$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	26-Nov-2016	17:10:38.839	Surface Ops	Optode Message	0.000000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.809,AirSaturation[%],99.773,Temperature[Deg.C],17.859,CalPhase[Deg],29.613,TCPhase[Deg],30.434,C1RPh[Deg],38.414,C2RPh[Deg],7.980,C1Amp[mV],587.2,C2Amp[mV],775.1,RawTemp[mV],312.3							
$	26-Nov-2016	17:10:38.839	Surface Ops	Optode Message	0.000000																											#							
$	26-Nov-2016	17:10:58.047	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,43,700,113,538								
$	26-Nov-2016	17:10:58.046	Surface Ops	Optode Message	0.010000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.616,AirSaturation[%],99.702,Temperature[Deg.C],17.856,CalPhase[Deg],29.622,TCPhase[Deg],30.443,C1RPh[Deg],38.422,C2RPh[Deg],7.979,C1Amp[mV],587.3,C2Amp[mV],775.0,RawTemp[mV],312.4							
$	26-Nov-2016	17:10:58.046	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	17:10:58.390	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,51,700,109,538								
$	26-Nov-2016	17:10:58.497	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	17:10:58.601	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:10:58.707	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:10:58.812	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	17:10:58.918	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	17:11:59.051	Surface Ops	FLBB Message	-0.010000																										99/99/99,99:99:99,695,45,700,115,538								
$	26-Nov-2016	17:11:57.981	Surface Ops	Optode Message	-0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.718,AirSaturation[%],99.754,Temperature[Deg.C],17.865,CalPhase[Deg],29.614,TCPhase[Deg],30.435,C1RPh[Deg],38.415,C2RPh[Deg],7.980,C1Amp[mV],587.2,C2Amp[mV],774.8,RawTemp[mV],312.1							
$	26-Nov-2016	17:11:57.984	Surface Ops	Optode Message	-0.010000																											#							
$	26-Nov-2016	17:11:59.395	Surface Ops	FLBB Message	-0.010000																										99/99/99,99:99:99,695,47,700,109,538								
$	26-Nov-2016	17:11:59.502	Surface Ops	FLBB Message	-0.010000																																		
$	26-Nov-2016	17:11:59.606	Surface Ops	FLBB Message	-0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:11:59.713	Surface Ops	FLBB Message	-0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:11:59.817	Surface Ops	FLBB Message	-0.010000																										Ave 3								
$	26-Nov-2016	17:11:59.921	Surface Ops	FLBB Message	-0.010000																										Pkt 2								
$	26-Nov-2016	17:12:04.997	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,52,700,111,538								
$	26-Nov-2016	17:12:05.105	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,49,700,113,538								
$	26-Nov-2016	17:12:05.210	Surface Ops	FLBB Message	0.000000																																		
$	26-Nov-2016	17:12:05.314	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:12:05.420	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:12:05.527	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	26-Nov-2016	17:12:05.631	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	26-Nov-2016	17:12:59.985	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,110,538								
$	26-Nov-2016	17:12:57.996	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.488,AirSaturation[%],99.703,Temperature[Deg.C],17.878,CalPhase[Deg],29.617,TCPhase[Deg],30.438,C1RPh[Deg],38.418,C2RPh[Deg],7.979,C1Amp[mV],587.3,C2Amp[mV],775.0,RawTemp[mV],311.7							
$	26-Nov-2016	17:12:57.999	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	17:13:00.326	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,49,700,117,538								
$	26-Nov-2016	17:13:00.433	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	17:13:00.543	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:13:00.647	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:13:00.751	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	17:13:00.856	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	17:13:58.050	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,47,700,109,538								
$	26-Nov-2016	17:13:58.048	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.308,AirSaturation[%],99.655,Temperature[Deg.C],17.884,CalPhase[Deg],29.622,TCPhase[Deg],30.443,C1RPh[Deg],38.422,C2RPh[Deg],7.979,C1Amp[mV],587.4,C2Amp[mV],775.0,RawTemp[mV],311.5							
$	26-Nov-2016	17:13:58.048	Surface Ops	Optode Message	0.000000																											#							
$	26-Nov-2016	17:13:58.404	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,50,700,110,538								
$	26-Nov-2016	17:13:58.512	Surface Ops	FLBB Message	0.000000																																		
$	26-Nov-2016	17:13:58.615	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:13:58.720	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:13:58.826	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	26-Nov-2016	17:13:58.932	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	26-Nov-2016	17:14:58.748	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,50,700,114,539								
$	26-Nov-2016	17:14:57.979	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.814,AirSaturation[%],99.773,Temperature[Deg.C],17.859,CalPhase[Deg],29.613,TCPhase[Deg],30.434,C1RPh[Deg],38.354,C2RPh[Deg],7.920,C1Amp[mV],587.2,C2Amp[mV],775.0,RawTemp[mV],312.3							
$	26-Nov-2016	17:14:57.982	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	17:14:59.091	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,47,700,114,538								
$	26-Nov-2016	17:14:59.196	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	17:14:59.300	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:14:59.408	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:14:59.513	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	17:14:59.617	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	26-Nov-2016	17:15:59.685	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,52,700,111,539								
$	26-Nov-2016	17:15:58.002	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.461,AirSaturation[%],99.653,Temperature[Deg.C],17.858,CalPhase[Deg],29.627,TCPhase[Deg],30.448,C1RPh[Deg],38.427,C2RPh[Deg],7.979,C1Amp[mV],587.5,C2Amp[mV],775.1,RawTemp[mV],312.3							
$	26-Nov-2016	17:15:58.006	Surface Ops	Optode Message	0.020000																											#							
$	26-Nov-2016	17:16:00.039	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,51,700,112,538								
$	26-Nov-2016	17:16:00.171	Surface Ops	FLBB Message	0.020000																																		
$	26-Nov-2016	17:16:00.275	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:16:00.379	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:16:00.487	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	26-Nov-2016	17:16:00.593	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	26-Nov-2016	17:17:00.588	Surface Ops	FLBB Message	0.040000																										99/99/99,99:99:99,695,48,700,116,538								
$	26-Nov-2016	17:16:57.981	Surface Ops	Optode Message	0.040000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.909,AirSaturation[%],99.764,Temperature[Deg.C],17.839,CalPhase[Deg],29.618,TCPhase[Deg],30.439,C1RPh[Deg],38.419,C2RPh[Deg],7.980,C1Amp[mV],587.3,C2Amp[mV],775.2,RawTemp[mV],312.9							
$	26-Nov-2016	17:16:57.984	Surface Ops	Optode Message	0.040000																											#							
$	26-Nov-2016	17:17:00.931	Surface Ops	FLBB Message	0.040000																										99/99/99,99:99:99,695,49,700,113,538								
$	26-Nov-2016	17:17:01.036	Surface Ops	FLBB Message	0.040000																																		
$	26-Nov-2016	17:17:01.142	Surface Ops	FLBB Message	0.040000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:17:01.246	Surface Ops	FLBB Message	0.040000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:17:01.353	Surface Ops	FLBB Message	0.040000																										Ave 3								
$	26-Nov-2016	17:17:01.457	Surface Ops	FLBB Message	0.040000																										Pkt 2								
$	26-Nov-2016	17:17:58.450	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,49,700,111,539								
$	26-Nov-2016	17:17:57.978	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],296.101,AirSaturation[%],99.792,Temperature[Deg.C],17.821,CalPhase[Deg],29.618,TCPhase[Deg],30.439,C1RPh[Deg],38.417,C2RPh[Deg],7.978,C1Amp[mV],587.3,C2Amp[mV],775.2,RawTemp[mV],313.5							
$	26-Nov-2016	17:17:57.981	Surface Ops	Optode Message	0.000000																											#							
$	26-Nov-2016	17:17:58.792	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,53,700,106,538								
$	26-Nov-2016	17:17:58.899	Surface Ops	FLBB Message	0.000000																																		
$	26-Nov-2016	17:17:59.005	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:17:59.109	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:17:59.214	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	26-Nov-2016	17:17:59.318	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	26-Nov-2016	17:18:59.556	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,52,700,111,539								
$	26-Nov-2016	17:18:57.982	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],296.098,AirSaturation[%],99.784,Temperature[Deg.C],17.818,CalPhase[Deg],29.620,TCPhase[Deg],30.441,C1RPh[Deg],38.420,C2RPh[Deg],7.980,C1Amp[mV],587.5,C2Amp[mV],775.4,RawTemp[mV],313.6							
$	26-Nov-2016	17:18:57.985	Surface Ops	Optode Message	0.010000																											#							
$	26-Nov-2016	17:18:59.900	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,47,700,112,538								
$	26-Nov-2016	17:19:00.007	Surface Ops	FLBB Message	0.010000																																		
$	26-Nov-2016	17:19:00.111	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:19:00.217	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:19:00.322	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	26-Nov-2016	17:19:00.431	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	26-Nov-2016	17:20:00.488	Surface Ops	FLBB Message	0.300000																										99/99/99,99:99:99,695,50,700,108,538								
$	26-Nov-2016	17:19:57.980	Surface Ops	Optode Message	0.300000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.800,AirSaturation[%],99.692,Temperature[Deg.C],17.822,CalPhase[Deg],29.629,TCPhase[Deg],30.450,C1RPh[Deg],38.429,C2RPh[Deg],7.979,C1Amp[mV],587.6,C2Amp[mV],775.3,RawTemp[mV],313.5							
$	26-Nov-2016	17:19:57.982	Surface Ops	Optode Message	0.300000																											#							
$	26-Nov-2016	17:20:00.833	Surface Ops	FLBB Message	0.300000																										99/99/99,99:99:99,695,50,700,113,538								
$	26-Nov-2016	17:20:00.938	Surface Ops	FLBB Message	0.300000																																		
$	26-Nov-2016	17:20:01.042	Surface Ops	FLBB Message	0.300000																										Ser FLBB2K-3310								
$	26-Nov-2016	17:20:01.146	Surface Ops	FLBB Message	0.300000																										Ver FLNTU 4.08								
$	26-Nov-2016	17:20:01.253	Surface Ops	FLBB Message	0.300000																										Ave 3								
$	26-Nov-2016	17:20:01.359	Surface Ops	FLBB Message	0.300000																										Pkt 2								
