$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	26-Nov-2016	16:10:08.412	Surface Ops	Optode Message	0.020000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.775,AirSaturation[%],99.742,Temperature[Deg.C],17.850,CalPhase[Deg],29.618,TCPhase[Deg],30.439,C1RPh[Deg],38.418,C2RPh[Deg],7.979,C1Amp[mV],587.6,C2Amp[mV],775.4,RawTemp[mV],312.6							
$	26-Nov-2016	16:10:08.412	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	16:10:13.559	Surface Ops	Unparsed SBE41 Message						S>  																													
$	26-Nov-2016	16:10:58.149	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,50,700,109,538								
$	26-Nov-2016	16:10:57.982	Surface Ops	Optode Message	0.000000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.378,AirSaturation[%],99.659,Temperature[Deg.C],17.875,CalPhase[Deg],29.623,TCPhase[Deg],30.444,C1RPh[Deg],38.423,C2RPh[Deg],7.979,C1Amp[mV],587.7,C2Amp[mV],775.1,RawTemp[mV],311.8							
$	26-Nov-2016	16:10:57.985	Surface Ops	Optode Message	0.000000																											#							
$	26-Nov-2016	16:10:58.490	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,48,700,110,538								
$	26-Nov-2016	16:10:58.597	Surface Ops	FLBB Message	0.000000																																		
$	26-Nov-2016	16:10:58.703	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:10:58.807	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:10:58.911	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	26-Nov-2016	16:10:59.015	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	26-Nov-2016	16:11:34.570	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,49,700,114,538								
$	26-Nov-2016	16:11:34.677	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,113,538								
$	26-Nov-2016	16:11:34.782	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	16:11:34.886	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:11:34.990	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:11:35.098	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	16:11:35.202	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	16:11:59.053	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,47,700,104,538								
$	26-Nov-2016	16:11:57.982	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.709,AirSaturation[%],99.711,Temperature[Deg.C],17.846,CalPhase[Deg],29.623,TCPhase[Deg],30.444,C1RPh[Deg],38.423,C2RPh[Deg],7.980,C1Amp[mV],587.8,C2Amp[mV],775.4,RawTemp[mV],312.7							
$	26-Nov-2016	16:11:57.985	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	16:11:59.393	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,53,700,111,538								
$	26-Nov-2016	16:11:59.499	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	16:11:59.605	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:11:59.709	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:11:59.813	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	16:11:59.923	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	16:12:59.989	Surface Ops	FLBB Message	0.570000																										99/99/99,99:99:99,695,52,700,109,538								
$	26-Nov-2016	16:12:57.997	Surface Ops	Optode Message	0.570000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.859,AirSaturation[%],99.735,Temperature[Deg.C],17.833,CalPhase[Deg],29.622,TCPhase[Deg],30.443,C1RPh[Deg],38.422,C2RPh[Deg],7.979,C1Amp[mV],587.8,C2Amp[mV],775.5,RawTemp[mV],313.1							
$	26-Nov-2016	16:12:58.001	Surface Ops	Optode Message	0.570000																											#							
$	26-Nov-2016	16:13:00.330	Surface Ops	FLBB Message	0.570000																										99/99/99,99:99:99,695,53,700,111,538								
$	26-Nov-2016	16:13:00.438	Surface Ops	FLBB Message	0.570000																																		
$	26-Nov-2016	16:13:00.544	Surface Ops	FLBB Message	0.570000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:13:00.648	Surface Ops	FLBB Message	0.570000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:13:00.752	Surface Ops	FLBB Message	0.570000																										Ave 3								
$	26-Nov-2016	16:13:00.858	Surface Ops	FLBB Message	0.570000																										Pkt 2								
$	26-Nov-2016	16:13:58.047	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.701,AirSaturation[%],99.705,Temperature[Deg.C],17.844,CalPhase[Deg],29.624,TCPhase[Deg],30.445,C1RPh[Deg],38.424,C2RPh[Deg],7.979,C1Amp[mV],587.9,C2Amp[mV],775.7,RawTemp[mV],312.8							
$	26-Nov-2016	16:13:58.047	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	16:13:58.288	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,110,538								
$	26-Nov-2016	16:13:58.393	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,114,538								
$	26-Nov-2016	16:13:58.501	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	16:13:58.604	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:13:58.709	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:13:58.818	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	16:13:58.922	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	16:14:58.751	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,48,700,110,538								
$	26-Nov-2016	16:14:57.980	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.600,AirSaturation[%],99.655,Temperature[Deg.C],17.837,CalPhase[Deg],29.630,TCPhase[Deg],30.451,C1RPh[Deg],38.431,C2RPh[Deg],7.980,C1Amp[mV],588.1,C2Amp[mV],775.7,RawTemp[mV],313.0							
$	26-Nov-2016	16:14:57.984	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	16:14:59.091	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,53,700,112,538								
$	26-Nov-2016	16:14:59.197	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	16:14:59.303	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:14:59.409	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:14:59.514	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	16:14:59.618	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	26-Nov-2016	16:15:59.685	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,51,700,107,538								
$	26-Nov-2016	16:15:58.004	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],296.008,AirSaturation[%],99.745,Temperature[Deg.C],17.813,CalPhase[Deg],29.625,TCPhase[Deg],30.446,C1RPh[Deg],38.425,C2RPh[Deg],7.979,C1Amp[mV],588.0,C2Amp[mV],775.8,RawTemp[mV],313.7							
$	26-Nov-2016	16:15:58.007	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	16:16:00.032	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,52,700,113,538								
$	26-Nov-2016	16:16:00.139	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	16:16:00.243	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:16:00.347	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:16:00.452	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	16:16:00.558	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	26-Nov-2016	16:17:00.469	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,47,700,109,538								
$	26-Nov-2016	16:16:57.980	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.524,AirSaturation[%],99.619,Temperature[Deg.C],17.831,CalPhase[Deg],29.635,TCPhase[Deg],30.456,C1RPh[Deg],38.435,C2RPh[Deg],7.978,C1Amp[mV],588.1,C2Amp[mV],775.8,RawTemp[mV],313.2							
$	26-Nov-2016	16:16:57.984	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	16:17:00.810	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,51,700,110,538								
$	26-Nov-2016	16:17:00.918	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	16:17:01.022	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:17:01.130	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:17:01.234	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	16:17:01.338	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	16:17:58.446	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,46,700,112,538								
$	26-Nov-2016	16:17:57.979	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.654,AirSaturation[%],99.667,Temperature[Deg.C],17.834,CalPhase[Deg],29.630,TCPhase[Deg],30.451,C1RPh[Deg],38.429,C2RPh[Deg],7.979,C1Amp[mV],588.1,C2Amp[mV],775.8,RawTemp[mV],313.1							
$	26-Nov-2016	16:17:57.982	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	16:17:58.789	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,48,700,111,538								
$	26-Nov-2016	16:17:58.898	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	16:17:59.004	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:17:59.108	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:17:59.212	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	16:17:59.316	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	26-Nov-2016	16:18:59.559	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,51,700,110,538								
$	26-Nov-2016	16:18:57.983	Surface Ops	Optode Message	0.010000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.753,AirSaturation[%],99.704,Temperature[Deg.C],17.835,CalPhase[Deg],29.625,TCPhase[Deg],30.446,C1RPh[Deg],38.425,C2RPh[Deg],7.978,C1Amp[mV],587.8,C2Amp[mV],775.6,RawTemp[mV],313.0							
$	26-Nov-2016	16:18:57.986	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	16:18:59.903	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,107,538								
$	26-Nov-2016	16:19:00.009	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	16:19:00.113	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:19:00.217	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:19:00.324	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	16:19:00.433	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	16:20:00.394	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,51,700,113,539								
$	26-Nov-2016	16:19:57.979	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.189,AirSaturation[%],99.559,Temperature[Deg.C],17.857,CalPhase[Deg],29.637,TCPhase[Deg],30.458,C1RPh[Deg],38.437,C2RPh[Deg],7.978,C1Amp[mV],588.1,C2Amp[mV],775.6,RawTemp[mV],312.3							
$	26-Nov-2016	16:19:57.982	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	16:20:00.734	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,50,700,113,538								
$	26-Nov-2016	16:20:00.842	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	16:20:00.945	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	16:20:01.054	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	16:20:01.158	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	16:20:01.262	Surface Ops	FLBB Message	0.020000																										Pkt 2								
