$	Date	Time	State	Comment	Pressure	Bel Pos	Batt Volt	Batt Amps	CTD Temp	CTD Sal	Raw Press	LPF Press	LPF Vel	Diff Vel	Diff 2 Vel	Slope of LPF Press.	CTD Sample Period	LPS331 Press	LPS311 Temp	SHT21 Humid	SHT21 Temp	PID SP	PID PV	PID Y1	PID Y2	PID I	PID v	PID u	PID uCPS	PID h	FLBB	Topt	Tphase	Rphase	NO3	pH	GPS	Irid Modem	
$	27-Nov-2016	00:54:42.159	Surface Ops	Optode Message	0.010000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.312,AirSaturation[%],99.621,Temperature[Deg.C],17.867,CalPhase[Deg],29.629,TCPhase[Deg],30.450,C1RPh[Deg],38.427,C2RPh[Deg],7.977,C1Amp[mV],583.5,C2Amp[mV],771.3,RawTemp[mV],312.0							
$	27-Nov-2016	00:54:42.159	Surface Ops	Optode Message	0.010000																											#							
$	27-Nov-2016	00:54:46.161	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	00:54:58.345	Surface Ops	FLBB Message	0.600000																										99/99/99,99:99:99,695,52,700,115,538								
$	27-Nov-2016	00:54:57.981	Surface Ops	Optode Message	0.600000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.596,AirSaturation[%],99.688,Temperature[Deg.C],17.853,CalPhase[Deg],29.624,TCPhase[Deg],30.445,C1RPh[Deg],38.424,C2RPh[Deg],7.979,C1Amp[mV],583.4,C2Amp[mV],771.2,RawTemp[mV],312.5							
$	27-Nov-2016	00:54:57.984	Surface Ops	Optode Message	0.600000																											#							
$	27-Nov-2016	00:54:58.683	Surface Ops	FLBB Message	0.600000																										99/99/99,99:99:99,695,51,700,113,538								
$	27-Nov-2016	00:54:58.791	Surface Ops	FLBB Message	0.600000																																		
$	27-Nov-2016	00:54:58.897	Surface Ops	FLBB Message	0.600000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:54:59.001	Surface Ops	FLBB Message	0.600000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:54:59.105	Surface Ops	FLBB Message	0.600000																										Ave 3								
$	27-Nov-2016	00:54:59.209	Surface Ops	FLBB Message	0.600000																										Pkt 2								
$	27-Nov-2016	00:55:01.031	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	00:55:59.557	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,50,700,111,538								
$	27-Nov-2016	00:55:57.978	Surface Ops	Optode Message	0.020000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.563,AirSaturation[%],99.687,Temperature[Deg.C],17.858,CalPhase[Deg],29.623,TCPhase[Deg],30.444,C1RPh[Deg],38.422,C2RPh[Deg],7.978,C1Amp[mV],583.4,C2Amp[mV],771.0,RawTemp[mV],312.3							
$	27-Nov-2016	00:55:57.981	Surface Ops	Optode Message	0.020000																											#							
$	27-Nov-2016	00:55:59.897	Surface Ops	FLBB Message	0.020000																										99/99/99,99:99:99,695,52,700,113,538								
$	27-Nov-2016	00:56:00.003	Surface Ops	FLBB Message	0.020000																																		
$	27-Nov-2016	00:56:00.107	Surface Ops	FLBB Message	0.020000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:56:00.211	Surface Ops	FLBB Message	0.020000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:56:00.317	Surface Ops	FLBB Message	0.020000																										Ave 3								
$	27-Nov-2016	00:56:00.423	Surface Ops	FLBB Message	0.020000																										Pkt 2								
$	27-Nov-2016	00:56:00.991	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	00:56:08.556	Surface Ops	FLBB Message	0.420000																										99/99/99,99:99:99,695,50,700,111,538								
$	27-Nov-2016	00:56:08.663	Surface Ops	FLBB Message	0.420000																										99/99/99,99:99:99,695,49,700,114,538								
$	27-Nov-2016	00:56:08.771	Surface Ops	FLBB Message	0.420000																																		
$	27-Nov-2016	00:56:08.874	Surface Ops	FLBB Message	0.420000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:56:08.979	Surface Ops	FLBB Message	0.420000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:56:09.083	Surface Ops	FLBB Message	0.420000																										Ave 3								
$	27-Nov-2016	00:56:09.191	Surface Ops	FLBB Message	0.420000																										Pkt 2								
$	27-Nov-2016	00:57:00.366	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,52,700,109,538								
$	27-Nov-2016	00:56:57.979	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.443,AirSaturation[%],99.667,Temperature[Deg.C],17.868,CalPhase[Deg],29.623,TCPhase[Deg],30.444,C1RPh[Deg],38.422,C2RPh[Deg],7.977,C1Amp[mV],583.3,C2Amp[mV],771.1,RawTemp[mV],312.0							
$	27-Nov-2016	00:56:57.982	Surface Ops	Optode Message	0.000000																											#							
$	27-Nov-2016	00:57:00.707	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,49,700,109,538								
$	27-Nov-2016	00:57:00.812	Surface Ops	FLBB Message	0.000000																																		
$	27-Nov-2016	00:57:00.916	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:57:01.025	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:57:01.129	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	27-Nov-2016	00:57:01.233	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	27-Nov-2016	00:57:01.024	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	00:57:58.546	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,50,700,111,538								
$	27-Nov-2016	00:57:57.980	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.028,AirSaturation[%],99.595,Temperature[Deg.C],17.901,CalPhase[Deg],29.625,TCPhase[Deg],30.446,C1RPh[Deg],38.425,C2RPh[Deg],7.978,C1Amp[mV],583.4,C2Amp[mV],771.1,RawTemp[mV],310.9							
$	27-Nov-2016	00:57:57.983	Surface Ops	Optode Message	0.000000																											#							
$	27-Nov-2016	00:57:58.892	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,50,700,109,538								
$	27-Nov-2016	00:57:59.001	Surface Ops	FLBB Message	0.000000																																		
$	27-Nov-2016	00:57:59.105	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:57:59.209	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:57:59.313	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	27-Nov-2016	00:57:59.419	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	27-Nov-2016	00:58:01.030	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	00:58:59.554	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,48,700,114,538								
$	27-Nov-2016	00:58:57.979	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.494,AirSaturation[%],99.696,Temperature[Deg.C],17.874,CalPhase[Deg],29.619,TCPhase[Deg],30.440,C1RPh[Deg],38.418,C2RPh[Deg],7.978,C1Amp[mV],583.2,C2Amp[mV],771.0,RawTemp[mV],311.8							
$	27-Nov-2016	00:58:57.982	Surface Ops	Optode Message	0.000000																											#							
$	27-Nov-2016	00:58:59.894	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,49,700,117,538								
$	27-Nov-2016	00:59:00.000	Surface Ops	FLBB Message	0.000000																																		
$	27-Nov-2016	00:59:00.103	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	27-Nov-2016	00:59:00.211	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	27-Nov-2016	00:59:00.316	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	27-Nov-2016	00:59:00.422	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	27-Nov-2016	00:59:00.991	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	01:00:00.366	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,111,538								
$	27-Nov-2016	00:59:57.978	Surface Ops	Optode Message	0.010000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.329,AirSaturation[%],99.648,Temperature[Deg.C],17.878,CalPhase[Deg],29.624,TCPhase[Deg],30.445,C1RPh[Deg],38.423,C2RPh[Deg],7.978,C1Amp[mV],583.2,C2Amp[mV],771.0,RawTemp[mV],311.7							
$	27-Nov-2016	00:59:57.982	Surface Ops	Optode Message	0.010000																											#							
$	27-Nov-2016	01:00:00.711	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,50,700,113,538								
$	27-Nov-2016	01:00:00.816	Surface Ops	FLBB Message	0.010000																																		
$	27-Nov-2016	01:00:00.920	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	27-Nov-2016	01:00:01.025	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	27-Nov-2016	01:00:01.131	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	27-Nov-2016	01:00:01.237	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	27-Nov-2016	01:00:01.024	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	01:00:58.546	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,50,700,112,538								
$	27-Nov-2016	01:00:57.980	Surface Ops	Optode Message	0.000000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.621,AirSaturation[%],99.744,Temperature[Deg.C],17.876,CalPhase[Deg],29.613,TCPhase[Deg],30.434,C1RPh[Deg],38.413,C2RPh[Deg],7.979,C1Amp[mV],583.1,C2Amp[mV],771.1,RawTemp[mV],311.7							
$	27-Nov-2016	01:00:57.983	Surface Ops	Optode Message	0.000000																											#							
$	27-Nov-2016	01:00:58.884	Surface Ops	FLBB Message	0.000000																										99/99/99,99:99:99,695,53,700,112,538								
$	27-Nov-2016	01:00:58.990	Surface Ops	FLBB Message	0.000000																																		
$	27-Nov-2016	01:00:59.097	Surface Ops	FLBB Message	0.000000																										Ser FLBB2K-3310								
$	27-Nov-2016	01:00:59.202	Surface Ops	FLBB Message	0.000000																										Ver FLNTU 4.08								
$	27-Nov-2016	01:00:59.306	Surface Ops	FLBB Message	0.000000																										Ave 3								
$	27-Nov-2016	01:00:59.413	Surface Ops	FLBB Message	0.000000																										Pkt 2								
$	27-Nov-2016	01:01:01.020	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	01:01:59.252	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,47,700,112,538								
$	27-Nov-2016	01:01:58.080	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.062,AirSaturation[%],99.604,Temperature[Deg.C],17.900,CalPhase[Deg],29.625,TCPhase[Deg],30.446,C1RPh[Deg],38.425,C2RPh[Deg],7.979,C1Amp[mV],583.3,C2Amp[mV],771.0,RawTemp[mV],311.0							
$	27-Nov-2016	01:01:58.083	Surface Ops	Optode Message	0.010000																											#							
$	27-Nov-2016	01:01:59.596	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,46,700,113,538								
$	27-Nov-2016	01:01:59.702	Surface Ops	FLBB Message	0.010000																																		
$	27-Nov-2016	01:01:59.805	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	27-Nov-2016	01:01:59.912	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	27-Nov-2016	01:02:00.018	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	27-Nov-2016	01:02:00.122	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	27-Nov-2016	01:02:01.088	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	01:03:00.174	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,46,700,113,539								
$	27-Nov-2016	01:02:57.981	Surface Ops	Optode Message	0.130000																											MEASUREMENT,4330,2592,O2Concentration[uM],295.063,AirSaturation[%],99.607,Temperature[Deg.C],17.902,CalPhase[Deg],29.624,TCPhase[Deg],30.445,C1RPh[Deg],38.424,C2RPh[Deg],7.979,C1Amp[mV],583.3,C2Amp[mV],771.0,RawTemp[mV],310.9							
$	27-Nov-2016	01:02:57.985	Surface Ops	Optode Message	0.130000																											#							
$	27-Nov-2016	01:03:00.517	Surface Ops	FLBB Message	0.130000																										99/99/99,99:99:99,695,49,700,112,538								
$	27-Nov-2016	01:03:00.622	Surface Ops	FLBB Message	0.130000																																		
$	27-Nov-2016	01:03:00.728	Surface Ops	FLBB Message	0.130000																										Ser FLBB2K-3310								
$	27-Nov-2016	01:03:00.832	Surface Ops	FLBB Message	0.130000																										Ver FLNTU 4.08								
$	27-Nov-2016	01:03:00.938	Surface Ops	FLBB Message	0.130000																										Ave 3								
$	27-Nov-2016	01:03:01.043	Surface Ops	FLBB Message	0.130000																										Pkt 2								
$	27-Nov-2016	01:03:01.043	Surface Ops	Unparsed SBE41 Message						S>  																													
$	27-Nov-2016	01:03:58.245	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,47,700,115,538								
$	27-Nov-2016	01:03:57.979	Surface Ops	Optode Message	0.010000																											%!MEASUREMENT,4330,2592,O2Concentration[uM],295.589,AirSaturation[%],99.752,Temperature[Deg.C],17.886,CalPhase[Deg],29.611,TCPhase[Deg],30.432,C1RPh[Deg],38.410,C2RPh[Deg],7.978,C1Amp[mV],582.9,C2Amp[mV],770.9,RawTemp[mV],311.4							
$	27-Nov-2016	01:03:57.982	Surface Ops	Optode Message	0.010000																											#							
$	27-Nov-2016	01:03:58.586	Surface Ops	FLBB Message	0.010000																										99/99/99,99:99:99,695,49,700,116,538								
$	27-Nov-2016	01:03:58.691	Surface Ops	FLBB Message	0.010000																																		
$	27-Nov-2016	01:03:58.795	Surface Ops	FLBB Message	0.010000																										Ser FLBB2K-3310								
$	27-Nov-2016	01:03:58.901	Surface Ops	FLBB Message	0.010000																										Ver FLNTU 4.08								
$	27-Nov-2016	01:03:59.008	Surface Ops	FLBB Message	0.010000																										Ave 3								
$	27-Nov-2016	01:03:59.112	Surface Ops	FLBB Message	0.010000																										Pkt 2								
$	27-Nov-2016	01:04:01.046	Surface Ops	Unparsed SBE41 Message						S>  																													
