopped  																													
$	25-Nov-2016	07:12:00.609	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	25-Nov-2016	07:12:03.302	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	25-Nov-2016	07:12:04.482	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	25-Nov-2016	07:12:04.504	Init Profile	Unparsed SBE41 Message						powering down  																													
$	25-Nov-2016	07:12:28.946	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	25-Nov-2016	07:13:31.235	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	07:13:37.312	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	25-Nov-2016	07:13:37.466	Dump CP Data	Unparsed SBE41 Message						samples = 169, maxPress = 9.60  																													
$	25-Nov-2016	07:13:45.415	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	25-Nov-2016	07:13:46.101	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	25-Nov-2016	07:13:46.309	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	25-Nov-2016	07:13:46.448	Dump CP Data	Unparsed SBE41 Message						  																													
$	25-Nov-2016	07:13:46.466	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	25-Nov-2016	07:13:52.492	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	25-Nov-2016	07:13:52.493	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:52.529	Dump CP Data	Unparsed SBE41 Message						    2.16,  18.5522,  35.1886, 19  																													
$	25-Nov-2016	07:13:52.579	Dump CP Data	Unparsed SBE41 Message						    4.04,  18.5518,  35.1887, 14  																													
$	25-Nov-2016	07:13:52.628	Dump CP Data	Unparsed SBE41 Message						    6.11,  18.5517,  35.1885, 13  																													
$	25-Nov-2016	07:13:52.678	Dump CP Data	Unparsed SBE41 Message						    8.16,  18.5521,  35.1885, 17  																													
$	25-Nov-2016	07:13:52.799	Dump CP Data	Unparsed SBE41 Message						    9.49,  18.5523,  35.1886, 106  																													
$	25-Nov-2016	07:13:52.799	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:52.907	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:52.908	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:53.014	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:53.015	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	07:13:53.016	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	07:13:58.621	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	25-Nov-2016	07:13:58.621	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.622	Dump CP Data	Unparsed SBE41 Message						00164878897513  																													
$	25-Nov-2016	07:13:58.642	Dump CP Data	Unparsed SBE41 Message						0028487889750E  																													
$	25-Nov-2016	07:13:58.668	Dump CP Data	Unparsed SBE41 Message						003D487889750D  																													
$	25-Nov-2016	07:13:58.693	Dump CP Data	Unparsed SBE41 Message						00524878897411  																													
$	25-Nov-2016	07:13:58.727	Dump CP Data	Unparsed SBE41 Message						005F487889756A  																													
$	25-Nov-2016	07:13:58.752	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.779	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.804	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.929	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.929	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	07:13:58.930	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	07:14:04.449	Dump CP Data	Unparsed SBE41 Message						S>  																													
