$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	25-Nov-2016	08:45:24.081	Surface Ops	Optode Message	0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],367.976,AirSaturation[%],125.939,Temperature[Deg.C],18.572,CalPhase[Deg],26.997,TCPhase[Deg],27.818,C1RPh[Deg],35.822,C2RPh[Deg],8.004,C1Amp[mV],493.6,C2Amp[mV],757.7,RawTemp[mV],289.6							
$	25-Nov-2016	08:45:24.325	Surface Ops	Optode Message	0.150000																											#							
$	25-Nov-2016	08:46:51.837	Surface Ops	FLBB Message	-0.150000																										99/99/99,99:99:99,695,51,700,312,539								
$	25-Nov-2016	08:46:51.944	Surface Ops	FLBB Message	-0.150000																										99/99/99,99:99:99,695,50,700,286,539								
$	25-Nov-2016	08:46:52.049	Surface Ops	FLBB Message	-0.150000																																		
$	25-Nov-2016	08:46:52.155	Surface Ops	FLBB Message	-0.150000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:46:52.263	Surface Ops	FLBB Message	-0.150000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:46:52.369	Surface Ops	FLBB Message	-0.150000																										Ave 3								
$	25-Nov-2016	08:46:52.473	Surface Ops	FLBB Message	-0.150000																										Pkt 2								
$	25-Nov-2016	08:46:52.368	Surface Ops	Optode Message	-0.150000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],304.563,AirSaturation[%],103.260,Temperature[Deg.C],18.112,CalPhase[Deg],29.185,TCPhase[Deg],30.006,C1RPh[Deg],38.008,C2RPh[Deg],8.001,C1Amp[mV],554.5,C2Amp[mV],761.2,RawTemp[mV],304.2							
$	25-Nov-2016	08:46:52.368	Surface Ops	Optode Message	-0.150000																											#							
$	25-Nov-2016	08:46:52.816	Surface Ops	FLBB Message	-0.150000																										99/99/99,99:99:99,695,50,700,154,539								
$	25-Nov-2016	08:46:52.923	Surface Ops	FLBB Message	-0.150000																										99/99/99,99:99:99,695,48,700,158,539								
$	25-Nov-2016	08:46:53.030	Surface Ops	FLBB Message	-0.150000																																		
$	25-Nov-2016	08:46:53.134	Surface Ops	FLBB Message	-0.150000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:46:53.239	Surface Ops	FLBB Message	-0.150000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:46:53.344	Surface Ops	FLBB Message	-0.150000																										Ave 3								
$	25-Nov-2016	08:46:53.450	Surface Ops	FLBB Message	-0.150000																										Pkt 2								
$	25-Nov-2016	08:50:06.743	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,113,539								
$	25-Nov-2016	08:50:06.576	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],302.185,AirSaturation[%],101.549,Temperature[Deg.C],17.681,CalPhase[Deg],29.449,TCPhase[Deg],30.270,C1RPh[Deg],38.269,C2RPh[Deg],7.998,C1Amp[mV],562.7,C2Amp[mV],765.6,RawTemp[mV],318.0							
$	25-Nov-2016	08:50:06.579	Anchor	Optode Message	9.510000																											#							
$	25-Nov-2016	08:50:07.084	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,116,539								
$	25-Nov-2016	08:50:07.189	Anchor	FLBB Message	9.510000																																		
$	25-Nov-2016	08:50:07.293	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:50:07.399	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:50:07.504	Anchor	FLBB Message	9.510000																										Ave 3								
$	25-Nov-2016	08:50:07.610	Anchor	FLBB Message	9.510000																										Pkt 2								
$	25-Nov-2016	08:50:08.577	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:50:20.168	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:51:07.531	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,51,700,112,539								
$	25-Nov-2016	08:51:06.576	Anchor	Optode Message	9.510000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],357.051,AirSaturation[%],122.189,Temperature[Deg.C],18.567,CalPhase[Deg],27.308,TCPhase[Deg],28.129,C1RPh[Deg],36.129,C2RPh[Deg],8.000,C1Amp[mV],500.7,C2Amp[mV],758.0,RawTemp[mV],289.7							
$	25-Nov-2016	08:51:06.577	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:51:06.579	Anchor	Optode Message	9.510000																											#							
$	25-Nov-2016	08:51:08.082	Anchor	FLBB Message	9.510000																										99/99/99,99:99:99,695,50,700,112,539								
$	25-Nov-2016	08:51:08.187	Anchor	FLBB Message	9.510000																																		
$	25-Nov-2016	08:51:08.291	Anchor	FLBB Message	9.510000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:51:08.397	Anchor	FLBB Message	9.510000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:51:08.503	Anchor	FLBB Message	9.510000																										Ave 3								
$	25-Nov-2016	08:51:08.607	Anchor	FLBB Message	9.510000																										Pkt 2								
$	25-Nov-2016	08:51:18.167	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:52:08.470	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,115,539								
$	25-Nov-2016	08:52:06.479	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],361.512,AirSaturation[%],123.728,Temperature[Deg.C],18.572,CalPhase[Deg],27.178,TCPhase[Deg],27.999,C1RPh[Deg],36.001,C2RPh[Deg],8.002,C1Amp[mV],497.0,C2Amp[mV],757.4,RawTemp[mV],289.5							
$	25-Nov-2016	08:52:06.482	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	08:52:08.811	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,109,539								
$	25-Nov-2016	08:52:08.916	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	08:52:09.022	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:52:09.126	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:52:09.230	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	08:52:09.336	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	08:52:07.578	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:52:19.168	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:53:09.169	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,50,700,112,538								
$	25-Nov-2016	08:53:06.578	Anchor	Optode Message	9.520000																											MEASUREMENT,4330,2592,O2Concentration[uM],363.895,AirSaturation[%],124.553,Temperature[Deg.C],18.576,CalPhase[Deg],27.109,TCPhase[Deg],27.930,C1RPh[Deg],35.933,C2RPh[Deg],8.003,C1Amp[mV],495.1,C2Amp[mV],757.1,RawTemp[mV],289.4							
$	25-Nov-2016	08:53:06.581	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	08:53:09.509	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,52,700,114,538								
$	25-Nov-2016	08:53:09.614	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	08:53:09.718	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:53:09.824	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:53:09.928	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	08:53:10.034	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	08:53:08.577	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:53:20.178	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:54:07.228	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,49,700,113,538								
$	25-Nov-2016	08:54:06.575	Anchor	Optode Message	9.520000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],364.983,AirSaturation[%],124.933,Temperature[Deg.C],18.579,CalPhase[Deg],27.078,TCPhase[Deg],27.899,C1RPh[Deg],35.843,C2RPh[Deg],7.944,C1Amp[mV],494.2,C2Amp[mV],757.0,RawTemp[mV],289.3							
$	25-Nov-2016	08:54:06.577	Anchor	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:54:06.578	Anchor	Optode Message	9.520000																											#							
$	25-Nov-2016	08:54:07.779	Anchor	FLBB Message	9.520000																										99/99/99,99:99:99,695,52,700,115,539								
$	25-Nov-2016	08:54:07.886	Anchor	FLBB Message	9.520000																																		
$	25-Nov-2016	08:54:07.989	Anchor	FLBB Message	9.520000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:54:08.094	Anchor	FLBB Message	9.520000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:54:08.200	Anchor	FLBB Message	9.520000																										Ave 3								
$	25-Nov-2016	08:54:08.303	Anchor	FLBB Message	9.520000																										Pkt 2								
$	25-Nov-2016	08:54:18.169	Anchor	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:55:08.502	Start CP Mode	FLBB Message	9.530000																										99/99/99,99:99:99,695,50,700,117,539								
$	25-Nov-2016	08:55:06.621	Start CP Mode	Optode Message	9.530000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],365.312,AirSaturation[%],125.089,Temperature[Deg.C],18.596,CalPhase[Deg],27.062,TCPhase[Deg],27.883,C1RPh[Deg],35.887,C2RPh[Deg],8.004,C1Amp[mV],493.8,C2Amp[mV],756.9,RawTemp[mV],288.8							
$	25-Nov-2016	08:55:06.624	Start CP Mode	Optode Message	9.530000																											#							
$	25-Nov-2016	08:55:08.840	Start CP Mode	FLBB Message	9.530000																										99/99/99,99:99:99,695,46,700,120,539								
$	25-Nov-2016	08:55:08.947	Start CP Mode	FLBB Message	9.530000																																		
$	25-Nov-2016	08:55:09.051	Start CP Mode	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:55:09.157	Start CP Mode	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:55:09.261	Start CP Mode	FLBB Message	9.530000																										Ave 3								
$	25-Nov-2016	08:55:09.365	Start CP Mode	FLBB Message	9.530000																										Pkt 2								
$	25-Nov-2016	08:55:07.577	Start CP Mode	Unparsed SBE41 Message						S>pts  																													
$	25-Nov-2016	08:55:07.667	Start CP Mode	Optode Message	9.530000																											MEASUREMENT,4330,2592,O2Concentration[uM],365.583,AirSaturation[%],125.171,Temperature[Deg.C],18.592,CalPhase[Deg],27.056,TCPhase[Deg],27.877,C1RPh[Deg],35.881,C2RPh[Deg],8.004,C1Amp[mV],493.6,C2Amp[mV],756.9,RawTemp[mV],288.9							
$	25-Nov-2016	08:55:07.670	Start CP Mode	Optode Message	9.530000																											#							
$	25-Nov-2016	08:55:09.917	Start CP Mode	FLBB Message	9.530000																										99/99/99,99:99:99,695,53,700,112,539								
$	25-Nov-2016	08:55:10.022	Start CP Mode	FLBB Message	9.530000																										99/99/99,99:99:99,695,49,700,115,538								
$	25-Nov-2016	08:55:10.128	Start CP Mode	FLBB Message	9.530000																																		
$	25-Nov-2016	08:55:10.232	Start CP Mode	FLBB Message	9.530000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:55:10.338	Start CP Mode	FLBB Message	9.530000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:55:10.444	Start CP Mode	FLBB Message	9.530000																										Ave 3								
$	25-Nov-2016	08:55:10.548	Start CP Mode	FLBB Message	9.530000																										Pkt 2								
$	25-Nov-2016	08:55:19.169	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:55:23.428	Start CP Mode	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:55:29.519	Start CP Mode	Unparsed SBE41 Message						S>startprofile5  																													
$	25-Nov-2016	08:55:29.520	Start CP Mode	Unparsed SBE41 Message						profile started, pump delay = 5 seconds  																													
$	25-Nov-2016	08:55:37.609	DeAnchor	Unparsed SBE41 Message						S>    9.55, 18.5566, 35.1887  																													
$	25-Nov-2016	08:57:45.393	Ascend	FLBB Message	4.980000																										99/99/99,99:99:99,695,50,700,115,539								
$	25-Nov-2016	08:57:45.503	Ascend	Optode Message	4.980000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],366.100,AirSaturation[%],125.349,Temperature[Deg.C],18.592,CalPhase[Deg],27.041,TCPhase[Deg],27.862,C1RPh[Deg],35.853,C2RPh[Deg],7.991,C1Amp[mV],493.7,C2Amp[mV],753.3,RawTemp[mV],288.9							
$	25-Nov-2016	08:57:45.503	Ascend	Optode Message	4.980000																											#							
$	25-Nov-2016	08:57:45.738	Ascend	FLBB Message	4.980000																										99/99/99,99:99:99,695,50,700,115,539								
$	25-Nov-2016	08:57:45.846	Ascend	FLBB Message	4.980000																																		
$	25-Nov-2016	08:57:46.194	Ascend	FLBB Message	4.970000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:57:46.298	Ascend	FLBB Message	4.970000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:57:46.402	Ascend	FLBB Message	4.970000																										Ave 3								
$	25-Nov-2016	08:57:46.507	Ascend	FLBB Message	4.970000																										Pkt 2								
$	25-Nov-2016	08:57:58.182	Ascend	FLBB Message	3.180000																										99/99/99,99:99:99,695,50,700,124,539								
$	25-Nov-2016	08:57:57.958	Ascend	Optode Message	3.180000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.376,AirSaturation[%],125.731,Temperature[Deg.C],18.570,CalPhase[Deg],27.014,TCPhase[Deg],27.835,C1RPh[Deg],35.838,C2RPh[Deg],8.003,C1Amp[mV],493.3,C2Amp[mV],757.5,RawTemp[mV],289.6							
$	25-Nov-2016	08:57:57.959	Ascend	Optode Message	3.180000																											#							
$	25-Nov-2016	08:57:58.521	Ascend	FLBB Message	3.180000																										99/99/99,99:99:99,695,48,700,114,539								
$	25-Nov-2016	08:57:58.626	Ascend	FLBB Message	3.180000																																		
$	25-Nov-2016	08:57:58.730	Ascend	FLBB Message	3.180000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:57:58.836	Ascend	FLBB Message	3.180000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:57:58.940	Ascend	FLBB Message	3.180000																										Ave 3								
$	25-Nov-2016	08:57:59.045	Ascend	FLBB Message	3.180000																										Pkt 2								
$	25-Nov-2016	08:58:14.002	Ascend	FLBB Message	1.530000																										99/99/99,99:99:99,695,50,700,120,539								
$	25-Nov-2016	08:58:13.886	Ascend	Optode Message	1.530000																											MEASUREMENT,4330,2592,O2Concentration[uM],367.792,AirSaturation[%],125.874,Temperature[Deg.C],18.571,CalPhase[Deg],27.002,TCPhase[Deg],27.823,C1RPh[Deg],35.768,C2RPh[Deg],7.945,C1Amp[mV],493.4,C2Amp[mV],757.5,RawTemp[mV],289.6							
$	25-Nov-2016	08:58:13.886	Ascend	Optode Message	1.530000																											#							
$	25-Nov-2016	08:58:14.582	Ascend	FLBB Message	1.350000																										99/99/99,99:99:99,695,50,700,115,538								
$	25-Nov-2016	08:58:14.687	Ascend	FLBB Message	1.350000																																		
$	25-Nov-2016	08:58:14.791	Ascend	FLBB Message	1.350000																										Ser FLBB2K-3310								
$	25-Nov-2016	08:58:14.898	Ascend	FLBB Message	1.350000																										Ver FLNTU 4.08								
$	25-Nov-2016	08:58:15.002	Ascend	FLBB Message	1.350000																										Ave 3								
$	25-Nov-2016	08:58:15.107	Ascend	FLBB Message	1.350000																										Pkt 2								
$	25-Nov-2016	08:58:16.209	Init Profile	Unparsed SBE41 Message						profile stopped  																													
$	25-Nov-2016	08:58:19.189	Init Profile	Unparsed SBE41 Message						S>stopprofile  																													
$	25-Nov-2016	08:58:20.368	Init Profile	Unparsed SBE41 Message						S>qsr  																													
$	25-Nov-2016	08:58:20.390	Init Profile	Unparsed SBE41 Message						powering down  																													
$	25-Nov-2016	08:58:44.749	Init Profile	Unparsed SBE41 Message						S>SBE 41CP V 7.2.5  																													
$	25-Nov-2016	08:59:47.196	Dump CP Data	Unparsed SBE41 Message						S>  																													
$	25-Nov-2016	08:59:53.173	Dump CP Data	Unparsed SBE41 Message						S>binaverage  																													
$	25-Nov-2016	08:59:53.344	Dump CP Data	Unparsed SBE41 Message						samples = 163, maxPress = 9.59  																													
$	25-Nov-2016	09:00:01.293	Dump CP Data	Unparsed SBE41 Message						rd: 0  																													
$	25-Nov-2016	09:00:01.979	Dump CP Data	Unparsed SBE41 Message						rd: 128  																													
$	25-Nov-2016	09:00:02.164	Dump CP Data	Unparsed SBE41 Message						avg: 0  																													
$	25-Nov-2016	09:00:02.369	Dump CP Data	Unparsed SBE41 Message						  																													
$	25-Nov-2016	09:00:02.370	Dump CP Data	Unparsed SBE41 Message						done, nbins = 11  																													
$	25-Nov-2016	09:00:08.446	Dump CP Data	Unparsed SBE41 Message						S>da  																													
$	25-Nov-2016	09:00:08.446	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.483	Dump CP Data	Unparsed SBE41 Message						    2.02,  18.5564,  35.1885, 18  																													
$	25-Nov-2016	09:00:08.533	Dump CP Data	Unparsed SBE41 Message						    4.01,  18.5566,  35.1888, 15  																													
$	25-Nov-2016	09:00:08.582	Dump CP Data	Unparsed SBE41 Message						    6.10,  18.5565,  35.1888, 12  																													
$	25-Nov-2016	09:00:08.632	Dump CP Data	Unparsed SBE41 Message						    8.13,  18.5563,  35.1888, 19  																													
$	25-Nov-2016	09:00:08.754	Dump CP Data	Unparsed SBE41 Message						    9.48,  18.5562,  35.1889, 99  																													
$	25-Nov-2016	09:00:08.754	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.860	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.861	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.869	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.923	Dump CP Data	Unparsed SBE41 Message						    0.00,   0.0000,   0.0000, 0  																													
$	25-Nov-2016	09:00:08.943	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	09:00:14.474	Dump CP Data	Unparsed SBE41 Message						S>dah  																													
$	25-Nov-2016	09:00:14.474	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.576	Dump CP Data	Unparsed SBE41 Message						0014487C897512  																													
$	25-Nov-2016	09:00:14.577	Dump CP Data	Unparsed SBE41 Message						0028487D89750F  																													
$	25-Nov-2016	09:00:14.577	Dump CP Data	Unparsed SBE41 Message						003D487C89750C  																													
$	25-Nov-2016	09:00:14.578	Dump CP Data	Unparsed SBE41 Message						0051487C897513  																													
$	25-Nov-2016	09:00:14.682	Dump CP Data	Unparsed SBE41 Message						005F487C897563  																													
$	25-Nov-2016	09:00:14.683	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.683	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.684	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.790	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.790	Dump CP Data	Unparsed SBE41 Message						00000000000000  																													
$	25-Nov-2016	09:00:14.791	Dump CP Data	Unparsed SBE41 Message						upload complete  																													
$	25-Nov-2016	09:00:20.410	Dump CP Data	Unparsed SBE41 Message						S>  																													
